We tested the cryo cooler and the turbo pump this afternoon. We ran the cryo cooler for two hours. The equalization pressure is 275psi(the pressure before we turn it on) and operating pressure (pressure after running for two hours) is 295psi(attachment 1). The operating pressure is lower than expected; the manual indicates the pressure is expected to be 300-320psi.
We cycled the turbo pump. It appears to be functioning properly.