Quick log establishing the maximum power for our thermal actuation:
Heater: HSA25100RJ from TE, unknown sourcing. Acetone wiping cleaned off p/n and markings from body, should engrave at next opportunity, but [Attachment 1] from many months ago shows the p/n. Note that this is not the current mounting configuration - [Attachment 2] is more similar to current mounting. Anyway, according to the datasheet (now added to the QIL wiki at Documentation > Manuals) this heater is rated for 25W and has a resistance of 100Ω.
Leads: unknown, and not super important unless we had tiny hair conductor - I am not in lab presently, but it appears from our connector (Lesker FTACIR19AC) that we must have 20-24 AWG,
Carrying the 7, the current through the Heater will be 0.25 A at max actuation, and the 20-24 AWG insulated copper leads will have plenty of ampacity for this load (plus, they are cooled, so normal current capacity considerations fly out the window a bit).
Conclusion: 25W actuation will be the limit that we will apply in the CTC100 temperature actuation routine.
From first trials yesterday, the response at ~20W (at starting temperatures around 80K) appears to be on the order of 1 degree per minute, which should be just fine for actuating to maintain a +/- 1 degree constant setpoint with static thermal loads. More to follow on trials implementing temperature control.
Note that the QIL Wiki points to the DCC (which contains a budget that was helpful resources to trace these purchases), the datasheets and other documentation, and also points to the QIL Cryo Vacuum Chamber photo album, which hosts the images below.