I hooked up interlock to the Innolight 2W 1064 nm in the enclosure. The manual shutter is closed on this unit.
SAFETY GLASSES REQUIRED !
Innolight 2W 1064nm main laser was turned off for more enclosure related work.
The electrician re rooting the power- conduit to the interlock under the floor and bring it through the new concrete tunnel.
Emergency laser power shut off switch was tested at entry door 104M. It worked fine.
No laser glasses required. Laser will be turned back on around 11 am today.
Atm1, AC power line to enclosure and interlock was extended.
Atm2, 24V line to interlock rerouted
Atm3, job is completed
SAFETY GLASSES REQUIRED ! SAFETY GLASSES REQUIRED!
The laser is turned on. Injection current set to 0.871A.
Actually some one turned the laser off last night and did not enter it into the elog ! Burned toast award is granted !
Please come forward voluntarily.
I turned off the laser last night to protect against the electricians. but failed to elog it. I accept the burnt toast completely.
The control room's north west corner smelled like propane gas yesterday around 16:30
We all agreed that the smell was real and I called the safety office. I was told that they received 6 other calls from different parts of the campus.
The smell disappeared in about a half an hour.
Our visiting graduate student Yuta Michimura received 40m specific basic safety training today.
Our new undergrad Joonho Lee received 40m specific basic safety training yesterday.
Yuta and Joonho are scheduled to participate in the LIGO-laser safety tutelage with Peter K on Oct 12
The 2 W Innolight shutter is closed and enclosure door removed. Beam path blocked. Do not change this condition.
The PSL output beam guide was upgraded from 2" to 8" OD . It is green ready. Shutter is open.
Yuta, Joonho, and Suresh received the Basic Laser Safety Training from Peter King today.
Now, we got homework.
Our new postdoc Suresh Doravari received 40m specific safety training last week.
I blocked the AP table's south west 10" ID port since it is obsolete with the new layout.
Reminder: items on the enclosure self can fall down in an earthquake. I moved oscilloscope and heavy calorimeter head from the edge of the cliff.
The #3 enclosure door was removed for the carpenter shop this morning. They will make 5 full size doors with window on each.
TEMPORARY aluminum sheet is clamped into its place. This sheet should not be removed till new doors arrive.
The inside Formica color will be changed from white to black to reduce the green scattering.
Larisa Thorne received 40m lab specific, basic safety training. She will attend P. King's Basic Laser Safety Training Session tomorrow.
The beam of IR for doubling is clipping on bnc cable to green beam transmitted pd.
Bryan Barr is visiting us from Glasgow for a month. He received 40m specific safety training on Friday.
Mike Caton of Konecranes inspected and loadtested all 3 of the 40m cranes at max reach trolley positions with 1 ton.
Konecrane representative gave crane operator training in the 40m. Koji has become a qualified, trained crane operator of the 40m lab.
The emphasis of this annual safety audit was on safe electrical housekeeping on March 3, 2011
1064 nm transmison were measured of 40m safety glasses as shown . Their performance did not degrade. They are as good as their labels.
40m surfs: Nicole Ing, Iswita Saikia and Sonali Mohapatra received 40m specific safety training today.
Yesterday, I fried the +15 V power supply rail on one of the Eurocrate extender cards while I was checking out the binary switching in the 1X5 rack. I will describe what I did it in the hopes that everyone else will be less stupid than me.
I wanted to monitor the voltage across a resistor on the suspension OSEM whitening board. Since I knew that both sides of the resistor would be at non-zero voltage (including possibly at the power-supply rail), I used a battery-operated scope with floating inputs, so that the scope would not try to pull the probe shield to ground. That should be OK, although not recommended, as you'll see, because you must be very careful to make sure that the scopes inputs are indeed floating.
Let's call the original signal 'A'. The trouble came when I then connected another signal (B), whose shield was connected to the ground on the whitening board, to the scope. Apparently the grounds on the scope inputs are connected, or were in the configuration I was using. When I connected the signal B, B's ground shorted A's shield to ground, which had been sitting at the +15V rail. That short circuit then fried the +15V supply line on the extender card I was using (escaping magic smoke was detected). Thankfully this only blew the extender card, and not the Eurocrate or the Eurocrate power supply or the whitening board or the scope etc, all of which would have been much worse.
The moral of the story is to be very careful when connecting power supply voltages to the shield or ground of a scope. In short, don't do it. I didn't ultimately need to, since I could have found other ways to measure the same signal.
Lightwave M126-1064-700 lasers sn415 at east end of the Y arm and sn201 at the AP table are connected individually to one each EMERGENCY LASER SHUT OFF SWITCH.
I found baked allen keys on the top of the clean optics cabinet. Somewhat heavy box that can come down in an earthquake on our heads.
NOTHING SHOULD GO ON THE TOP OF THE CABINETS OR RACKS except small plastic boxes that storing our clean clothing.
Today I learned some important circuit-building lessons while testing my photosensor circuit box (i.e. how NOT to test a circuit and, conversely, things that should be done instead).
I blew my first circuit today. The victim is in the photo below (bottom 7805 voltage regulator). The plastic covering fell off after I removed the fried regulator. After checking various components, I figured out that I blew the circuit because I had forgotten to ground the regulator. Although this was very unfortunate, I did make an important discovery. While testing the voltage output of the 7805 voltage regulator (I put a new one), I discovered that contrary to the claims of the datasheet, an input voltage of 5V will not produce a steady 5V supply. I found that at 5V, my regulator was only producing 4.117 V. I was using a 5 V supply because I wanted to use only 1 power supply (I was using a two-channel power supply that had a fixed 5V output to produce the +15, -15, ground, and 5 V I need for my photosensor circuit box). After seeing this, I got a second power supply and am now using 10V to as an input for the regulator to produce 4.961V. I found that from a voltage range of 10V to 15 V, the regulator produced a steady 4.961 V supply. I have decided to use 10V as an input. My newly-grounded voltage regulator did not smoke or get hot at 10V.
After several more debugging trials (my LED was still not lighting up, according to the infared viewer), I learned another painful lesson. I learned DO NOT USE CLIP LEADS TO TEST CIRCUITS!!!! Initally, I was powering my circuit and making all of my connections between the photosensor head (2 photodiodes and 1 LED) with clip leads. This was a BAD IDEA BECAUSE CLIP LEADS ARE UNSTABLE AND IT IS VERY EASY TO SHORT A CIRCUIT IF THEY ACCIDENTALLY TOUCH! I did not realize this important lesson until my photosensor circuit was once again burning. Confused as to why my circuit was once again burning, I foolishly touched the voltage regulator. As you can see on the top voltage regulator in the photo below, my finger left its mark on the smoldering voltage regulator. As you cannot see the wincing on my face as I try to type this long elog, I will painfully type that the voltage regulator left its own mark on my finger (an ugly sore little welt). Suresh has taught me a valuable lesson: WHEN DEALING WITH SOMETHING OF QUESTIONABLE/UNKNOWN TEMPERATURE, USE YOUR NOSE, NOT YOUR FINGER TO DETERMINE IF THAT COMPONENT IS HOT!!!!
To make my circuit-testing safer, upon the suggestion of Suresh, I have since removed the clip leads and inserted a 12 pin IDC component (pictured below). There are 12 pins for the 6 inputs I will get from each of the 2 photosensor heads. I have requested orders for a 16 pin IDC connector, 15 pin Dsub male part, 15 pin Dsub feed-thru, 9 pin Dsub male part (2), and 9 pin Dsub feed-thru (2). After receiving these components, I should be able to safely test my circuit.
In the meanwhile, I can explore SimMechanics and try to figure out how to use the accelerometer
There is should be a few IDC connectors in the lab (and some ribbon cable) using which you can proceed with the testing of the circuit, if you prefer. If not we can get them from our ever helpful electronics division at Downs. In any case there is no need to lose time waiting for parts to arrive.
Our surf student Manuel Marchiò received 40m specific safety training today.
The hazardous waste people are moving chemicals around outside our door, and have roped off our regular front door.
Please go around, and use the control room door to enter and exit. It is currently unlocked, although I'll lock up when I leave for LIGOX.
Paul, Mirko and Katrin visiting grad students received the 40m basic safety training.
First aid kits are located close vicinity of entry doors and under circuit breaker panels.
Bob, Callum and Daphen noted that our keeping a JDSU HeNe (max power <4mW) is against somebody's SOP. So I cleared everything that relates to 40m SOS suspending to the bottom shelf of the 2nd cabinet in the cleanroom (the back set of cabinets nearest the flow benches). The door has a nifty label. Things that are in there include:
QPD and micrometer mount
microscope and micrometer mount
Al beam block
Magnet gluing fixture
dumbbell gluing fixture
The electronics that we use (HeNe's power supply, 'scope, QPD readout) are still on the roll-y thing under the flow bench.
It started with fire alarm test yesterday at 14:50 All alarms are functioning VERY loud and their flashers are bright. Evacuation drill followed. We assembled at north west corner of the 40m building and counted 6 heads.
Nobody was left sleeping inside. Bob carried the success report of the drill to PMA office immediately.
Emergency exit lights were inspected: 2 out of 13 batteries have to be replaced
One of the Halon fire extinguishers needs to be recharged out of 8
Please do participate in preparation for the upcoming safety audit on Feb 28
Batteries replaced and cylinder recharged. Please clean up your experimental set up if it is blocking breakers or entry way etc.
I will start the final clean up 2pm today.
Correction list by visiting safety committee, Haick Issaian is not shown:
1, update laser, crane operator list and post it
2, check fire extinguishers monthly, date and initials must be on the tags
3, move drinking water tower so it does not block fire extinguisher
4, post updated crane doc at cranes
5, post present phone lists at IFO room phones
6, emergency laser shutoff at the south end must be mounted without C-clamp
7, use heavy cable tie to insure position of mag-fan on cabinet top
Additional to do list:
a, safety glasses to be cleaned
b, let the electrical shop fix Rack-AC power to optical tables at the ends
c, measure transmission of laser safety glasses
d, update IFO outside door info signs
e, update laser inventory and post it
f, schedule annual crane inspection and renew maintenance contract
g, PSL enclosure inner shelf needs a good clean up so it is earthquake safe
Today I've attended the laser safety seminar.
ETMY sus damping was disabled. Green locking laser and associated electronics turned off. Computers and power supplies turned off at rack 1Y4
The electricians picking up ac power from 1Y4 manual disconnect box and installing conduit line to ISCT-ETMY east end optical table.
There will be no more daisy chaining this way.
The power is back on at ETMY . c1iscey has not been restarted.
Now I'm turning ac power off at ETMX for the same job to be done.
Bob cleaned all safety glasses in 10 % Liquinox soap in water solution first. The transmittance of glasses were checked at 100 mW 1064 nm S polarization, beam diameter 0.5 mm at 0-20 deg incident.
Coherent FieldMate power meter measured T= < 1 mW of all glasses.
Completed with the exception of d and g
Masha, Eric, Yaakov, Liz and Sasha received 40m specific basic safety training.
Caltech security called me at 1am Sunday. Control room emergency exit door was found unlocked!
It is your responsibility to lock door if you unlocked it !
I unlocked the door on Tuesday in order to move the red cart.
After that I confirmed that the door was locked.
It is your responsibility to lock door if you unlocked it !
Koji just found the emergency exit door unlocked again. NOT GOOD.
We have determined that if you use the emergency door to enter the lab, it leaves the door unlocked, unless you go back outside and deliberately lock it. This means that someone has been using the emergency exit as a regular entrance.
It's fine to leave by that door, but you should make a habit of entering through the regular door. Using the back door as an entrance is a special case situation, when they have the main door blocked.
Rijuparna Chakraborty and Elli Elenora King received 40m specific basic safety training in the 40mLab
Eric Quintero and Mike Jenson received 40m specific basic safety training.
Den Martynov received 40m specific safety training.
We were sitting trying to lock MICH (hooooorraaaayy!!!), and the emergency lights above the control room door came on, and then ~1 minute later turned off. Steve, can you see what's up?
Ayaka Shoda, visiting graduate student received basic 40m specific safety training today
Charles Blakemore our new undergrad help received 40m specific safety training today.