When both cavities are locked, I measured
1) power fluctuation behind ACAV (DC@0.55V) and RCAV (DC@2.2V)
2) power fluctuation behind PMC (DC@3.17V)and RCAV (DC@2.2V)
the HEPA filter was on, the SLOWDC loop was engaged.
SR785 is set to AC couple mode, auto range on both channels.
The level of RIN from RCAV from two measurements are a bit different,see fig2.
RIN from ACAV is not significantly higher than that of RCAV. This is good news, I thought AOM might pose more pointing instability to the system.