Attached is captured beatnote frequency between Aug 27th 2019 to Today, Sep 5th 2019.
I have attached a few more zoomed-in plots as well. This code will serve us in the future as well. Only the green regions are where both cavities were locked and cavity heater PID was engaged. This data involves the experiment time of CTN:2406 and hence have the disturbances of that time as well.
Essentially, there is no fixed stability number one can really quote for this PID. It looks stable in regions, but at times it shoots up or down randomly. Maybe some of them are because of me doing something on the table, but some are late at night which can only be explained by the movement of ghosts.
fromFBread.py now has an optional flag -d for decimating the read data, so that smaller files are created. Example: -d 160 will decimate the data by a factor of 160 making a sampling rate of 0.1 Hz. It calculates mean of the data, for a block size of 160 (corresponding to 10s) and also calculates standard deviation in this block and adds that as additional columns in the read data. Hence the plots attached here have uncertainties as well.
Code and Data