All parameters are temporary:
Test mass size: D150mm x L140mm
Intermediate mass size W152.4mm x D152.4mm x H101.6mm
TM Magnets: 70mm from the center
Height from the bottom of the base plate
- Test mass: 5.0" (127mm) ==> 0.5" margin for the thermal insulation etc (for optical height of 5.5")
- Suspension Top: 488.95mm
- Top suspension block bottom: 17.75" (450.85mm)
- Intermediate Mass: 287.0mm (Upper pendulum length 163.85mm / Lower pendulum length 160mm)
- IM OSEMs: Top x2 (V/P)<-This is a mistake (Nov 3 fixed), Face x3 (L/Y/P), Side x 1 (S)
- TM OSEMs: Face x4
- OSEM insertion can be adjusted with 4-40 screws
- EQ Stops / Cradle (Nov 3 50% done)
- Space Consideration: Is it too tight?
- Top Clamp: We are supposed to have just two wires (Nov 3 50% done)
- Lower / Middle / Upper Clamps & Consider installation procedure
- Fine alignment adjustment
- Pendulum resonant frequencies & tuning of the parameters
- Utility holes: other sensors / RTDs / Cabling / etc
- Top clamp options: rigid mount vs blade springs
- Top plate utility holes
- IM EQ stops
Discussion with Rana
- Hoe do we decide the clear aperture size for the TM faces?
- OSEM cable stays
- Thread holes for baffles
- Light Machinery can do Si machining
- Thermal conductivity/expansion
- The bottom base should be SUS... maybe others Al except for the clamps
- Suspension eigenmodes separation and temperature dependence
# Deleted the images because they are obsolete.
Some more progress:
- Shaved the height of the top clamp blocks. We can extend the suspension height a bit more, but this has not been done.
- The IM OSEM arrangement was fixed.
- Some EQ stops were implemented. Not complete yet.
Does this work? Is this insane?
Today we looked at possible locations for where we will be setting up Mariner Suspension and Cryo chamber. The first option was the far left table in the CAML lab but it seems that there is going to be an issue with height clearance, so we have come up with another solution which takes a table from Koji's lab which is 3'x4' ft and moving it into CAML lab in the back right of the lab. To move the table we may need to call facilities to help us because we will most likely need to take the table apart to get it out of the lab. The aisle space in Koji's lab is about 43 inches, but the doorway, which is the tightest space, is 35 inches.
After we have set up the table in CAML we are planning on moving the Chamber in DOPO-lab to CAML. We plan to use skyhook with has a load limit of 500lbs/227kg this should be more than enough to move the chamber. We still need to get the wheeled base for skyhook we are in the works in doing so.
Also, We want to remove the previous setup from the chamber and leave it at DOPO-lab. Stephen is going to figure out how to keep it clean (sort of). Besides these transportation logistics, I am also working on the electronics as an immediate task and the electrical arrangement in the chamber.
to do list
- Check the table height
- Check the chamber height (base/cap)
- Check how much the chamber cap needs to be lifted (so that we can remove it)
- Is the weight capacity sufficient?
- B246/QIL Skyhook
- OMC Lab
Table moving effort in the OMC lab: See https://nodus.ligo.caltech.edu:8081/OMC_Lab/412