RefCav is locked , the beam is more stable than yesterday setup. I'll write down the values of the setting for a quick reference.
I'm not sure what universal names for all these channels are. I just explain them in more details for my future reference and next generation archeologists.
Common Gain( for both fast and PC paths): 23.6 dB (fast path controls the PZT which changes the length of the NPRO, PC path controls the phase
shift of the beam)
Fast Gain ( for PC path only): 12.5 dB
Phase shift: 0 + 180 degree. ("+180 degree" means phase flip)
RF Amplifier Adj (power for 35.5 MHz sidebands): 6.28 V.
Thermal control ADj (a voltage calibrator connected to slow channel of the laser controller): -0.010V
Servo gain Adj (over all gain of the demodulated signal): 27.75 dB
Output DC offset (offset voltage that governs the length of the PMC): -3.12 V
Phase shift: 2.87 V + 180 degree
RF Amp Adj (power for 21.5 MHz sidebands): 5.36V.
Now I'm working on ACav path. I made a cable for a photo diode.
I'm not sure if the last PD is a working 35.5 MHz PD, I'll see if it works or not.
Now I'm using two 2-channel monitors to simultaneously see the beams after PMC and RefCav. It will be more convenient if I use a 4 channel monitor, I'll clear some space for it.