Coating Ring-down Measurement Lab elog
Not logged in
Tue Jun 27 14:17:47 2017
Further plots and improving models
I built a new model of the ESD to determine whether or not the spikes in the electric field at the corners was affecting the results enough that it had to be accounted for in further models. To create the model, I created a 2D profile of the arm used in my original model and filleted the corners at a radius of .05 mm, since the electrode model is .1 mm thick, this made completely rounded edges. In creating this model I caught an earlier mistake in the original one, I only set one half of the surface of the electrodes to have a potential or to ground, the "bottom" was left with no charge. I fixed this mistake and then compared the two models at a potential of 1000 V. For speed of computation I ran both models with a finer mesh size and then calculated the electric field at approximately the middle of the ESD, 1mm above the fourth electrode arm. For the rounded electrodes the field had a value of 84024 V/m and for the rectangular electrodes the field had a value of 80728 V/m, which is less than a 4% difference in field magnitude. Furthermore, the field shapes appear nearly indistinguishable; I am confident from this test that I can proceed modelling the arms of the ESD as rectangles.