Noise floor from PD and SR560 are measured, then converted from V/rHz to frequncy/rHz by the calibration from IFR2023 (0.7 MHz/Volt).
SR560 has the same setting (gain invert 100, low pass filter at 1 Hz) as it did yesterday (when I measured the power spectrum
of the feed back signal.) The power spectrum of the feedback signal (from yesterday) is plotted in gray, noise floor from PD( mixed with 160MHz signal from IFR2023) is plotted in pink, and noise from SR560 is in blue. I'll get the RC noise data on 2010Feb09, so I can plot them on the same graph.