I found that, in fact, I had lowered the modulation depth since when I measured it to be 0.45 rads --> Psb = 0.1.
Here is the sweep measurement:
This is Psb = 0.06 --> gamma = 0.35 rads.
This changes the "raw transmission" and "coupling", but not the inferred visibility:
I also measured the cavity AMTF at three powers today: 0.5 mW, 10 mW, and 45 mW input.
They look about the same. If anything, the cavity pole seems slightly lower with the higher power, which is counterintuitive. The expected shift is very small (~10%), since the decay rate is still totally dominated by the mirror transmissions even for the supposed high-loss state (Sam and Sheon estimated the roundtrip loss at high power to be ~1400 ppm, while the combined coupling mirrors' T is 1.6%). I have not been able to fit the cavity poles consistently to within this kind of error.