C3:PSL-NPRO_PWRMON channel, and npro.db file are added
for monitoring power output of the NPRO (reflected beam from Faraday Isolator)
I haven't reset the crate yet, so the channel may not appear in DAQ yet.
It's connected to VMIVME-3113 at #C0 S60.
The photo diode is Thorlab PDA55 with RG1000 filter.
Calibration for power is 1.81 mW/V, now it reads 5.56V.