For several of the channels on the PEM ADCU, zeros are occuring at the same time. Does anyone know why that might happen or how to fix it?
Added zita to the martian host table, and configured his network accordingly:
zita A 192.168.113.217
I installed 'nfs-client' on zita (the StripTool terminal). It now has mounted all the shared disks, but still can't do StripTool since its a 32-bit machine and our StripTool is 64.
I'll finish the rest of the config to turn it into a full-fledged workstation tomorrow.
Here is a comparison of the response of various DoFs in our various RFPD sensors for two different CARM offsets. Even in the case of the smaller CARM offset of ~1kHz, we are several linewidths away from the resonance. Need to do some finesse modeling to make any meaningful statement about this - why is the CARM response in REFL11 apparently smaller for the smaller CARM offset?
If you mistrust my signal processing, the GPS times for which I ran the sensing lines are:
CARM offset = ~30kHz (arm transmission <0.02) --- 1257064777+5min
CARM offset = ~1kHz (arm transmission ~5) --- 1257065566+5min
There seems to be stronger-than-expected coupling between CARM and the 3f sensors.