Coating Ring-down Measurement Lab elog
Not logged in
Wed Aug 2 13:47:47 2017
Arm width Sweep
I ran a sweep of the width of the ESD arms. There appears to be a linear relationship across the modes except for mode 25. Mode 25 exhibits a very similar behavior as in the arm gap sweep. I realized that the abrupt change in direction (also noticeable in mode 14) is likely caused by the fact that the force profile is calculated as absolute value, there might be an exponential relationship that gets converted into that shape by the absolute value function.