40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 40m Log, Page 288 of 336 Not logged in
ID Date Author Type Category Subject
13159   Wed Aug 2 14:47:20 2017 KojiSummaryComputersChiara backup situation summary

I further made the burt snapshot directories compressed along with ELOG 11640. This freed up additional ~130GB. This will eventually help to give more space to the local backup (/dev/sdc1)

controls@chiara|~> df -m Filesystem     1M-blocks    Used Available Use% Mounted on /dev/sda1         450420   11039    416501   3% / udev               15543       1     15543   1% /dev tmpfs               3111       1      3110   1% /run none                   5       0         5   0% /run/lock none               15554       1     15554   1% /run/shm /dev/sdb1        2064245 1581871    377517  81% /home/cds /dev/sdd1        1877792 1426378    356028  81% /media/fb9bba0d-7024-41a6-9d29-b14e631a2628 /dev/sdc1        1877764 1698489     83891  96% /media/40mBackup

13183   Thu Aug 10 14:13:23 2017 KiraSummaryPEMtemperature sensor

Goal is to build a temperature sensor accurate to 1 mK. Schematic is shown below. This does not take into account the DC gain that occurs.

Parts that would be used for this: LM317 regulator, AD592 temperature transducer, OP amp (low input noise and high impedance), 100K (or maybe 10k) resistor. This is what is currently proposed, but the exact parts we use could be changed to better fit the sensor. The resistor and the OP amp will be decided depending on the output of the AD592.

Once this is built, I would like to create a few copies of it and put them into an insulated container and measure the output from each one. This would allow us to calculate the temperature noise of the circuit, as we can take out the variations due to temperature changes inside the container by comparing the outputs.

I can also model the noise in the circuit to see how much noise there is before building it. There are three terms to the noise that we have, and we need to decide which one dominates at low frequencies.

Our final goal is to create an additional circuit that could cancel out the DC gain. I have attached an additional schematic proposed by Rana that would help with this issue. I will leave this second half for when the first part works.

13188   Thu Aug 10 21:22:06 2017 ranaSummaryPEMtemperature sensor
• Should we use the AD590 or the AD592? They're sort of the same, but have slightly different packages and specs.
• Given the sensitivity of the AD590 to power supply drift, I would recommend using a precision voltage reference like the AD581 or AD587. Take a look at the datasheets and order a few different varieties so we can see what works best for us. I believe that the voltage regulators have too much drift to use for precision temperature control.
• The AD590 datasheet has a few example circuits showing how we can subtract off the offset which comes the 1 uA/K coefficient of the AD590 (i.e. we would have 295 uA at room temperature, trying to stabilize to +/- 0.1 K)
• For the first prototypes its fine to use some solderable protoboard assembly, but we will eventually have to figure out how to package this thing and interface it with our Acromag slow controls system.

also, I've attached some temperature noise spectra from the LISA group at the AEI in Hannover. It will be interesting to see if we get the same results.

13209   Tue Aug 15 11:50:21 2017 KiraSummaryPEMtemp sensor packaging/mount

For the final packaging/mounting of the sensor to the seismometer, I have thought of two options.

1. Attach circuit to a PCB board and place it inside the can, while leaving the AD590 open to the air inside the can.

• This makes sure that the sensor gets a direct measurement of the temperature of the air in the can, as it is exposed to the air.
• But, it takes a limited area of measurement, so it could be the case that the area we place it in happens to be a hot or cold pocket, and the measurement would be inaccurate.
• This can be solved by placing multiple copies of the circuit in various places of the can and averaging the values.

2. Attach the AD590 to a copper plate with thermal paste and put it into a pomona box.

• This solves the problem of having a limited sample area the first option had, as the copper plate should have a uniform temp distribution, thus we are sampling the temp of that whole area.
• Need to make sure that the response time to the temperature variations of copper is less than the frequency that we are measuring.
• This can be calculated using equations for heat transfer (listed below).

If anyone has input on which method is preferred or any additional options that we may have, I would appreciate it.

Heat transfer:

q = k A dT / s

• k = thermal conductivity
• A = area
• s = thickness

For copper, k = 401 W/mK, x = 1.27 mm, A = 2.66x10^-3 m^2 (for the particular copper plate I measured), dT = 1K (assume). Thus the heat transfer will be 839 J/s.

I'm not completely sure what to do with this yet, but it could help us decide whether the copper plate option will be useful for us.

13235   Mon Aug 21 20:11:25 2017 johannesSummaryGeneralLoss measurements plan

There are three methods we (will soon) have available to evaluate the round-trip dissipative losses in the arms that do not suffer from the ITM loss dominance:

• DC reflection method:
• Compare reflected light levels from [ITM only] vs [arm cavity on resonance]
• Basler CCDs:
• Infer large (or small) angle scatter loss with calibrated CCDs
• Reflection ringdowns:
• Need AS port light injection, principle is similar to DC method but better (?)

### DCREFL

The DC method comparing reflectivities has been used in the past and is relatively easy to do. After the recent vacuum troubles the first step should be to re-perform these as CDS permits (needs some ASS functionality and of course the MC to behave). It wouldn't hurt to know the parameters this depends on, aka mode overlap and modulation depths with better certainty. Maybe the SURF scripts for mode-spectroscopy can be applied?

### CCDs

With the new CCD cameras calibrated, pre-vent we can determine the magnitude of the large-angle scatter loss (assuming isotropic scatter) of ETMX and possibly ETMY. Can we look past ETMX/ETMY from the viewports? Then we can probably also look at the small angle scatter of ITMX and ITMY. If not, once we open one of the chambers there's the option of installing mirrors as close as possible to the main beam path. The easiest is probably to look at ITMX, since there is plenty of space in the XEND chamber, and the camera is already installed.

### ASPORT

This requires a lot of up-front work. We decided to use the spare 200mW NPRO. It will be placed on the PSL table and injected into an optical fiber, which terminates on the AS table. The again free space beam there needs to be sort-of mode-matched into the SRC ("sort-of" because mode-spectroscopy). We want to be able to phaselock this secondary beam to the PSL with at least a couple kHz bandwidth and also completely extinguish the beam on time-scales of a few microseconds. We will likely need to purchase a few components that we can salvage from other labs, I'm still going through the inventory and will know more soon (more detailed post to follow). We need to settle for the polarization we want to send in from the back.

## Tentative Schedule (aggressive)

#### Week Aug 21 - Aug 27:

• Update mode-overlap estimates
• Obtain current DC refl estimates
• Spatial profile of auxiliary NPRO
• CCD software prep work

#### Week Aug 28 - Sep 3:

• Re-evaluate modulation indices if necessary
• Optical beat AS Port Auxiliary Laser (ASAL) - PSL
• PLL setup
• CCD large angle prep work

#### Week Sep 4 - Sep 10:

• PLL CDS integration
• Amplitude-modulation preparation
• CCD large angles

#### Week Sep 11 - Sep 17:

• Fiber-injection
• AS table preliminary mode-matching
• CCD small angle prep work

#### Week Sep 18 - Sep 24:

• ASAL amplitude switching
• CCD small angles

#### Week Sep 25 - Oct 1:

• AS port ringdowns

13236   Mon Aug 21 21:26:41 2017 gautamSummaryGeneralLoss measurements plan

In case you want to use it, I had profiled the Lightwave NPRO sometime back, and we were even using it as the AUX X laser for a short period of time.

As for using the AS laser for mode spectroscopy: don't we want to match the beam into the cavity as best as possible, and then use some technique to disturb the input mode (like the dental tooth scraper technique from Chris Mueller's thesis)?

Johannes and I did an arm scan of the X arm today (arm controlled with ALS, monitoring IR transmission) - only 2 IR FSRs were scanned, but there should be sufficient information in there to extract the modulation depth and mode matching - can we use Kaustubh's/Naomi's code?. The Y arm ALS needs to be touched up so I don't have a Y arm scan yet. Note that to get a good arm scan measurement, the High Gain Thorlabs PD should be used as the transmission PD.

Quote:

#### Week Aug 21 - Aug 27:

• Update mode-overlap estimates
• Obtain current DC refl estimates
• Spatial profile of auxiliary NPRO
• CCD software prep work

13241   Tue Aug 22 16:56:54 2017 johannesSummaryGeneralAS laser existing components inventory

I surveyed the lab today to see what we may need to buy for the AS laser setup.

## We have:

NPRO 200 mW + Driver

ISOMET Model 1201E: This is a free space AOM I found in the modulator cabinet. It needs to be driven at 40MHz (to be confirmed) with ~6W of electrical power. For a 500 micron beam it can allegedly achieve rise times of '93' [units not specified, could this be nanoseconds?]. I did not find a dedicated driver for it, however there was a 5W minicircuits amplifier ZHL-5W-1 in the RF cabinet and a switch ZSDR-230, which has a typical switch time of 2 microseconds, however I'm not sure how this translates to rise/fall times of the deflected power. It seems we have everything to set this up, so we'll by the end of the week if we can use a combination of these things or if we need to buy additional driver electronics.

New Focus model 4004 broadband phase modulator which is labeled as dusty, and in fact quite dirty when looking through. We should attempt to clean this thing and maybe we can use it here or at the ends.

Probably all the optics we need for the PSL table setup.

## We need:

Beat PD: How about one of these: EOT ET-3000A? I didn't find a broadband PD for the beat with the PSL

Fiber Stuff: coupler & polarization maintaining fiber 20m & collimator. There are a couple options here, which we can discuss in the meeting.

Faraday Isolator: If we want to inject P-polarization. If S is okay we can use a polarizing plate beamsplitter instead.

Possibly some large lenses for mode-matching to IFO (TBD)

13250   Thu Aug 24 18:02:16 2017 GabrieleSummaryLSCFirst cavity length reconstruction with a neural network

## 1) Introduction

In brief, I trained a deep neural network (DNN) to recosntuct the cavity length, using as input only the transmitted power and the reflection PDH signals. The training was performed with simulated data, computed along 0.25s long trajectories sampled at 8kHz, with random ending point in the [-lambda/4, lambda/4] unique region and with random velocity.

The goal of thsi work is to validate the whole approach of length reconstruction witn DNN in the Fabry-Perot case, by comparing the DNN reconstruction with the ALS caivity lenght measurement. The final target is to deploy a system to lock PRMI and DRMI. Actually, the Fabry-Perot cavity problem is harder for a DNN: the cavity linewidth is quite narrow, forcing me to use very high sampling frequency (8kHz) to be able to capture a few samples at each resonance crossing. I'm using a recurrent neural network (RNN), in the input layers of the DNN, and this is traine using truncated backpropagation in time (TBPT): during training each layer of RNN is unrolled into as many copies as there are input time samples (8192 * 0.25 = 2048). So in practice I'm training a DNN with >2000 layers! The limit here is computational, mostly the GPU memory. That's why I'm not able to use longer data stretches.

But in brief, the DNN reconstruction is performing well for the first attempt.

## 2) Training simulation

In the results shown below, I'm using a pre-trained network with parameters that do not match very well the actual data, in particular for the distribution of mirror velocity and the sensing noises. I'm working on improving the training.

I used the following parameters for the Fabry-Perot cavity:

The uncertaint is assumed to be the 90% confidence level of a gaussian distribution. The DNN is trained on 100000 examples, each one a 0.25/8kHz long trajectory with random velocity between 0.1 and 5 um/s, and ending point distributed as follow: 33% uniform on the [-lambda/4, lambda/4] region, plus 33% gaussian distribution peaked at the center with 5 nm width. In addition there are 33% more static examples, distributed near the center.

For each point along the trajectory, the signals TRA, POX11_I and POX11_Q are computed and used as input to the DNN.

## 3) Experimental data

Gautam collected about 10 minutes of data with the free swinging cavity, with ALS locked on the arm. Some more data were collected with the cavity driven, to increase the motion. I used the driven dataset in the analysis below.

### 3.1) ALS calibration

The ALS signal is calibrated in green Hz. After converting it to meters, I checked the calibration by measuring the distance between carrier peaks. It turned out that the ALS signal is undercalibrated by about 26%. After correcting for this, I found that there is a small non-linearity in the ALS response over multiple FSR. So I binned the ALS signal over the entire range and averaged the TRA power in each bin, to get the transmission signals as a function of ALS (in nm) below:

I used a peak detection algorithm to extract the carrier and 11 MHz sideband peaks, and compared them with the nominal positions. The difference between the expected and measured peak positions as a function of the ALS signal is shown below, with a quadratic fit that I used to improve the ALS calibration

The result is

z_initial = 1e9 * L*lamba/c *1.26. * ALS
z_corrected = 2.1e-06 z^2  -1.9e-02 z  -6.91e+02

The ALS calibrated z error from the peak position is of the order of 3 nm (one sigma)

## 3.2) Mirror velocity

Using the calibrated ALS signal, I computed the cavity length velocity. The histogram below shows that this is well described by a gaussian with width of about 3 um/s. In my DNN training I used a different velocity distribution, but this shouldn't have a big impact. I'm retraining with a different distirbution.

## 4) DNN results

The plot below shows a stretch of time domain DNN reconstruction, compared with the ALS calibrated signal. The DNN output is limited in the [-lambda/4, lambda/4] region, so the ALS signal is also wrapped in the same region. In general the DNN reconstruction follows reasonably well the real motion, mostly failing when the velocity is small and the cavity is simultanously out of resonance. This is a limitation that i see also in simulation, and it is due to the short training time of 0.25s.

I did not hand-pick a good period, this is representative of the average performance. To get a better understanding of the performance, here's a histogram of the error for 100 seconds of data:

The central peak was fitted with a gaussian, just to give a rough idea of its width, although the tails are much wider. A more interesting plot is the hisrogram below of the reconstructed position as a function of the ALS position, Ideally one would expect a perfect diagonal. The result isn't too far from the expectation:

The largest off diagonal peak is at (-27, 125) and marked with the red cross. Its origin is more clear in the plot below, which shows the mean, RMS and maximum error as a function of the cavity length. The second peak corresponds to where the 55 MHz sideband resonate. In my training model, there were no 55 MHz sidebands nor higher order modes.

## 5) Conclusions and next steps

The DNN reconstruction performance is already quite good, considering that the DNN couldn't be trained optimally because of computation power limitations. This is a validation of the whole idea of training the DNN offline on a simulation and then deploy the system online.

I'm working to improve the results by

• training on a more realistic distribution of velocity
• adding the 55 MHz sidebands
• tune the DNN architecture

However I won't spend too much time on this, since I think the idea has been already validated.

13251   Thu Aug 24 18:51:57 2017 KojiSummaryLSCFirst cavity length reconstruction with a neural network

Phenomenal!

13256   Sat Aug 26 09:56:34 2017 GabrieleSummaryLSCFirst cavity length reconstruction with a neural network

## Update

I included the 55 MHz sideband and higher order modes in my training examples. To keep things simple, I just assumed there are higher order modes up to n+m=4 in the input beam. The power in each HOM is randomly chosen from a random gaussian distribution with width determined from experimental cavity scans. I used a value of 0.913+-0.01 rad for the Gouy phase (again estimated from cavity scans, but in reasonable agreement with the nominal radius of curvature of ETMX)

Results are improved. The plot belows show the performance of the neural network on 100s of experimental data

For reference, the plots below show the performance of the same network on simulated data (that includes sensing noise but no higher order modes)

13258   Mon Aug 28 08:47:32 2017 JamieSummaryLSCFirst cavity length reconstruction with a neural network
 Quote: Phenomenal!

truly.

13269   Tue Aug 29 15:41:17 2017 KiraSummaryPEMheater circuit

I worked with Kevin and Gautam to create a heater circuit. The first attachment is Kevin's schematic of the circuit. The OP amp connects to the gate of the power MOSFET, and the power supply connects to the drain, while the source goes into the heater. We set the power supply voltage to 22V and varied the voltage of the input to the OP amp. At 6V to the OP amp, we got a current of 0.35A flowing through the heater and resistor. This was the peak current we got due to the OP amp being saturated (an increase in either of the power supplies did not change the current), but when we increased the voltage of the supply rails of the OP amp from 15V to 20V, we got a current of 0.5A. We would want a higher current than this, so we will need to get a different OP amp with a higher max voltage rating, and a resistor that can take more power than this one (it currently takes 5W of power, and is the best one we could find).

Kevin and I created a simulation of this circuit using CircuitLab to understand why the current was so low (second attachment). The horizontal axis is the voltage we supply to the OP amp. The blue line shows the voltage at the point between the output of the OP amp and the gate of the MOSFET. The orange line is the voltage at the point between the source of the MOSFET and the heater. The brown line is the voltage at the point between the heater and resistor. Thus, we can see that saturation occurs at about 2.1V. At that point, the gate-source voltage is the difference between the blue curve and the orange curve, which is about 4V, which is what we measured. Likewise, the voltage across the heater is the difference between the orange curve and the brown curve, which comes out to around 8V, which is also what we measured. Lastly, the voltage across the resistor is the brown curve, which is about 2V, which matches our observations. The circuit works as it should, but saturates too soon to get a high enough current out of it.

Gautam noted that it is important to measure the current correctly. We can't just use an ammeter and place it across the resistor or heater, because the internal resistance of the ammeter (~0.5 ohm) is comparable to the resistance we want to measure, so the current gets split between the circuit and the ammeter and we get an equivalent resistance of 1/R = 1/R0 + 1/Ra, where R0 is the resistance of the part we want to measure the current across, and Ra is the ammeter resistance. Thus, the new resistance will be lower and the ammeter will show a higher current value than what is actually there. So to accurately measure the current, we must place the ammeter in series with the part we want to measure. We initially got a 1A reading on the heater, which was not correct, and our setup did not heat up at all basically. When we placed the ammeter in series with the heater, we got only 0.35A.

The last two images are the setup for testing of the heater. We wrapped it around an aluminum piece and covered it with a few layers of insulating material. We can stick a thermometer in between the insulation and heater to see the temperature change. In later tests, we may insulate the whole piece so that less heat gets dissipated. In addition, we used a heat sink and thermal paste to secure the MOSFET to it, as it got very hot.

Our next steps will be to get a resistor and an OP amp that are better suited for our purposes. We will also run simulations with components that we choose to make sure that it can provide the desired current of 1A (the maximum output of the power supply is 24V, and the heater is 24 ohm, so max current is 1A). Kevin is working on that now.

13272   Wed Aug 30 06:45:32 2017 KevinSummaryPEMNew Heater Circuit

I changed the heater circuit described in this elog to a current sink. The new and old circuits are shown in the attachment. The heater is $R_h$ and is currently 24Ω; the sense resistor $R$ is currently 6Ω. The op-amp is still an OP27 and the MOSFET is still an IRF630.

The current through the old circuit was saturating because the gate voltage on the MOSFET was saturating at the op-amp supply rails. This is because the source voltage is relatively high: $V_S = I(R + R_h)$.

In the new circuit the source voltage is lower and the op-amp can thus drive a large enough $V_{GS}$ to draw more current (until the power supply saturates at 25V/30Ω = 0.8A in this case). The source and DAC voltages are equal in this case$V_{\mathrm{DAC}} = V_S$ and so the current is $I = V_{\mathrm{DAC}}/R$. Since this is the same current through the heater, the drain voltage is $V_D = V_{cc} - IR_h$. I observed this behavior in this circuit until the power supply saturated at 0.8A. Note that when this happens $V_D = V_S$ and the gate voltage saturates at the supply rails in an attempt to supply the necessary current.

13274   Wed Aug 30 11:04:08 2017 GabrieleSummaryLSCFirst look at neural network reconstruction of PRMI motion

## Introduction

I trained a deep neural network (DNN) to reconstruct MICH and PRCL degrees of freedom in the PRMI configuration. For details on the DNN architecture please refer to G1701455 or G1701589. Or if you really want all the details you can look at the code. I used the following signals as input to the DNN: POPDC, POP22_Q, ASDC, REFL11_I/Q, REFL55_I/Q, AS55_I/Q.

Gautam took some PRMI data in free swinging and driven configuration:

• 1187819331 + 10mins: Free swinging PRMI (after first locking PRMI on carrier and dither aligning).
• 1187820070 + 5mins: PRM driven at low freq.
• 1187820446 + 5mins: BS driven at low freq.

In contrast to the Fabry-Perot cavity case, we don't have a direct measurement of the real PRCL/MICH degrees of freedom, so it's more difficult to assess if the DNN is working well.

## Results

All MICH and PRCL values are wrapped into the unique region [-lambda/4, lambda/4]^2. It's even a bit more complicated than simpling wrapping. Indeed, MICH is periodic over [-lambda/2, lambda/2]. However, the Michelson interferometer reflectivity (as seen from PRC) in the first half of the segment is  the same as in the second half, except for a change in sign. This change of sign in Michelson reflectivity can be compensated by moving PRCL by lambda/4, thus generating a pi phase shift in the PRC round trip propagation that compensate for the MICH sign change. Therefore, the unit cell of unique values for all signals can be taken as [-lambda/4, lambda/4] x [-lambda/4, lambda/4] for MICH x PRCL. But when we hit the border of the MICH region, PRCL is also affected by addtion of lambda/4. Graphically, the square regions A B C below are all equivalent, as well as more that are not highlighted:

This makes it a bit hard to un-wrap the resonstructed signal, especially when you add in the factor that in the reconstruction the wrapping is "soft".

The plot below shows an example of the time domain reconstruction of MICH/PRCL during the free swinging period.

It's hard to tell if the positions look reasonable, with all the wrapping going on.

## Two-dimensional maps of signals

Here's an attempt at validating the DNN reconstruction. Using the reconstructed MICH/PRCL signal, I can create a 2d map of the values of the optical signals. I binned the reconstructed MICH/PRCL in a 51x51 grid, and computed the mean value of all optical signals for each bin. The result is shown in the plot below, directly compared with the expectation from a simulation.

The power signals (POP_DC, AS_DC, PO22_Q) looks reasonably good. REFL11_I/Q also looks good (please note that due to an early mistake in my code, I reversed the convention for I/Q, so PRCL signal is maximized in Q instead than in I). The 55MHz signals look a bit less clear...

## Steps forward

• I'm quite confident in the tuning of demodulation phase and signs for REFL11 and POP22, but less so for REFL55 and not sure at all for AS55. So it would be useful to measure a full sensing matrix of PRCL and MICH against those signals, to compare with my simulation
• I'm working on an idea to fine tune the DNN using the real interferometer data, more to follow when the idea crystallizes in a clear form.
13279   Thu Aug 31 00:46:57 2017 ranaSummaryCDSallegra -> Scientific Linux 7.3

## I made a 'LiveCD' on a 16 GB USB stick using this command after the GUIs didn't work and looking at some blog posts:

sudo dd if=SL-7.3-x86_64-2017-01-20-LiveCD.iso of=/dev/sdf

Quote:

Debian doesn't like EPICS. Or our XY plots of beam spots...Sad!

Quote:
 Quote: No, not confused on that point. We just will not be testing OS versions at the 40m or running multiple OS's on our workstations. As I've said before, we will only move to so-called 'reference' systems once they've been in use for a long time.

Ubuntu16 is not to my knowledge used for any CDS system anywhere.  I'm not sure how you expect to have better support for that.  There are no pre-compiled packages of any kind available for Ubuntu16.  Good luck, you big smelly doofuses. Nyah, nyah, nyah.

K Thorne recommends that we use SL7.3 with the 'xfce' window manager instead of the Debian family of products, so we'll try it out on allegra and rossa to see how it works for us. Hopefully the LLO CDS team will be the tip of the spear on solving the usual software problems we have when we "~up" grade.

13292   Tue Sep 5 09:47:34 2017 KiraSummaryPEMheater circuit calculations

I decided to calculate the fluctuation in power that we will have in the heater circuit. The resistors we ordered have 50 ppm/C and it would be useful to know what kind of fluctuation we would expect. For this, I assumed that the heater itself is an ideal resistor that has no temperature variation. The circuit diagram is found in Kevin's elog here. At saturation, the total resistance (we will have a $1\Omega$ resistor instead of $6\Omega$ for our new design) will be $R_{tot}=R+R_{h}=1\Omega +24\Omega =25\Omega$. Therefore, with a 24V input, the saturation current should be $I=\frac{V_{in}}{R_{tot}}=\frac{24V}{25\Omega}=0.96A$.  Therefore, the power in the heater should be (in the ideal case) $P=I^2R{_{h}}=22.1184W$

Now, in the case where the resistor is not ideal, let's assume the temperature of the resistor changes by 10C (which is about how much we would like to heat the whole thing). Therefore, the resistor will have a new value of $R_{new}=R+50ppm/C\times 10C\times 10^{-6}=1.0005\Omega$. The new current will then be $I_{new}=\frac{V_{in}}{R_{new}}=0.95998A$ and the new power will be $P_{new}=I_{new}^{2}R_{h}=22.1175W$. So the difference in power going through the heater is about 0.00088W.

We can use this power difference to calculate how much the temperature of the metal can we wish to heat up will change. $\Delta T=\Delta P\times (1/\kappa) /x$ where $\kappa$ is the thermal conductivity and x is the thickness of the material. For our seismometer, I calculated it to be 0.012K.

13294   Tue Sep 5 16:37:47 2017 GabrieleSummaryLSCImproved PRMI deep learning reconstruction

This is an update on the results already presented earlier (refer to elog 13274 for more introductory details). I improved significantly the results with the following tricks:

• I retuned the demodulation phase of AS55, this time ensuring that the (alleged) MICH motion is visible mostly in Q when crossing a carrier resonance. Further fine tunings of phases will be possible once we have a measurement of the length optical matrix

• I fine tuned the netwrok by training it again using the real data. The ides is the following. I started with the network trained on the simulated data, and froze the parameters of the input recurrent layers. I fed the real signal to the network, computed the reconstructed PRCL/MICH, and fed them to my PRMI model to compute simulated signals. I allowed some of the parameters of the models to vary (expecially demodulation phases). I then trained again the network by trying to match the model predicted signals with the real input signals. I allowed only the parameters of the fully connected layers to vary (mostly for technical reasons, I'm working on re-training also the recurrent layers)

An example of time domain reconstruction is visible below. It already looks better than the old results:

As before, to better evaluate the performance I plotted averaged values of the real signals as a function of the reconstructed MICH and PRCL positions. The results are compared with simulation below. They match quite well (left real data, right simualtion expectation)

One thing to better understand is that MICH seems to be somewhat compressed: most of the output values are between -100 and +100 nm, instead of the expected -lambda/4, lambda/4. The reason is still unclear to me. It might be a bug that I haven't been able to track down yet.

13304   Fri Sep 8 12:08:32 2017 GabrieleSummaryLSCGood reconstruction of PRMI degrees of freedom with deep learning

## Introduction

This is an update of my previous reports on applications of deep learning to the reconstruction of PRMI degrees of freedom (MICH/PRCL) from real free swinging data. The results shown here are improved with respect to elog 13274 and 13294. The training is performed in two steps, the first one using simulated data, and the second one fine tuning the parameters on real data.

## First step: training with simulation

This step is exactly the same already described in the previous entries and in my talks at the CSWG and LVC. For details on the DNN architecture please refer to G1701455 or G1701589. Or if you really want all the details you can look at the code. I used the following signals as input to the DNN: POPDC, POP22_Q, ASDC, REFL11_I/Q, REFL55_I/Q, AS55_I/Q. The network is trained using linear trajectories in the PRCL/MICH space, and signals obtained from a model that simulates the PRMI behavior in the plane wave approximation. A total of 150000 trajectories are used. The model includes uncertainties in all the optical parameters of the 40m PRMI configuration, so that the optical signals for each trajectory are actually computed using random optical parameteres, drwn from gaussian distributions with proper mean and width. Also, white random gaussian sensing noise is added to all signals with levels comparable to the measured sensing noise.

The typical performance on real data of a network pre-trained in this way was already described in elog 13274, and although being reasoble, it was not too good.

## Second step: training with real data

Real free swinging data is used in this step. I fine tuned the demodulation phases of the real signals. Please note that due to an old mistake, my convention for phases is 90 degrees off, so for example REFL11 is tuned such that PRCL is maximized in Q instead of I. Regardless of this convention confusion, here's how I tuned the phases:

• REFL11: PRCL is all in Q when crossing the carrier resonance
• REFL55: PRCL is all in Q when crossing the carrier resonance
• AS55: MICH is all in Q when crossing the PRCL carrier resonance
• POP22: signal peaking in Q when crossing carrier or sideband resonances. Carrier resonance crossing gives positive sign

Then I built the following training architecture. The neural network takes the real signals and produces estimates of PRCL and MICH for each time sample. Those estimates are used as the input for the PRMI model, to produce the corresponding simulated optical signals. My cost function is the squared difference of the simulated versus real signals. The training data is generated from the real signals, by selection 100000 random 0.25s long chunks: the history of real signal over the whole 0.25s is used as input, and only the last sample is used for the cost function computation. The weights and biases of the neural network, as well as the model parameters are allowed to change during the learning process. The model parameters are regularized to suppress large deviations from the nominal values.

One side note here. At first sight it might seems weird that I'm actually fedding as input the last sample and at the same time using it as the reference for the loss function. However, you have to remember that there is no "direct" path from input to output: instead all goes through the estimated MICH/PRCL degrees of freedom, and the optical model. So this actually forces the network to tune the reconstruction to the model. This approach is very similar to the auto-encoder architectures used in unsupervised feature learning in image recognition.

## Results

After trainng the network with the two previous steps, I can produce time domain plots like the one below, which show MICH and PRCL signals behaving reasonably well:

To get a feeling of how good the reconstruction is, I produced the 2d maps shown below. I divided the MICH/PRCL plane in 51x51 bins, and averaged the real optical signals with binning determined by the reconstructed MICH and PRCL degrees of freedom. For comparison the expected simulation results are shown. I would say that reconstructed and simulated results match quite well. It looks like MICH reconstruction is still a bit "compressed", but this should not be a big issue, since it should still work for lock acquisition.

## Next steps

There a few things that can be done to futher tune the network. Those are mostly details, and I don't expect significant improvements. However, I think the results are good enough to move on to the next step, which is the on-line implementation of the neural network in the real time system.

13365   Fri Oct 6 12:56:40 2017 gautamSummaryLSCRTCDS NN

[gabriele, gautam]

Gabriele had prepared a C code implementation of his NN for MICH/PRCL state estimation. He had been trying to get it going on some of the machines in WB, but was unsuccessful. The version of RCG he was trying to compile and run the code on was rather dated so we decided to give it a whirl on our new RCG3.4 here at the 40m. Just noting down stuff we tried here:

• Code has been installed at /opt/rtcds/userapps/release/cds/c1/src/nn.This is to facilitate compilation by the RCG.
• There is also a simulink block diagram (x3tst.mdl) in there which we copied and pasted into c1pem. Changed the appropriate paths in the C-Code block to point to the location in the previous bullet point.
• c1pem was chosen for this test as it runs at 2k which is what the network is designed for.
• Since we were running the test on c1sus and expected the machine to crash, I shutdown all the watchdogs for the test.
• Code compiled without any problems (i.e. rtcds make c1pem and rtcds install c1pem executed successfully). There were some warning messages related to C-Code blocks but these are not new, they show up in all models in which we have custom C-code blocks.
• Unfortunately, the whole c1sus FE crashed when we tried rtcds restart c1pem.
• We tried a couple of more iterations, and attempted to monitor dmesg during the restart process to see if there were any clues as to why this wasn't working, but got nothing useful.

All models have been reverted to their state prior to this test, and everything on the CDS_OVERVIEW MEDM screen is green now.

13383   Tue Oct 17 17:53:25 2017 jamieSummaryLSCprep for tests of Gabriele's neural network cavity length reconstruction

I've been preparing for testing Gabriele's deep neural network MICH/PRCL reconstruction.  No changes to the front end have been made yet, this is all just prep/testing work.

## Background:

We have been unable to get Gabriele's nn.c code running in kernel space for reasons unknown (see tests described in previous post).  However, Rolf recently added functionality to the RCG that allows front end models to be run in user space, without needing to be loaded into the kernel.  Surprisingly, this seems to work very well, and is much more stable for the overall system (starting/stopping the user space models will not ever crash the front end machine).  The nn.c code has been running fine on a test machine in this configuration.  The RCG version that supports user space models is not that much newer than what the 40m is running now, so we should be able to run user space models on the existing system without upgrading anything at the 40m.  Again, I've tested this on a test machine and it seems to work fine.

The new RCG with user space support compiles and installs both kernel and user-space versions of the model.

## Work done:

• Create 'c1dnn' model for the nn.c code.  This will run on the c1lsc front end machine (on core 6 which is currently empty), and will communicate with the c1lsc model via SHMEM IPC.  It lives at:
• /opt/rtcds/userapps/release/isc/c1/models/c1dnn.mdl
• Got latest copy of nn.c code from Gabriele's git, and put it at:
• /opt/rtcds/userapps/release/isc/c1/src/nn/
• Checked out the latest version of the RCG (currently SVN trunk r4532):
• /opt/rtcds/rtscore/test/nn-test
• Set up the appropriate build area:
• /opt/rtcds/caltech/c1/rtbuild/test/nn-test
• Built the model in the new nn-test build directory ("make c1dnn")
• Installed the model from the nn-test build dir ("make install-c1dnn")

## Test:

I tried a manual test of the new user space model.  Since this is a user space process running it should have no affect on the rest of the front end system (which it didn't):

• Manually started the c1dnn EPICS IOC:
• $(cd /opt/rtcds/caltech/c1/target/c1dnn/c1dnnepics && ./startupC1) • Tried running the model user-space process directly: • $ taskset -c 6 /opt/rtcds/caltech/c1/target/c1dnn/bin/c1dnn -m  c1dnn

Unfortunately, the process died with an "ADC TIMEOUT" error.  I'm investigating why.

Once we confirm the model runs, we'll add the appropriate SHMEM IPC connections to connect it to the c1lsc model.

13390   Wed Oct 18 12:14:08 2017 jamieSummaryLSCprep for tests of Gabriele's neural network cavity length reconstruction
 Quote: I tried a manual test of the new user space model.  Since this is a user space process running it should have no affect on the rest of the front end system (which it didn't): Manually started the c1dnn EPICS IOC: $(cd /opt/rtcds/caltech/c1/target/c1dnn/c1dnnepics && ./startupC1) Tried running the model user-space process directly: $ taskset -c 6 /opt/rtcds/caltech/c1/target/c1dnn/bin/c1dnn -m c1dnn Unfortunately, the process died with an "ADC TIMEOUT" error.  I'm investigating why. Once we confirm the model runs, we'll add the appropriate SHMEM IPC connections to connect it to the c1lsc model.

I tried moving the model to c1ioo, where there are plenty of free cores sitting idle, and the model seems runs fine.  I think the problem was just CPU contention on the c1lsc machine, where there were only two free cores and the kernel was using both for all the rest of the normal user space processes.

So there are two options:

• Use cpuset on c1lsc to tell the kernel to remove all other processes from CPU6 and save it just for the c1dnn model.  This should not have any impact on the running of c1lsc, since that's exactly what would be happening if we were running the model in kernel space (e.g. isolating the core for the front end model).  The auxilliary support user space processes (epics seq/ioc, awgtpman) should all run fine on CPU0, since that's what usually happens.  Linux is only using the additional core since it's there.  We don't have much experience with cpuset yet, though, so more offline testing will be required first.
• Run the model on c1ioo and ship the needed signals to/from c1lsc via PCIe dolphin.  This is potentially slightly more invasive of a change, and would put more work on the dolphin network, but it should be able to handle it.

I'm going to start testing cpuset offline to figure out exactly what would need to be done.

13395   Thu Oct 19 15:42:03 2017 jamieSummaryLSCMICH/PRCL reconstruction neural network running on c1lsc

Gabriele's PRCL/MICH reconstruction neural network is now running on c1lsc.  Summary:

• front-end model is called c1dnn, and is running as an experimental user-space process
• c1dnn is getting most of it's needed inputs from existing SHMEM IPC outputs from c1lsc
• none of the output signals from the network are being sent anywhere yet (grounded)
• c1dnn has not been integrated in any way, into the DAQ etc.  it is being run manually by hand, and will be completely shut down after this test

Simple MEDM screen I made to monitor the input/output signals:

The RTS process seems to run fine, but there is quite a bit of jitter in the CPU_METER, at the 50% level:

It's not running over the limit, but it is jumping around more than I think it should be.  Will look into that...

## cpuset for cpu isolation for user-space model

The c1dnn model is running on CPU6 on c1lsc.  CPU6 was isolated from the rest of the system using cpuset.  The "cset" utility was used to create a "system" CPU set that was assigned to CPU0, and the kernel was instructed to move all running processes to that set:

controls@c1lsc:~ 2$sudo cset set cset: Name CPUs-X MEMs-X Tasks Subs Path ------------ ---------- - ------- - ----- ---- ---------- root 0,6 y 0 y 343 0 / controls@c1lsc:~ 0$ sudo cset set -c 0 -s system --cpu_exclusive
cset: --> created cpuset "system"
controls@c1lsc:~ 0$sudo cset set cset: Name CPUs-X MEMs-X Tasks Subs Path ------------ ---------- - ------- - ----- ---- ---------- root 0,6 y 0 y 342 1 / system 0 y 0 n 0 0 /system controls@c1lsc:~ 0$ sudo cset proc --move -f root -t system -k
cset: moving all tasks from root to /system
cset: moving 292 userspace tasks to /system
cset: moving 0 kernel threads to: /system
cset: --> not moving 50 threads (not unbound, use --force)
[==================================================]%
cset: done
controls@c1lsc:~ 0$sudo cset set cset: Name CPUs-X MEMs-X Tasks Subs Path ------------ ---------- - ------- - ----- ---- ---------- root 0,6 y 0 y 50 1 / system 0 y 0 n 292 0 /system controls@c1lsc:~ 0$ sudo cset proc --move -f root -t system -k --force
cset: moving all tasks from root to /system
cset: moving 50 kernel threads to: /system
[==================================================]%
cset: **> 29 tasks are not movable, impossible to move
cset: done
controls@c1lsc:~ 0$sudo cset set cset: Name CPUs-X MEMs-X Tasks Subs Path ------------ ---------- - ------- - ----- ---- ---------- root 0,6 y 0 y 29 1 / system 0 y 0 n 313 0 /system controls@c1lsc:~ 0$

I then created a set for the RTS process ("rts-c1dnn") on CPU6, and executed the c1dnn model in that set:

controls@c1lsc:~ 0$sudo cset set -c 6 -s rts-c1dnn --cpu_exclusive cset: --> created cpuset "rts-c1dnn" controls@c1lsc:~ 0$ sudo cset set
cset:
Name       CPUs-X    MEMs-X Tasks Subs Path
------------ ---------- - ------- - ----- ---- ----------
root        0,6 y       0 y    24    2 /
rts-c1dnn          6 y       0 n     0    0 /rts-c1dnn
system          0 y       0 n   340    0 /system
controls@c1lsc:~ 0$sudo cset proc -s rts-c1dnn --exec /opt/rtcds/caltech/c1/target/c1dnn/bin/c1dnn -- -m c1dnn cset: --> last message, executed args into cpuset "/rts-c1dnn", new pid is: 27572 sysname = c1dnn .... When done I just hit Ctrl-C. I left the cpusets as they are, with all system processes in the "system" set. This should not pose any problems since it's the identical configuration as would be if a normal kernel-level model was running in CPU6. The c1dnn process and it's EPICS sequencer were shutdown after this test. 13400 Tue Oct 24 20:14:21 2017 jamieSummaryLSCfurther testing of c1dnn integration; plugged in to DAQ In order to try to isolate CPU6 for the c1dnn neural network reconstruction model, I set the CPUAffinity in /etc/systemd/system.conf to "0" for the front end machines. This sets the cpu affinity for the init process, so that init and all child processes are run on CPU0. Unfortunately, this does not affect the kernel threads. So after reboot all user space processes where on CPU0, but the kernel threads were still spread around. Will continue trying to isolate the kernel as well... In any event, this amount of isolation was still good enough to get the c1dnn user space model running fairly stably. It's been running for the last hour without issue. I added the c1dnn channel and testpoint files to the daqd master file, and restarted daqd_dc on fb1, so now the c1dnn channels and test points are available through dataviewer etc. We were then able to observe the reconstructed signals: We'll need to set the phase rotation of the demodulated RF PD signals (REFL11, REFL55, AS55, POP22) to match them with what the NN expects... 13401 Wed Oct 25 09:32:14 2017 GabrieleSummaryLSCfurther testing of c1dnn integration; plugged in to DAQ  Quote: We'll need to set the phase rotation of the demodulated RF PD signals (REFL11, REFL55, AS55, POP22) to match them with what the NN expects... Here are the demodulation phases and rotation matrices tuned for the network. For the matrices, I am assuming that the input is [I, Q] and the output is [I,Q]. POP22 phi = 153 degrees [[-0.894, 0.447], [-0.447, -0.894]] REFL11 phi = 93 degrees [[-0.058, 0.998], [-0.998, -0.058]] REFL55 phi = -90 degrees [[0.000, -1.000], [1.000, 0.000]] AS55 phi = 7 degrees [[0.993, 0.122], [-0.122, 0.993]] 13405 Sun Oct 29 16:40:17 2017 ranaSummaryComputersdisk cleanup Backed up all the wikis. Theyr'e in wiki_backups/*.tar.xz (because xz -9e gives better compression than gzip or bzip2) Moved old user directories in the /users/OLD/ 13411 Mon Nov 6 18:22:48 2017 jamieSummaryLSCcurrent procedure for running c1dnn code This is the current procedure to start the c1dnn model:$ ssh c1lsc
$sudo systemctl start rts-epics@c1dnn$ sudo systemctl start rts-awgtpman@c1dnn
$sudo /usr/bin/cset proc -s rts-c1dnn --exec /opt/rtcds/caltech/c1/target/c1dnn/bin/c1dnn -- -m c1dnn ... Then to shutdown: ... Ctrl-C$ sudo systemctl stop rts-awgtpman@c1dnn
\$ sudo systemctl stop rts-epics@c1dnn

The daqd already knows about this model, so nothing should need to be done to the daqd to make the dnn channels available.

13421   Thu Nov 9 10:51:37 2017 gautamSummaryLSCcurrent procedure for compiling and installing c1dnn code

Jamie pointed out that the compile and install instructions are different for c1dnn:

cd /opt/rtcds/caltech/c1/rtbuild/test/nn-test
make c1dnn
make install-c1dnn

I think these build instructions have to be run on the c1lsc frontend - in the past, I have been able to compile and install models on any computer with the shared drive mounted (including the control room workstations), but I'm guessing that something has changed since the RCG upgrade. Jamie can correct me on this if I'm wrong.

13425   Fri Nov 10 18:57:41 2017 ranaSummaryElectronicsIthaca 1201 vs. SR560

I characterized the black Ithaca 1201 pre-amp that we had sitting in the racks. It works fine and the input referred noise is < 10 nV/rHz. I also checked that the filter selection switches on the front panel did what they claim and that the gain knob gives us the correct gain.

For comparison I have also included the G=100, 1000 input referred noise of one of the best SR560 that we have (s/n 02763) in the lab. Above a few Hz, the SR560 is better, but for low frequency measurements it seems that the 1201 is our friend.

As with the SR560, you don't actually get low noise performance for G < 100, due to some fixed output noise level.

Steve:  sn48332 of Ithaca 1201

13427   Tue Nov 14 16:02:43 2017 KiraSummaryPEMseismometer can testing

I made a model for our seismometer can using actual data so that we know approximately what the time constant should be when we test it out. I used the appendix in Megan Kelley's report to make a relation for the temperature in terms of time.

$\frac{dQ}{dt}=mc\frac{dT(t)}{dt}$ so $T(t)=\frac{1}{mc}\int \frac{dQ}{dt}dt=\frac{1}{mc}\int P_{net}dt$ and $P_{net}=P_{in}-P_{out}$

In our case, we will heat the can to a certain temerature and wait for it to cool on its own so $\inline P_{in}=0$

We know that $\inline P_{out}=\frac{kA\Delta T}{d}$ where k is the k-factor of the insulation we are using, A is the area of the surface through which heat is flowing, $\inline \Delta T$ is the change in temperature, d is the thickness of the insulation.

Therefore,

$T(t)=\frac{1}{mc}\int_{0}^{t}\frac{kA}{d}[T_{lab}-T(t')]dt'=\frac{kA}{mcd}(T_{lab}t-\int_{0}^{t}T(t')dt')$

We can take the derivative of this to get

$T'(t)=\frac{kAT_{lab}}{mcd}-\frac{kA}{mcd}T(t)$, or $T'(t)=B-CT(t)$

We can guess the solution to be

$T(t)=C_{1}e^{-t/\tau}+C_{2}$ where tau is the time constant, which we would like to find.

The boundary conditions are $\inline T(0)=40$ and $\inline T(\infty)=T_{lab}=24$. I assumed we would heat up the can to 40 celcius while the room temp is about 24. Plugging this into our equations,

$\inline C_{1}+C_{2}=40, C_{2}=24$, so $\inline C_{1}=16$

We can plug everything back into the derivative T'(t)

$T'(t)=-\frac{16}{\tau}e^{-t/\tau}=B-C[16e^{-t/\tau}+24]$

Equating the exponential terms on both sides, we can solve for tau

$\frac{16}{\tau}e^{-t/\tau}=16Be^{-t/\tau}, \frac{1}{\tau}=B, \tau=\frac{1}{B}=\frac{mcd}{kA}$

Plugging in the values that we have, m = 12.2 kg, c = 500 J/kg*k (stainless steel), d = 0.1 m, k = 0.26 W/(m^2*K), A = 2 m^2, we get that the time constant is 0.326hr. I have attached the plot that I made using these values. I would expect to see something similar to this when I actually do the test.

To set up the experiment, I removed the can (with Steve's help) and will place a few heating pads on the outside and wrap the whole thing in a few layers of insulation to make the total thickness 0.1m. Then, we will attach the heaters to a DC source and heat the can up to 40 celcius. We will wait for it to cool on its own and monitor the temperature to create a plot and find the experimental time constant. Later, we can use the heatng circuit we used for the PSL lab and modify the parts as needed to drive a few amps through the circuit. I calculated that we'd need about 6A to get the can to 50 celcius using the setup we used previously, but we could drive a smaller current by using a higher heater resistance.

13460   Fri Dec 1 17:09:29 2017 Udit KhandelwalSummaryGeneral

Current objectives and statuses:

• CAD Model of 40m lab (facility, chambers, invacuum components etc)
• Status: On hold since I'm unable to acquire general dimensions of
13472   Wed Dec 13 17:46:08 2017 Udit KhandelwalSummaryGeneralSummary of Current Tasks

1. 40m_bldg.dwg has 2D drawing of the 40m building

• After importing file as a 2D sketch into solidworks, make sure to retrace all the lines before performing any 3D extrusion stuff.

2. 40m_VE.dwg has the Vaccuum Envelope.

• Divided the file into individual sketches for the tubes, test mass, and beam splitter chambers (so they can be individually modified later if required).

3. 40melev.dwg has the relative positioning between (1) and (2).

• Using this file to position objects inside building cad.

4. All files can be found in Dropbox folder [40m SOS Modeling], which should be renamed to [40m CAD].

5. Next step would be to add the optical table, mirrors.

Tip-Tilt Suspension

1. Current objective: (refer to D070172) - Increase the length of the side arms (so it matches the dimensions of D960001), while keeping the test mass subassembly at the same height.
2. Future objective: Resonant frequency FEM of the frame (sans the test mass), and then change height to get the desired frequency.

Past Work

• Completed solidworks model of SOS (D960001). I understand this is not the focus right now so this is for reference that the model is ready to be used.

• I will be in India from 16th December until 6th January so this is my final visit for this year. I have enough material to work from home, and will correspond with Koji over email regarding Lab CAD and tip-tilt suspension.
13484   Fri Dec 15 18:24:46 2017 ranaSummaryOptical LeversPRM

Today Angelina and I looked at the PRM OL with an eye towards installing a 2nd QPD. We want to try out using 2 QPDs for a single optic to see if theres a way to make a linear combination of them to reduce the sensitivity to jitter of the HeNe laser or acoustic noise on the table.

The power supply for the HeNe was gone, so I took one from the SP table.

There are WAY too many optics in use to get the beam from the HeNe into the vacuum and then back out. What we want is 1 steering mirror after the laser and then 1 steering mirror before the QPD. Even though there are rumors that this is impossible, I checked today and in fact it is very, very possible.

More optics = more noise = bad.

13489   Wed Dec 20 00:43:58 2017 KevinSummaryGeneralDAC noise contribution to squeezing noise budget

Gautam and I looked into the DAC noise contribution to the noise budget for homodyne detection at the 40m. DAC noise is currently the most likely limiting source of technical noise.

Several of us have previously looked into the optimal SRC detuning and homodyne angle to observe pondermotive squeezing at the 40m. The first attachment summarizes these investigations and shows the amount of squeezing below vacuum obtainable as a function of homodyne angle for an optimal SRC detuning including fundamental classical sources of noise (seismic, CTN, and suspension thermal). These calculations are done with an Optickle model. According to the calculations, it's possible to see 6 dBvac of squeezing around 100 Hz.

The second attachment shows the amount of squeezing obtainable including DAC noise as a function of current noise in the DAC electronics. These calculations are done at the optimal -0.45 deg SRC detuning and 97 deg homodyne angle. Estimates of this noise are computed as is done in elog 13146 and include de-whitening. It is not possible to observe squeezing with the current 400 Ω series resistor which corresponds to 30 pA/rtHz current noise at 100 Hz. We can get to 0 dBvac for current noise of around 10 pA/rtHz (1.2 kΩ series resistor) and can see 3 dBvac of squeezing with current noise of about 5 pA/rtHz at 100 Hz (2.5 kΩ series resistor). At this point it will be difficult to control the optics however.

So it seems reasonable to reduce the DAC noise to sufficient levels to observe squeezing, but we will need to think about the controls problem more.

13490   Thu Dec 21 19:25:48 2017 KevinSummaryGeneralDAC noise contribution to squeezing noise budget

Gautam and I redid our calculations, and the updated plot of squeezing as a function of DAC current noise per coil is shown in the attachment. The current noise is calculated as the maximum of the filtered DAC noise and the Johnson noise of the series resistor. The total noise is for four optics with four coils each.

The numbers are worse than we quoted before: according to these calculations we can get to 0 dBvac for current noise per coil of about 2.4 pA/rtHz at 100 Hz.

 Quote: Gautam and I looked into the DAC noise contribution to the noise budget for homodyne detection at the 40m. DAC noise is currently the most likely limiting source of technical noise. Several of us have previously looked into the optimal SRC detuning and homodyne angle to observe pondermotive squeezing at the 40m. The first attachment summarizes these investigations and shows the amount of squeezing below vacuum obtainable as a function of homodyne angle for an optimal SRC detuning including fundamental classical sources of noise (seismic, CTN, and suspension thermal). These calculations are done with an Optickle model. According to the calculations, it's possible to see 6 dBvac of squeezing around 100 Hz. The second attachment shows the amount of squeezing obtainable including DAC noise as a function of current noise in the DAC electronics. These calculations are done at the optimal -0.45 deg SRC detuning and 97 deg homodyne angle. Estimates of this noise are computed as is done in elog 13146 and include de-whitening. It is not possible to observe squeezing with the current 400 Ω series resistor which corresponds to 30 pA/rtHz current noise at 100 Hz. We can get to 0 dBvac for current noise of around 10 pA/rtHz (1.2 kΩ series resistor) and can see 3 dBvac of squeezing with current noise of about 5 pA/rtHz at 100 Hz (2.5 kΩ series resistor). At this point it will be difficult to control the optics however. So it seems reasonable to reduce the DAC noise to sufficient levels to observe squeezing, but we will need to think about the controls problem more.

13491   Fri Dec 22 09:40:19 2017 ranaSummaryGeneralDAC noise contribution to squeezing noise budget
1. Should not count the ITMs. On those we can use big resistors / filters to cut out the noise.
2. For the initial LIGO, we used 7 K resistors and the mass was 10 kg. But...the output driver went +/- 150 V.

So we had a max F/m = (20 mA * 0.064 N/A)/(10 kg) = 0.0001. For the 40m, to get the same thing, we would need 40x less current (~0.5 mA). At the moment we have (12 V / 400 Ohm) = 30 mA.

We need to get a spectrum and times series of the required coil current for acquiring and holding the DRMI, and also the single arm. Then we can see where to make noise reductions to allow this drastic force reduction.

13494   Sun Dec 31 12:43:50 2017 ranaSummaryElectronicsSR560: reworking

I have ordered some LSK389A (in both the SOIC-8 and TO-71 packages) to replace the SR560's default front end FET pair (NPD5565).

I'm going to rework s# 00619 once these new FETs come in. Also ordered 100 of the SOIC-8 to DIP-8 adapter boards from Digikey.

This plot shows the current performance compared to the Rai Low Noise box. I expect the FETs should let us get to ~1.5 nV/rHz with the SR560.

13516   Mon Jan 8 20:50:01 2018 ranaSummaryElectronicsSR560: reworking

I replaced the NPD5565 with a LSK389 (SOIC-8 with DIP adapter). There was a noise reduction of ~30%, but not nearly as much as I expected. I wonder if I have to change the DC bias current on these to get the low noise operation?

https://photos.app.goo.gl/hsMwsif7NLscsgpx1

13544   Fri Jan 12 20:35:34 2018 Udit KhandelwalSummaryGeneral2018/01/12 Summary
1. Worked further on positioning vacuum tubes and chambers in the building.
2. Next step would be to find some drawings for optical table positions and vibration isolation stack. Need help with this!
2. Tip Tilt Suspension (D070172)
1. Increased the length of side arms. The overall height of D070172 assembly matches that of D960001.
2. The files are present in dropbox in [40mShare] > [40m_cad_models] > [TT - Tip Tilt Suspension]
13560   Fri Jan 19 15:22:19 2018 Udit KhandelwalSummaryGeneral40m CAD update 2018/01/19

1. All parts will be now named according to the numbering system in this excel sheet: LIGO 40m Parts List in dropbox folder [40mShare] > [40m_cad_models] > [40m Lab CAD]
2. I've placed optical tables in the chambers at 34.82" from the bottom for now. This was chosen by aligning the centre of test mass of SOS assembly (D960001) with that of vacuum tube (Steve however pointed out last week they might not necessarily be concentric).

13566   Mon Jan 22 12:48:48 2018 KojiSummaryGeneralBeat setup for aLIGO EOM test

I'm planning to construct a beat setup between the PSL and AUX beams. I am going to make it in the area shown in a blue square in the attached photo. This does not disturb Johannes' and PSL setups. The beams are obtained from the PBS reflection of the PSL and the dumped beam of the aux path (0th or 1st order beam of the AOM).

13567   Mon Jan 22 20:54:58 2018 KojiSummaryGeneralAUX-PSL beat setup

The beat setup has been made on the PSL table. The BS and the PD were setup. The beat was found at 29.42degC and 50.58degC for the PSL and AUX crystal temperatures, respectively.
We are ready for the EOM test. I have instruments stacked around the PSL table. Please leave them as they are for a while. If you need to move them, please contact with me. Thanks.

A picked-off PSL after the main modulator was used as the PSL beam. This was already introduced close to the setup thanks to the previous 3f cancellation test ELOG 11029. The AUX beam was obtained from the transmission of 90% mirror. Both paths have S polarization. The beams are combined with a S-pol 50% BS. The combined beam is detected by a new focus 1GHz PD.

The PSL crystal temp (actual) was 50.58degC. The AUX crystal temp was swept upward and the string beat was found at 50.58degC. After a bit of alignment, the beat strength was -18dBm (at 700V/A RF transimpedance of NF1611) .

13575   Wed Jan 24 13:55:04 2018 KiraSummaryPEMSeismometer can insulation test

Gautam and I set up the insulated seismometer can in the lab today. I had previously wired up the two heaters I placed onto the sides of the can in parallel to get a total resistance of 12.5 ohms and then I wrapped the whole can in 3 layers of insulation (k-factor 0.25). We placed it on a large sheet of insulation as to not crush the wires leading out the bottom of the can. I stuck on one of my AD590 sensors to the inside of the can onto the copper lining using duct tape, though this is only a temporary solution. In the future, it would be nice to have some sort of thermal clamp to secure the sensor to the can. To provide power to the heater circuit board and the temperature sensor board, we got a powerstrip and plugged in two power supplies and a function generator into it. The heater circuit (attachment 3) is powered by one of the power supplies and the function generator, while the temperature sensor (attachment 5) is stuck to the side of the can and is powered by the second power supply. The heater circuit's MOSFET (IRF640, attachment 4) is placed on a metal block and sandwiched between two more to make sure it doesn't move around. The temperature sensor is connected by a long BNC cable to the channels in attachment 6.

gautam: we plugged the BNC output of Kira's temperature sensor circuit to J7 on the AA input chassis in 1X2 - this corresponds to ADC1 input 12 in c1ioo. I then made a "PEM" namespace block inside the c1als model, and placed a single CDS filter module inside it (this can be used for calibration purposes). The filter module is named "C1:PEM-SEIS_EX_TEMP", and has the usual CDSfilt channels available. I DQ'ed the output of the filter module (@256 Hz, probably too high, but I'm holding off on a recompile for now). Recompilation and model restart of c1als went smoothly.

2 bench power supplies are being used for this test, we can think of a more permanent solution later.

**25 Jan noon: Added another filter module, "C1:PEM-SEIS_EX_TEMP", to which Kira is hooking up a second temperature sensor, which will serve as a monitor of the "Ambient" lab temperature. Added DQ channel for the output of this filter module, fixed sampling to 32Hz. Compile and restart went smooth.

13601   Fri Feb 2 21:12:46 2018 Udit KhandelwalSummaryGeneralSummary - 2018/02/02

Discussed with Koji about motivation to simplify the design of this assembly, which has many unnecessary over-constraints. I have started to cad alternate parts with the aim of removing these over-constraints.

Acquired a stack of original engineering drawings of the vacuum chambers from Steve which I will take home, get scanned, and then use as reference for the cad i'm working on.

Other:
Started paperwork at west bridge office to get paid as an "occasional employee". Hopefully I receive old money.

13602   Fri Feb 2 22:47:00 2018 KojiSummaryGeneralAP1053: Packaging & Performance

I've packaged an AP1053 in a Thorlabs box. The gain and the input noise level were measured. It has the gain of ~10 and the input noise of ~0.6nV/rtHz@50MHz~200MHz.

Details

AP1053 was soldered on Thorlabs' PCB EEAPB1 (forgot to take a picture). The corresponding chassis is Thorlabs' EEA17. There is a 0.1uF high-K ceramic cap between DC and GND pins. The power is supplied via a DC feedthru capacitor (Newark / Power Line Filter / 90F2268 / 5500pF) found in the WB EE shop. The power cable has a connector to make the long side of the wires detachable. Because I did not want to leave the RF signal path just mechanically touched, the SMA connectors were soldered to the PCB. As the housing has no access hole, I had to make it at one of the sides.

The gain of the unit was measured using the setup shown in the upper figure of Attachment 2. When the unit was energized, it drew the current of about 0.1A. The measued gain was compensated by the pick off ratio of the coupler (20dB). The gain was measured with the input power of -20, -10, 0, 10, and 15dBm. The measurement  result is shown in Attachment 3. The small signal gain was actually 10dB and showed slight degradation above 100MHz. At the input of 10dB some compression of the gain is already visible. It looks consistent with the specification of +26.0dBm output for 1dB compression above 50MHz and +24.0dBm output below 50MHz.

The noise level was characterized with the setup shown in the bottom figure of Attachment 3. The noise figure of the amplifier is supposed to be 1.5dB above 200MHz and 3.5dB below 200MHz. This is quite low and the output noise of AP1053 can not be measured directly by the analyzer. So, another LN amplifier (ZFL-500HLN) was stacked. The total gain of the system was measured in the same way as above. The measured noise level was ~0.7nV/rtHz between 50MHz and 200MHz. Considering the measurement noise level of the system, it is consistent with the input referred noise of 0.6nV/rtHz. I could not confirm the advertized noise figure of 1.5dB above 200MHz. The noise goes up below 50MHz. But still 2nV/rtHz at 3MHz. I'd say this is a very good performance.

13638   Fri Feb 16 21:03:17 2018 Udit KhandelwalSummaryGeneralSummary 2018/02/16

Updated the dimensions of and fleshed out the chambers in greater detail, by referring to the engineering drawings that Steve gave to me. I have scanned and uploaded most of these drawings to Dropbox in [40mShare]>[40m_cad_models]>[Vacuum Chamber Drawing Scans]. The excel file "LIGO 40m Parts List" in the [40m Lab CAD] folder also lists the Steve drawings I referenced for dimensions of each part.

Next steps:
1. Finish details of all chambers.
2. Start placing representative blocks on the optical table.

13654   Fri Feb 23 20:46:04 2018 Udit KhandelwalSummaryGeneralCAD Summary 2018/02/23

I have more or less finished cadding the test mass chamber by referring to the drawings Steve gave me. Finer details like lugs and bolts and window flaps can be left for later. Here's a quick render:

13662   Wed Feb 28 21:14:34 2018 gautamSummaryPEMChannel admin

Since we decided to use the Acromag for readback of the temperature sensor for Kira's seismometer temperature control, I enabled logging of the channel Johannes had reserved for this purpose last week. Kira has made the physical connection of a temperature sensor to the BNC input for this channel - it reads back -2.92 V right now, which is around what I remember it being when Kira was doing her benchtop tests. I edited C0EDCU.ini to enable logging of this channel at 16 Hz. Presumably, a study of the ADC noise of the Acromag at low frequencies has to be made to ensure appropriate whitening (if any) can be added. Channel name is C1:PEM-SEIS_EX_TEMP_MON. Similarly, there is C1:PEM_SEIS_EX_TEMP_CTRL which is meant to be the control channel for the servoing. Calibration of the temperature sensor readback into temperature units remains. It also remains to be verified if we can have these slow EPICS channels integrated with a fast control model, or if the PID temperature control will be purely custom-script based as we have for the FSS slow loop.

I removed the fast channels I had setup temporarily in c1als. Recompilation and restart of the model went smoothly.

 Quote: I then made a "PEM" namespace block inside the c1als model, and placed a single CDS filter module inside it (this can be used for calibration purposes). The filter module is named "C1:PEM-SEIS_EX_TEMP", and has the usual CDSfilt channels available. I DQ'ed the output of the filter module (@256 Hz, probably too high, but I'm holding off on a recompile for now). Recompilation and model restart of c1als went smoothly.
13677   Fri Mar 9 20:35:41 2018 Udit KhandelwalSummaryGeneralSummary 2018/03/09

1. Optical Table Layout

I had discussed with Koji a way to record coordinates of optical table equipments in a text file, and load to solidworks. The goal is to make it easier to move things around on the table in the CAD. While I have succeeded in importing coordinates through txt files, there is still a lot of tediousness in converting these points into sketches. Furthermore, the task has to be redone everytime a coordinate is added to or changed in the txt file. Koji and I think that this can all be automated through solidworks macros, so I will explore that option for the next two weeks.

Steve will help find manufacturing drawings of the BS chamber. I have completed the ETM chambers, while the ITM ones are identical to them so I will reuse parts for the CAD.

13683   Thu Mar 15 16:00:25 2018 Larry WallaceSummaryComputersCert renewal for NODUS

The cert for nodus has been renewed for another 2 years.

The following is the basic procedure for getting a new cert: (Note certs are only good for two years as of 2018)
openssl req -sha256 -nodes -newkey rsa:2048 -keyout nodus.ligo.caltech.edu.key -out nodus.ligo.caltech.edu.csr
Country Name (2 letter code) [AU]:US
State or Province Name (full name) [Some-State]:CaliforniaLocality Name (eg, city) []:Pasadena
Organization Name (eg, company) [Internet Widgits Pty Ltd]:California Institute of Technology
Organizational Unit Name (eg, section) []:LIGO
Common Name (eg, YOUR name) []:nodus.ligo.caltech.edu

Leave the e-mail address, challenge password and optional company name blank. A new private key will be generated.
chown root nodus.ligo.caltech.edu.key
chgrp root nodus.ligo.caltech.edu.key
chmod 0600 nodus.ligo.caltech.edu.key

The nodus.ligo.caltech.edu.csr file is what is sent in for the cert.
This file should be sent to either ryan@ligo.caltech.edu or security@caltech.edu and copy wallace_l@ligo.caltech.edu.

Once the files are downloaded, the new cert and intermediate cert, they can be copied and renamed.

The PEM-encoded host certificate by itself is saved at:

/etc/httpd/ssl/nodus.ligo.caltech.edu.crt

The nodus.ligo.caltech.edu.key file should be in the same directory or whichever directory is indicated in the ssl.conf located in /etc/httpd/conf.d/  directory.

httpd will need to be restarted in order for it to see the new cert.

13686   Mon Mar 19 07:37:00 2018 Angelina PanSummary Proposed QPD Optical Arrangement

I am currently working on an optical arrangement consisting of a QPD that measures the fluctuations of an incoming HeNe laser beam that is reflected by a mirror. The goal is to add a second QPD to the optical arrangement to form a linear combination that effectively cancels out the (angular) fluctuations from the laser beam itself so that we can only focus on the fluctuations produced by the mirror.

In order to solve this problem, I have written a program for calculating the different contributions of the fluctuations of the HeNe laser and fluctuations from the mirror, for each QPD (program script attached). The goal of the program is to find the optimal combination of L0, L1, L2, and f2 that cancels the fluctuations from the laser beam (while retaining  solely the fluctuations from the mirror) when adding the fluctuations of QPD 1 and QPD 2 together.

By running this program for different combinations of distances and focal lengths, I have found that the following values should work to cancel out the effects of the oscillations from the HeNe laser beam (assuming a focal length of 0.2 m for the lens in front of the original QPD):

• L0 = 1.0000 m (distance from laser tube to mirror)
• L1 = 0.8510 m (distance from mirror to lens in front of QPD 1)
• L2 = 0.9319 m (distance from beamsplitter to lens in front of QPD 2)
• f2 = 0.3011 m (focal length of lens in front of QPD 2)

Based on these calculations, I propose to try the following lens for QPD 2:

1’’ UV Fused Silica Plano-Convex Lens, AR-Coated: 350 - 700 nm (focal length 0.3011 m). https://www.thorlabs.com/newgrouppage9.cfm?objectgroup_id=6508

ELOG V3.1.3-