Following my preliminary conclusion from yesterday (CRYO:1201), I set out to confirm or deny this seeming decrease in Q for a given clamp when going from the simple vacuum chamber to the cryostat.
One potential source of extra damping I considered was the wires attached to the block for the power resistor and RTD, so, while I still had the clamp in the cryostat assembly, I just disconnected these wires and pumped down the cryostat to see if I saw an improvement. I did see an increase in Q from ~3000 to ~5500, but not to the full 7000 I saw before in the standalone chamber. So, I conclude that there is some appreciable damping added by this kapton wiring. We need to use less rigid wire for the last stretch between the coldplate-mounted strain releif and the block.
The last step was to transport the clamp back into the simple chamber and see if I could recover the Q of 7000 that I measured initially. I did, completing the circle of repeatablility. I'm not sure what else could be causing the excess damping in the cryostat.
It is a shame, because I would be very interested to see what this particular silicon sandwich clamp looks like at 120 K, but I seem to have now way of doing so without the extra losses empirically associated with putting it in the cryostat.