| ATF eLog
||Not logged in
Message ID: 2334
Entry time: Sun May 5 20:52:16 2019
In reply to: 2333
Frequency noise measurement of 2 micron source
Looks good! A few comments:
- The LPF should be a 1.9 MHz LP from Mini-circuits
- The SR560 should be set with a 1 MHz low pass only
- No Rubidium needed. We just want to use two IFR/Marconi synthesizers so that they can share the same time standard.
- We will need a RF Amp to drive the AOM hard enough - check the manual and see what level is required to get the max diffraction.