ach[Joe, Phillip, Koji, Stephen]
Work from 17.04.2019
First contact cleaning of OMC optics
We cleaned the OMC optic with first contact. After a first cleaning run all mirrors except for two looked
fine. One had some first contact residuals on the left at center height and another had some particle sitting
near the center area. As the ionized nitrogen gun didn't help we applied another round of first contact which resolved
the two issues. Unfortutanely the second run of cleaning again left some residuals of first contact at the edges.
We were able to peal these off with tweezers.
Placement of Optics at the breadboard
We cleaned the contact surfaces for the bonds using optic wipes and pure isopropanol. The placement wen't well for 3 of the 5 optics (low number of newtonian rings).
One was recleaned and placed on the breadboard again which seemed fine. For the 5th no newtonian rings could be seen (either verry ood or bad) we planed on trying it in the current set-up. Mirrors used can be seen in attachment 3.