Calibration of the transimpedance
Use KEITHLEY 2450 as a calibrated current source. Model 2450 has the current source accuracy of 0.020%+1.5uA at 10mA range. For 6mA current output, the error is 3uA (0.05%).
The output of the current amp at 103 Ohm setting was 6.0023V when -6.000mA current was applied. i.e R_trans = 1000.4 +/- 0.5 Ohm. This is a negligible level.
QE of the diodes (As of 07/30/2016)