The heater was turned on at 3:13pm on Friday 2/4.
We specified a set temperature of 123K. However, the CTC100 PI control included a 1 W lower limit on the input to the heater, so there was a steady load of 1 W applied to the Silicon Workpiece over the weekend.
At 16:01 the cryocooler was turned off to start the warmup.
The CTC100 PI control was configured with a setpoint of 250 K on the Workpiece RTD, to aid in the warmup, and an allowable power range from 0 W to 22 W.