We started rebuilding the DOPO in the lab even though the new optical table hasn't arrived. For this reason, we are using a 1 ft x 3 ft x 0.5 in anodized aluminum breadboard which we can then attach handles to move the setup. This also makes the prototype's footprint smaller. The first thing we did as usual was settle on a beam height. The beam height used before was ~ 3in (~ 75mm), and since the EOM, Faraday Isolator, and RFPD are nominally at that height from the breadboard, the only thing we had to fix was the pump laser head. The bare height is 55 mm, so we stacked two 9 mm thorlabs bases together, bolted them down to the breadboard and then mounted the NPRO laser head on the top. Finally, using a level we secured it to the breadboard using the three points and long 1/4-20 screws while being careful as we didn't want to flex the head too much.
Next up is aligning the laser to the EOM and Faraday Isolator. For this, we will use the beam profiles measured late last year. Another task ahead is to implement the new mount for the cavity.