Ever since I attached the vaccan air springs to the cylinder, it has been rapidly losing air pressure. It is now down to ~1100 psi, where on Thursday it was ~1800 psi. At this rate we will have to order a new cylinder, and figure out how to make the air springs less leaky, as this is affecting our alignment over time.
For now, I'll relock the cavities, replace the shims, and turn off the air springs as they are causing more harm than good. We will rely on the floating table until then.
I added a pressured pipe T to the nitrogen cylinder in our lab so we could float the table and vaccan air springs from the same cylinder, as opposed to before when the air springs were floated from the wall. This way we know that the air pressure is not changing every fifteen minutes.
The cylinder pressure is ~1800 psi right now, and I set the pressure regulator to 40 psi again (I took before and after pictures, pics 1 and 3). I also took a picture of the wall gauge before I disconnected the air springs from it, it was set to ~40 psi as well.
The final plot is the same nine channels I looked at this morning, 25 minutes of data. It seems our guess was right, because there are no longer ~15 minute TRANS and REFL waves. Cool.
Our initial impression of cavity power fluctuations was that temperature fluctuations in the EAOMs were the cause. To check this, I made some REFL DC monitors yesterday.
Plotted is one hour of data from today. Some notes:
1) TRANS DC and REFL DC for both cavites are breathing together every fifteen minutes, and are anticorrelated (one goes up, the other goes down).
2) The temperature monitors are not fluctuating with the same regularity as the power monitors.
3) The REFL DC for the north PMC is fluctuating with temperature
This leads me to believe our power fluctuations are caused by changes of alignment into the cavities from the air springs holding up our vacuum can.
Right now, the air springs are hooked up to the wall. There is probably some pressure regulator which switches on and off every fifteen minutes. To fix this, I'm going to switch the vacuum air springs over to our nitrogen cylinder we have in the lab and see if the fluctuations go away.