We can't generate any arbitary signal with the real time model, since awg is not working properly yet. For the moment being I added a uniform random number generator in the model (only option I found for noise) and send it into the ESD filter bank. In this way I can generate band-passed noise.
I plugged in the DAC output to the HV amplifier input, and I could send white noise to the electrostatic drive. Behold: I was able to excite quite a few modes. In the following trace blue is a reference and red is right after I sent white noise (3 V peak to peak) to the disk for a while (less than 1 minute). Excitation stopped at 4:56pm LT.
Using COMSOL and tuning the disk thickness at 1.018 mm I could hit the frequency of the first butterfly mode (1109 Hz) and get a reasobly good estimate of the other modes.
After about half a hour of ring down, most of the modes are gone but the two lowest are still going strong.
Note that the flattish background noise seems to be generated by some sort of glitches. I tried to swap the laser and the power supply, without change. More investigations are needed.
Note that the roughing pump was still on during the test.