Today, I added a new out-of-loop transmission PD (Thorlabs PDA10CS) for the south path. This will be helpful in future measurements of RIN coupling to beatnote noise. This PD is added at (1, 40) using the dumped light. The optical layout would be updated in a few days. I've confirmed that this photodiode is reading the same RIN as read earlier in CTN:2555. I've also connected Acromag channel for South Transmission DC to this photodiode, so the transmitted power channels and the mode matching percentage channel of South Cavity are meaningful again.