Today we improved alignment of the lens-camera arm. We discovered earlier that this alignment affects the amount of "snowfall" on the scattering images. Looking at the latest 405nm video (see attachment), one can still see snowfall, but it is considerably weaker now and the true scatter image is clearly visible. We took a set of scatter images at certain scattering angles and produced BSDF curves. The shape of these curves has partially to do with the snowfall contribution, but one also has to keep in mind that the mirror quality is much worse than what has been used in the Fullerton measurement. We still need to calibrate these curves. The calibration factor is different for the two images so that you cannot even compare them at the moment except for their shape.
Today we also got the new broadband lens for the camera arm. First measurements show that image quality is better. Playing a bit around with distances between object mirror, lens and image plane, we also found that image quality becomes better when the lens and camera get closer to the mirror (which is only an issue for the 405nm measurement since 633nm and 532nm look very good anyway). So we are thinking to change the camera arm setup to make it much shorter.