This is the schematic for PSL setup.
At this point, Pre Mode Cleaner (PMC) and Reference Cavity (RefCav) are locked. The rest will be locking Analyzer Cavity (ACav) and setting up for beat noise measurement.
ACav's beam path will have double pass AOM [Crystaltech 3080 194]. We'll use +1st order beam. When hook up the VCO, make sure that the power is on only when the VCO and the AOM are connected, otherwise the VCO dies.
Next is aligning the AOM. A good alignment will maximize the power of the +1st order beam. The beam should get close to the AOM's transducer as much as possible to minimize time delay.
The beam at the AOM will be focused to 75 um.
The mirror that reflects the beam back to the AOM is a 0.3m concave mirror, which will be placed 0.3 m away from the AOM. The reflected beam should completely overlap on itself. This will neutralize
the pointing instability when the modulating frequency shifts.
Then we can align ACav, this time I'll try not to remove the PMC when I scan the beam frequency (at~3-10Hz.) If the PMC cannot catch up with the laser, increase the gain of the PMC, sideband power.
ACav should be locked before Monday June 8.