The inventory is in Clickup, which is a new organizational tool I'm trying out. There' an easy csv export, so I can get it elsewhere if/when we want. We have a wide variety of lenses:
I found fewer curved mirrors, but there were a couple.
Steps I took for the temperature sensor:
1. Tried to see what's the temperature by reading the current temperature in ndscope or dataviewer. In dataviewer, go to 'Signal' tab, and enter the channel name or find it on the list of slow channels. For ndscope...
No data appears on either. I restart cymac1, which seemed frozen, but still nothing.
2. What's going on with these channels?
Looks like the channel is reading zero.
3. I traced the cable from the particle counter and found that it sends data to cominaux, the common auxiliary machine for the lab.
This is the database file that defines the channels on cominaux. I search for 'LAB_TEMP_F' and find the epics record for the temperature channels. The epics records are all "calc" records, and the temperature in Kelvin is taken from X1:AUX-ACROXT_AI_15. This corresponds to channel 15 of the acromag slow ADC.
That's starting to make sense, the cable from the particle counter didn't go to the acromag ADC. Starting from the ADC channel 15, I traced the cable back to what used to be the AD590 temperature transducer.
4. Where did the IC temperature sensor go? Searching the elog and my dusty memory... neither readily recalls where it went. Let's get another one, they are cheap and easy to use.