Not logged in
Wed Oct 8 13:40:02 2008, Dmass, Laser, PSL, Chiller Noise
Sat Nov 8 21:38:17 2008, Dmass, Laser, PSL, Chiller Noise
Sat Nov 8 21:38:17 2008
In reply to:
The new chiller was making a noise (as reported to me by John).
It was emitting a continuous high pitched tone, and the display alternated between the set point temp, and "d1" or "di".
I hit enter twice, and it stopped. To be safe,
I have turned the 35W off
until I spend some time down here and make sure
that the chiller is ok. I have left the chiller on.
The chiller was making the noise again. Again it said di on the screen. The noise and warning went away when I pushed any button on the front panel. I think it's just a warning that the water no longer qualifies as DI, as the light on the back of the chiller was red (it's green when the resistance is high, red when low - either change the di cartridge or figure out how toe disable the noise. As we don't necessarily want to use DI water because of the metal plumbing,
I am inclined to disable the warning noise if I can.