Suddenly something dirty emerged in the lab. What is this? It looks like an insulation foam or similar, but is quite degraded and emits a lot of particulates.
This does not belong to the lab. I don't see piping above this area which shows broken insulation or anything. All the pipes in the room are painted white.
The only possibility is that it comes from the hole between the next lab (CRIME Lab). I found that the A.C. today is much stronger and colder than last week. And there is a positive pressure from CRIME Lab. Maybe the foam was pushed out from the hole due to the differential pressure (or any RF cable action).
The Windows laptop for WincamD/Beam'R2 (DELL Vostro3300) was not functional.
- Windows 7 got stuck in the starting up process (Google "startup repair loop")
- The battery can't charge and the adapter connection is flaky
I decided to newly install Win10.
I made a new bootable Win10 DVD from the ISO downloaded from IMSS. The ISO file was converted to CDR using Disk Utility on Mac.
This deleted the past disk partitions. The installation process has no trouble and Win10 ran successfully. The machine is slow but still acceptable for our purpose.
Dataray Version 7.1H25Bk was downloaded from the vendor website https://dataray.com/blogs/software/downloads and installed successfully.
The devices ran as expected by connecting the heads and selecting the proper device in the software.
Then, the Win10 fell into "Hibernation Loop" and "Shutdown loop" (after disabling hibernation in the safe mode).
This is probably the combination of extremely slow windows update (feature update i.e. beta OS update) and the occasional shutdown due to the flakiness of the AC connection
Win10 was reinstalled and automatic Win update was disabled via windows policy manager or something like that. Still, it tries to download and update some of the updates (what's happening there!?
Here are my strong recommendations on how to use this laptop
Aaron took the set to Cryo lab
Completed work of the previous months: [Jul] [Aug] [Sep] [Oct] [Nov] [Dec]
We took apart the unit removed from the 3rd IFO (Unit serial number aLIGO #3, XTAL 10252004) to see what makes it tick. Koji has done a fine job of adding the plots of the impedance data to this log book. Attached are some details of the physical construction showing the capacitor values used in shunt before the coils.
Last Friday, new workbenches were installed. Vladimir got a new table and a cleanroom stool.
The other two workbenches were also nicely set.
Suspension cage and transportation box: 250.8lb
Suspension cage and transportation box: 150.2lb ==> 100.6lb ==> 45,630 g
Metal Breadboard: 7261 g
Glass Breadboard and transportation fixture: 16382 g
Transportation fixture only: 9432 g ==> 6950 g
Added mass (up to now): 300 g ==> 7250 g
We worked on the OMC test over the weekend.
- At the beginning, the measured OMC transmission was ~85% even after subtracting the junk light and sidebands from the calculation.
- A pretty visible (by eye) dust were on CM1. Also a small residue of First Contact was found on the same mirror.
- We applied FC only on CM1 to remove these.
- The measued transmission went up to the level of 96%.
- We swept the incident power from 0.3mW to 30mW in order to see the dependence of the transmission against the incident power.
- The variation of the transmission ~10% was observed (attached figure 1, Red). This was compared with the similar dependence measured at Caltech (Magenta)
- So, the reduction of the transmission was observed as in eLIGO, although the measurements at Caltech and LLO are not consistent.
- Can this be attributed to the dependence of the PD efficiency? We measured the incident power on the PDs together with the preamp DC output. (Figure.2)
This gives us how the responsivity changes with the incident power.
- Nevertheless, the dependence remains. We'll make more accurate measurement today.
[Koji Zach Suresh]
The OMC arrived at LLO without any destruction!
OMC installed in HAM6!
For various reasons, I had to switch NPROs (from the LightWave 126 to the Innolight Prometheus).
I installed the laser, realigned the polarization and modulation optics, and then began launching the beam into the fiber, though I have not coupled any light yet.
A diagram is below. Since I do not yet have the AOM, I have shown that future path with a dotted line. Since we will not need to make AMTFs and have a subcarrier at the same time, I have chosen to overload the function of the PBS using the HWP after the AEOM. We will operate in one of two modes:
One thing that concerns me slightly: the Prometheus is a dual-output (1064nm/532nm) laser, with separate ports for each. I have blocked and locked out the green path physically, but there is some residual green light visible in the IR output. Since we are planning to do the OMC transmission testing with a Si-based Thorlabs power meter---which is more sensitive to green than IR---I am somewhat worried about the ensuing systematics. I *think* we can minimize the effect by detuning the doubling crystal temperature, but this remains to be verified.
EDIT (ZK): Valera says there should be a dichroic beam splitter in the lab that I can borrow. This should be enough to selectively suppress the green.
The breadboard (SN2) was loaded on the transportation fixture.
The laser side template was installed and the cavity mirrors were placed.
The laser beam will be resonated in the cavity next week.
POSTED to 40m ELOG
2nd optical test http://nodus.ligo.caltech.edu:8080/40m/13725
Monday, November 14, 2022
Camille and Koji did a "deep cleaning" of OMC#1:
1) Applied First Contact to the mirror surfaces. Removed first contact after ~10 minutes.
2) Acetone scrub of the mirror surfaces with a cotton swab.
3) Applied First Contact again. Removed after ~10 minutes. We left the FC paint on for the work on Thu.
The foggy spot on the input mirror was unchanged after the first round of First Contact. But the foggy spot came off during the acetone scrub.
EDIT (ZK): Koji points out that (1 - Ti) should really be the non-resonant reflectivity of the aligned cavity, which is much closer to 1. However, it should *actually* be the non-resonant reflectivity of the entire OMC assembly, including the steering mirror (see bottom of post). The steering mirror has T ~ 0.3%, so the true results are somewhere between my numbers and those with (1 - Ti) -> 1. In practice, though, these effects are swamped by the other errors.
More information about the power-dependent visibility measurement:
As a blanket statement, this measurement was done by exact analogy to those made by Sam and Sheon during S6 (c.f. LHO iLog 11/7/2011 and technical note T1100562), since it was supposed to be a verification that this effect still remains. There are absolutely better ways to do (i.e., ways that should give lower measurement error), and these should be investigated for our characterization. Obviously, I volunteer.
All measurements were made by reading the output voltages produced by photodetectors at the REFL and TRANS ports. The REFL PD is a BBPD (DC output), and the TRANS is a PDA255. Both these PDs were calibrated using a Thorlabs power meter (Controller: PM100D; Head: S12XC series photodiode-based---not sure if X = 0,2... Si or Ge) at the lowest and highest power settings, and these results agreed to the few-percent level. This can be a major source of error.
The power was adjusted using the HWP/PBS combination towards the beginning of the experiment. For reference, an early layout of the test setup can be seen in LLO:5978 (though, as mentioned above, the REFL and TRANS PDs have been replaced since then---see LLO:5994). This may or may not be a "clean" way to change the power, but the analysis should take the effect of junk light into account.
Below is an explanation of the three traces in the plot. First:
Now, the traces
The error bars in the measurement were dominated, roughly equally, by 1) systematic error from calibration of the PDs with the power meter, and 2) error from noise in the REFL_L measurement (since the absolute AC noise level in TRANS and REFL_L is the same, and TRANS >> REFL_L, the SNR of the latter is worse).
(1) can be helped by making ALL measurements with a single device. I recommend using something precise and portable like the power meter to make measurements at all the necessary ports. For REFL_L/UL, we can place a beam splitter before the REFL PD, and---after calibrating for the T of this splitter very well using the same power meter---both states can be measured at this port.
(2) can probably be helped by taking longer averaging, though at some point we run into the stability of the power setting itself. Something like 30-60s should be enough to remove the effects of the REFL_L noise, which is concentrated in the few-Hz region in the LLO setup.
One more thing I forgot was the finite transmission of the steering mirror at the OMC input (the transmission of this mirror goes to the QPDs). This will add a fixed error of 0.3%, and I will take it into account in the future.
I found that, in fact, I had lowered the modulation depth since when I measured it to be 0.45 rads --> Psb = 0.1.
Here is the sweep measurement:
This is Psb = 0.06 --> gamma = 0.35 rads.
This changes the "raw transmission" and "coupling", but not the inferred visibility:
I also measured the cavity AMTF at three powers today: 0.5 mW, 10 mW, and 45 mW input.
They look about the same. If anything, the cavity pole seems slightly lower with the higher power, which is counterintuitive. The expected shift is very small (~10%), since the decay rate is still totally dominated by the mirror transmissions even for the supposed high-loss state (Sam and Sheon estimated the roundtrip loss at high power to be ~1400 ppm, while the combined coupling mirrors' T is 1.6%). I have not been able to fit the cavity poles consistently to within this kind of error.
Tonight, we locked the "fauxMC". We obtained a visibility of >99%.
Koji had aligned it roughly last night, but we wanted to have a couple steering mirrors in the path for this practice cavity (the periscope mirrors will serve this function in the real setup), so we marked the alignment with irises and installed two extra mirrors.
After obtaining flashes with the WinCam placed at the output coupler, we removed the WinCam and put a CCD camera at one of the curved mirror transmissions and used this to get a strong TEM00 flash. Then, we installed the REFL PD/CCD, swept the laser PZT and optimized the alignment by minimizing the REFL dips. Finally, we connected the RF electronics and locked the cavity with the LB box. We used whatever cables we had around to trim the RF phase, and then Koji made some nice SMA cables at the 40m.
One thing we noticed was that we don't have enough actuation range to keep the cavity locked for very long---even with the HV amp (100V). We are going to offload to the NPRO temperature using an SR560 or pomona box circuit. We may also make an enclosure for the cavity to protect it from the HEPA blasting.
Tomorrow, after we do the above things, we will practice measuring the transmission, length (FSR) and mode spectrum of the cavity before moving on to the real McCoy.
That is an expanding fire pillow, also known as firebrick. It is used to create a fire block where holes in fire-rated walls are made and prevents lab fires from spreading rapidly to adjacent labs. I had to pull cable from B254 to our labs on either side during a rather narrow window of time. Some of the cable holes are partially blocked, making it difficult to reach the cable to them. The cable is then just guided to the hole from a distance. With no help, it's not possible to see this material getting shoved out of the hole. I can assure you that I took great pains not to allow the CYMAC coax to fall into any equipment, or drag against any other cables.
Friday, Nov 11th, 2022
Setting up OMC #1 for transmission measurements:
The laser beam was aligned to the OMC cavity. The OMC cavity was locked and the transmission measurements were recorded.
The measured total optical loss of the OMC was
1st: 0.015 +/- 0.003
2nd: 0.085 +/- 0.005
3rd: 0.0585+/- 0.0008
4th: 0.047 +/- 0.002
In avegrage the estimated loss is
Loss = 0.055 +/- 0.014
This is unchanged from the measurement at LLO after the FC cleaning
Loss = 0.053 +/- 0.010