Initial pointing or IP-ANG is a pointing monitor of the MC. This beam is launched after the second pzt steering mirror.
IP-ANG is missing the pick up mirror by a few inches at ETMYchamber
1000 days plot show last appearance in Feb 2010
Two extender plates ready for cleaning. The existing optical table tops have 38" OD. Using two of these the OD will be 44"
Emergency exit lights were inspected: 2 out of 13 batteries have to be replaced
One of the Halon fire extinguishers needs to be recharged out of 8
Please do participate in preparation for the upcoming safety audit on Feb 28
Oplev transfer functions PIT UGF were optimized to be at 2-3 Hz with 60 degree minimum phase margin by adjuting oplev gains.
Additional Notes by KI:
Kiwamu (or whoever is here last tonight): please run the free-swing/kick script (/opt/rtcds/caltech/c1/scripts/SUS/freeswing) before you leave, and I'll check the matrices and update the suspensions tomorrow morning.
All suspentions were restored and MC locked. PRM side osem RMS motion was high.
Atm2, Why the PRM is 2x as noisy as the SRM ?
Batteries replaced and cylinder recharged. Please clean up your experimental set up if it is blocking breakers or entry way etc.
I will start the final clean up 2pm today.
The BS and the PRM have 3.3 Hz resonant gain filters that kill the phase margins.
Correction list by visiting safety committee, Haick Issaian is not shown:
1, update laser, crane operator list and post it
2, check fire extinguishers monthly, date and initials must be on the tags
3, move drinking water tower so it does not block fire extinguisher
4, post updated crane doc at cranes
5, post present phone lists at IFO room phones
6, emergency laser shutoff at the south end must be mounted without C-clamp
7, use heavy cable tie to insure position of mag-fan on cabinet top
Additional to do list:
a, safety glasses to be cleaned
b, let the electrical shop fix Rack-AC power to optical tables at the ends
c, measure transmission of laser safety glasses
d, update IFO outside door info signs
e, update laser inventory and post it
f, schedule annual crane inspection and renew maintenance contract
g, PSL enclosure inner shelf needs a good clean up so it is earthquake safe
Green welding glass 7" x 9" shade #14 with 40 mm hole and mounting fixtures are ready to reduce scatter light on SOS
PEEK 450CA shims and U-shaped clips will keep these plates damped.
All oplev qpd quadrons were zeroed by offset in blocked dark condition.
Type: How not to
Please do not leave optical tables open! You will be held responsible for creating dirty optics.
Optical spectrum analyzers like the Attachment made by Coherent , Meles Griot- CVI and Spectral Product are all discontinued.
The 40m have Coherent models C240 analyzer with controller C251 Their Finesse measured in 2004: sn205408 F302, sn205409 F396,
Jenne borrowed Jan's Meles Griot model 13SAE006, Peter King has the same model. FSR 300 MHZ, finnees 200 minimum
Over-sized local laser emergency switch was held by large C clamp at the south end. This was replaced by a smaller one and it is mounted with magnets.
The Innolight laser was turned off, while the interlock was wired.
Dichroic mirror quotes are in the wiki.
ATF is pricy.
We got a good price from Laseroptik, but the wedges are 5 arcminutes. The fused silica grade is 0F, meaning the homogeneity is 5 ppm instead of 1ppm. I requested an other large wedge quote on the substrates.We may have to get substrates from somebody else and ship it to Germany
MLT quote is outrageously high
REO is not interested in this low volume job.
Today is janitor day. It still does not explain why the 2W Innolight tripped off about an hour ago. All back to normal.
.......................................................I asked Keven later, he admitted hitting the emergency shut off next to the chemical storage cabinet.
Fred Goodbar of Konecrane has completed the annual certified crane inspection and maintenance of our cranes as required in safety document.
They are in good working condition and safe to use.
ETMY sus damping was disabled. Green locking laser and associated electronics turned off. Computers and power supplies turned off at rack 1Y4
The electricians picking up ac power from 1Y4 manual disconnect box and installing conduit line to ISCT-ETMY east end optical table.
There will be no more daisy chaining this way.
The power is back on at ETMY . c1iscey has not been restarted.
Now I'm turning ac power off at ETMX for the same job to be done.
The power was turned back on at 4pm It took some time for Suresh to restart the computers. We have damping but things are not perfect yet. Auto BURTH did not work well.
The Laseroptik quote is here.The 2 degrees wedge cost is $40 on each optics! See wiki
Bob cleaned all safety glasses in 10 % Liquinox soap in water solution first. The transmittance of glasses were checked at 100 mW 1064 nm S polarization, beam diameter 0.5 mm at 0-20 deg incident.
Coherent FieldMate power meter measured T= < 1 mW of all glasses.
Completed with the exception of d and g
Our existing 300 series SS plungers from McMastercar #8476A43 are silver plated as Atm2 shows.
Problems: 1, they become magnetized after years being close to the magnets
2, they oxidize by time so it is hard to turn them
I looked around to replace them.
Titanium body, nose and beryllium copper spring. None magnetic for UHV enviorment.
Can be made in 7 weeks at an UNREASONABLE $169.00 ea at quantity of 50
In order to get a better price from Vlier's Tom Chen I changed Ti body back to SS304L-siver plated and music wire spring. The price is still ~$120 ea. at quantity 50
I will talk to Mike G about modifying the McMaster plunger with a hex nut.
Conclusion: There is no need to silver plate the SS plunger at the torque level we hold the OSEMs
Mike Gerfen will be done with 50 pieces by April 4 and he will give them to Bob for cleaning. They should be cleaned and baked in a jig so the springs would be compressed for better venting.
Schott, green welding glass, shade 14, 3 mm thick was measured in the beam path of 1.2W, S polarization of 1064nm at ~1 mm diameter size as MC reflected path.
Absorption 95%, R 5% at incident angle 25-50 degrees. It looks like the perfect material for beam trap.
The CIT Chemistry Glass Shop cuts turned out to be sloppy using diamond disc blade cutter.
East coast Precision Glass & Optics offered scribed cut and polished side. Their quote price was high and time consuming.
The GLASS HOUSE shop in Pasadena 626 / 796-9151 on Walnut did a good and cheap job. Oscar the cutter will finish the rest of the cutting by next Friday, April 6
He used scribed cutting technic and his 1" x 0.5" pieces are good. Bob will pick up them up.
Local eq shakes the lab
We received 6 new chairs stools for the work benches that have 36" heights.
The control room east side and general office desks require 18" seat height. The new chairs stools have minimum seat heights 26"
Please label chairs that we are getting rid of.
I added an U channel based bottom shelf at the south end today.
I'm working on similar shelf at ETMY. Precondition: NPRO in bypass mode, heater for doubling in bypass........since power outage? Optical level servo turned off.......
U-channel based shelf in place. Oplev servo is back on at 11:15am The table may moved. The oplev return is missing the quad by a few milimeter.
ETMY sus damping restored.
ETMX sus damping restored
Guralp 1 is on the south side of IOO chamber
Aluminum posts and SS clamps for green glass traps in vacuum will be out of the shops by June 6, 2012 the latest
The typical sign of a dying gas laser is that it glows for a few minutes only. The power supplies are fine.
Two new JDS - Uniphase 1103P lasers ( NT64-104 ) arriving on Monday, May 21
ETMX oplev had 6 mm diameter beam on the qpd. I relayed the beam path with 2 lenses to get back 3 mm beam on the qpd
BRC 037 -100 Bi _concave lens and PCX 25 200 VIS do the job. Unfortunately the concave lens has the AR 1064.
The uncoated bi-concave lens was replaced by AR coated one: KBC 037 -100 AR.14 resulting 35% count increase on qpd
I locked the PMC and the MC followed instantly.
Yesterday I swapped in new He/Ne laser with output power 3.5 mW The return spot on qpd is large ~6mm in diameter and 20,500 counts
The spot size reduction require similar layout as ETMX oplev.
New self checking emergency back up lights were installed at 13 locations in the 40m lab.
The PMC behavior is not changed.
3 strip charts monitoring on 24 hours time scale: Vac, PSL-IFO, SUS and 10 channel video monitoring inside - outside of 40m lab
IOO Angle & IOO Position QPDs centered.
The oplev path is relayed and the spot size on the qpd is reduced. I still have to clean up and replace "Miki Mouse" lens holder.
There was no IP-ANG coming out of the chamber at this time!
...will be helpful for acquiring lock after the vent. We should install a camera at ETMX.
Jenne and Steve
Sony CCD in place needs alignment and ND filter
Something bad happened to c1sus and c1iscex ~20 min ago. They both have "0x2bad" 's. I restarted the daqd on the framebuilder, and then rebooted c1sus, and nothing changed. The SUS screens are all zeros (the gains seem to be set correctly, but all of the signals are 0's).
If it's not fixed when I get in tomorrow, I'll keep poking at it to make it better.
PMC trend of 400 and 1200 days
The Innolight 2W based PSL- IOO was implemented in the ~ summer of 2010
Mike Pedraza came by today to install a new wireless network router configured for the 40m lab network. It has a 'secret' SSID i.e. not meant for public use outside the lab. You can look up the password and network name on the rack. Pictures below show the location of the labels.
Mike P swapped in a new network router Linksys E1000
For some reason the state of the oplevs is completely different for almost every suspension. They have different sets of filters in the bank, and different filters engaged. wtf? How did this happen? Is this correct? Do we expect that the state of the oplevs should be different on all the different suspensions? I wouldn't have thought so.
I discovered this because the PRM is unstable with the oplevs engaged. I don't think it was yesterday. Is something hidden changing the oplev settings?
As of February 23, 2012 when oplev PIT and YAW transfer functions were taken
What not to do:
The PRM oplev servo was left on and it was driving this oscillation overnight.
The ref cavity ion pump was running at 7.7kV instead of 5kV
This Digitel SPC-1 20 l/s ion pump should be running at 5kV
I noticed that the ion pump was turned off.
It was turned ON. It showed 0.00 microA at 5kV The current display is not sensitive enough. There must be some small outgassing or leak. It adds up if we stop pumping.
We want to keep the reference cavity in pristine condition. It required the ion pump running all times.
Air conditioning maintenance is scheduled for tomorrow from 8 to 11am