to have real data for comparison i measured the noise of the SR560 for different (high) gain settings.
Will add lower gain settings and line powered measurements later.
test setup: SR560, AC-coupled, low-noise setting, battery powered measured with SR785
datafiles contain raw output noise values, so for input referred noise plz divide by gain given in filename. first col = frequency, second col = noise spectral density