40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 40m Log, Page 99 of 344 Not logged in
ID Date Author Type Category Subject
4300   Tue Feb 15 11:56:17 2011 josephbUpdateCDSUpdated some DAQ channel names
That is my fault for not running the activateDAQ.py script after a round of rebuilds. I have run the script this morning, and confirmed that the oplev channels are showing up in dataviewer.

Quote:

Although Joe and Kiwamu claim that they have inserted the correct DAQ names for the OPLEVs (e.g. PERROR and YERROR) back in Jan. 11, when I look today, I see that these channels are missing!

## I want my PERROR/YERRORs back!

4302   Tue Feb 15 15:06:25 2011 josephbUpdateCDSCDS todo list for tomorrow morning

Currently, there is a test directory called /opt/rtcds/caltech/c1/new_core where we have the latest svn checkout.  Tomorrow (after everything works), it will become the core directory.

1) Modify on the fb machine the /diskless/root/etc/ld.so.cache file.  This is done by logging into fb, going to /etc/ld.so.conf.d/, modifying epics-x86_64.conf to only have .10 stuff , and running sudo /sbin/ldconfig.  Copy the newly generated /etc/ld.so.cache file to /diskless/root/etc/.

2) Modify the rc.local file on the fb machine in /diskless/root/etc/ to take advantage of the new subscripts and init.d/ start scripts.

3) Add the no_rfm_dma to all the iop models (c1x01,c1x02,c1x03,c1x04,c1x05).

4) Rebuild all front end models with new code.  Install.

5) Build awgtpman and mx_streams with new code.

6) Rerun activateDaq.py (to fix channel names from all the rebuilt code).

7) Double check Burt request files have the switch fix.

8) Restart the front ends.

9)Restart the frame builder.

9) Check channels, exitations, RFM connections.

10) Check Monit is working.

4308   Wed Feb 16 12:16:14 2011 josephbUpdateCDSFixed Optical level SUM channel names

[Joe,Valera]

Valera pointed out the OPLEV SUM channels were incorrect.  We changing the optical level sum channel to _OPLEV_SUM when it should have been OL_SUM.  This has been fixed in the activateDAQ.py script.

4311   Thu Feb 17 11:20:04 2011 josephbUpdateCDSstart scripts no longer need sudo

I've modified the rc.local file to run the IOC codes as controls, which means they no longer write root permission log files on startup.

The awgtpman, which was the other permission issue with the start scripts, is started by a run script now.  This new version seems to be content to keep the permissions of the current log file, which is set to controls.

This should prevent the issue of sudo wiping your path environment variable for just that command. (Try "sudo which burtwb" versus "which burtwb" for example).  This apparently a security feature of sudo.

If you should happen to use sudo to run a start script, the easiest solution to fix the permissions is just got to the target directory (type "target") and run "sudo chown controls:controls -R *" on one of the workstations (the front ends don't handle the groups properly at the moment).

This should allow the scripts to properly use burtrb and burtwb to write and backup burt files.

4312   Thu Feb 17 11:49:48 2011 josephbUpdateCDSFront end start/stop scripts go to /scripts/FE again

I modified the core/advLigoRTS/Makefile to once again place the startc1SYS and killc1SYS scripts in the scripts/FE/ directory.

It had been reverted in the SVN update.

4313   Thu Feb 17 11:51:14 2011 josephbUpdateCDSLockin filter names too long - broke loading

Problem:

Could not load filters into the C1:SUS-ETMX_LOCKIN1_SIG filter bank.

Reason:

Apparently the filter bank name was too long.  I'm not sure why this isn't caught by the real time code generator, I'm planning on asking Alex and Rolf about it today.

Solution:

Reduce the name of the components.  Basically LOCKIN1 needs to become something like LOCK1 or LIN1.

In related news, it looks like the initial filters are hard coded to be 2048 Hz.  Given that they start out empty they won't cause things to break immediately, and if you're editing the file you can update the rate as you add the filter.  I'll also bring this up with Alex and Rolf and see if the RCG can't be more intelligent about its filter generation.

4320   Thu Feb 17 23:56:53 2011 josephbUpdateCDSDaqd was rebuilt, now reverted.

As one of the trouble shooting steps for the daqd (i.e. framebuilder) I rebuilt the daqd executable.  My guess is somewhere in the build code is some kind of GPS offset to make the time correct due to our lack of IRIG-B signal.

The actual daqdrc file was left untouched when I did the new install, so the symmetricom gps offset is still the same, which confuses me.

I'll take a look at the SVN diffs tomorrow to see what changed in that code that could cause a 300000000 or so offset to the GPS time.

4323   Fri Feb 18 13:41:22 2011 josephbUpdateCDSCDS fixes

I talked to Alex today and had two things fixed:

First the maximum length of filter names (in the foton C1SYS.txt files in /chans) has been increased to 40, from 20.  This does not increase EPICS channel name length (which is longer than 20 anyways).

This should prevent running into the case where the model doesn't complain when compiled, but we can't load filters.

Additionally, we modified the feCodeGen.pl script in /opt/rtcds/caltech/c1/core/advLigoRTS/src/epics/util/ to correctly generate names for filters in all cases.  There was a problem where the C1 was being left off the file name when in the simulink .mdl file the filter was located in a box which had "top_names"  set.

4343   Wed Feb 23 10:37:02 2011 josephbSummaryIOOMyterious data loss: FB needs investigation

Friday:

In addition to the other fixes, Alex rebuilt the daqd process. I failed to elog this. When he rebuilt it, he needed change the symmerticom gps offset in the daqdrc file (located in /opt/rtcds/caltech/c1/target/fb).

On Friday night, Kiwamu contacted me and let me know the frame builder had core dumped after a seg fault.  I had him temporarily disable the c1ass process (the only thing we changed that day), and then replaced Alex's rebuilt daqd code with the original daqd code and restarted it.  However, I did not change the symmetricom offset at this point.  Finally, I restarted the NDS process.  At that point testpoints and  trends seemed to be working.

Sunday:

The daqd process was restarted sometime on Sunday night (by Valera i believe).  Apparently this restart finally had the symmetricom gps offset kick in (perhaps because it was the first restart after the NDS was restarted?).  So data was being written to a future gps time.

Monday:

Kiwamu had problems with testpoints and trends and contacted me.  I tracked down the gps offset and fixed it, but the original daqd process only started once successfully, after that is was segfault, core dump non-stop. I tried Alex's rebuilt daqd (along with putting the gps offset to the correct value for it), and it worked.  Test points, trends, excitations were checked at the point and found working.

I still do not understand the underlying causes of all these segmentation faults with both the old and new daqd codes.  Alex has suggested some new open mx drivers be installed today.

 Quote: Looks like there was a mysterious loss of data overnight; since there's nothing in the elog I assume that its some kind of terrorism. I'm going to call Rolf to see if he can come in and work all night to help diagnose the issue.

4344   Wed Feb 23 11:53:30 2011 josephbUpdateCDSUpdated mx drivers

Alex and I updated the open mx drivers from 1.3.3 to 1.3.901 (1.4 release candidate).  We downloaded the drivers from: http://open-mx.gforge.inria.fr/

We put them in /root/open-mx-1.3.901 on the fb machine (and thus get mounted by all the front ends.).  We did configure and make and make install.

We did a quick check with /opt/mx/bin/mx_info on the fb machine at this point and realized the MAC addresses and host names were all messed up, including two listings for c1iscex with different mac addresses (neither of which was c1iscex).

We then brought all the front ends mx_streams down, brought the fb down, then cleared all the peer names with mx_hostname.  We then brought the fb up, and the front end mx_stream processes.

/opt/mx/mx_info now shows a clean and correct set of hostnames and mac addresses.  Testpoints and trends are working.

4364   Mon Feb 28 11:22:40 2011 josephbSummaryGeneralto do list
 Quote: - Where is the CDS TO DO ==> Joe

## CDS To Do:

1) Get ETMY working - figure out why signals are not getting past the AI board (D000186) to the coils.

2) Get TDS and command line AWG stuff working

3) Get c1ass and new c1lsc (with Koji) fully integrated with the rest of the system.

4) Get CDS software instructions up to date and well organized.

5) Redo cabling  and generally make it a permanent installation instead of hack job:

a) Measure cable lengths, check connectors, wire with good routes and ensure strain relief.  Ensure proper labeling

b) Get correct length fiber for c1sus RFM and timing.

c) Fix up final BO adapter box and DAC boxes.

d) Make boxes for the AA filter adapters which are currently just hanging.

e) Get two "faceplates" for the cards in the back of the ETMY IO chassis so they can screwed down properly.

f) Remove and properly store old, unused cables, boards, and anything else.

6) Create new documentation detailing the current 40m setup, both DCC documents and interactive wiki.

7)  Setup an Ubuntu work station using Keith's wiki instructions

## Simulated Plant To Do:

1)Create simulated plant to interface with current end mass controls (say scx).

2) Create proper filters for pendulum and noise generation, test suspension.

3) Propagate to all other suspensions.

4) Working on simulated IFO plant to connect to LSC.  Create filters for near locked (assume initial green control perhaps) state.

5) Test LSC controls on simulated IFO.

6) Fix c code so there's seamless switching between simulated and real controls.

## CDS Status:

 MC damp dataviewer diaggui AWG c1lsc c1ioo c1sus c1iscex c1iscey RFM The Dolphins Sim.Plant Frame builder TDS Cabling
4380   Mon Mar 7 17:22:39 2011 josephbUpdateCDSNew simulated plant work

[Joe, Jamie]

We modified the c1scx model to have a switch to go between simulated and real plants.  The channel is currently C1:SCX-SIM_SWITCH.

When this channel is zero, the simulated plant channels are going to the ADCs and zeros  are going out to the real DACs.  When this channel is one, the real ADCs are coming in, and real data is going out to the DACs.

Jamie will be adding a big green/red light to the suspension screens which indicate the state of the simulated plant.  We will also eventually add this to the overall status screen.

A control screen for the simulated plant is located at /opt/rtcds/caltech/c1/medm/c1spx/master/C1SUP_ETMX.adl.  These are currently a work in progress.

4388   Tue Mar 8 16:59:47 2011 josephbUpdateCDSSimulated Plant Work

The screens for the simplified c1spx model have been updated.  I re-introduced the suspension point information into the sensor output matrix so we can take into account the fact that as the entire supporting structure moves, the osems moves relative to the optic.

Master screens for the noise filters (i.e. 60 Hz, suspension point motion, and optic noise) have been created.

I have currently set the matrix values of the c1spx model to handle just longitudinal motion.  I.e. Coils drive only in the POS degree of freedom and sensor read outs are also only in the POS degree of freedom.  I've turned off all the noise inputs.

I added a simple double pole at 1 Hz in the C1:SUP_ETMX_PL_F2P_0_0 filter bank.

4396   Thu Mar 10 13:44:56 2011 josephbUpdateCDSAdded digitization noise to the c1spy model for simulated ADCs/DACs

To simulate digitization noise, the easiest way I found was to use the MathFunction block, found in the CDS_PARTS model, under simLinkParts.

The MathFunction block supports square of input value, square root of input value, reciprocal of input value, and modulo of two input values.

The last is useful because it casts the input values as integers before taking the modulo.By placing this block after the saturation block (set to +/- 32768), adding 32768.5, choosing the 2nd input to be larger than 2 * 32768 (100,000 in this case), and then subtracting 32768, we wind up with a rounding function.

The above method has been applied to the c1spy model in the CI and SO out sub-blocks.

4404   Fri Mar 11 11:33:24 2011 josephbUpdateCDSFixed mistake in Matrix of Filter banks naming convention

While fixing up some medm screens and getting spectra of the simulated plant, I realized that the naming convention for the Matrices of Filter banks was backwards when compared to that of the normal matrices (and the rest of the world).  The naming was incorrectly column, row.

This has several ramifications:

1) I had to change the suspensions screens for the TO_COIL  output filters.

2) I had to change the filters for the suspension with regards to the TO_COIL output filters so they go in the correct filter banks.

3) Burt restores to times previous March 11th around noon, will put your TO_COIL output filters in a funny state that will need to be fixed.

4) The simplant RESPONSE filters had to be moved to the correct filter banks.

5) If you have some model I'm not aware of that uses the FiltMuxMatrix piece, it is going to correctly build now, but you're going to have to move filters you may have created with foton.

4409   Sun Mar 13 16:46:48 2011 josephbUpdateCDSETMY Sim work

4. The blue Output Filters  section has been changed to agree with the new filter of matrices row, column labeling.  My fault for not testing it and realizing it was broken.  The change was made in /opt/rtcds/caltech/c1/medm/master/C1SUS_DEFAULTNAME.adl and then ,/generate_master_screens.py was run, updating all the screens.

5.  I have swapped the logic for the sensor filter banks (ULSEN, URSEN, etc).  It now sends a "1" to the Binary Output board controlling the OSEM analog whitening when the FM1 filter is ON.  This has been done for all the suspensions (BS, ITMX,ITMY, SRM, PRM, MC1, MC2,MC3, ITMX, ITMY).

I am also updating the first sensor filter banks for the BS, ITMX, ITMY, SRM, PRM,MC1,MC2,MC3, called "3:30", to match the Y and X ends.

8.  I can't find any documentation on how to get a momentary button press to toggle states.  I could stick a filter bank in and use the on/off feature of that part, but that feels like a silly hack.  I've decided for the moment to split the TM offset button into 2, one for ON, one for OFF.  I'll put in on the list of things to have added to the RCG code (either a method, or documentation if it already exists).

EDIT:  TM offset still doesn't work.  Will worry about it next week.

9.  Fixed a connection in SPY/SPX models where  the side senor path that was missing a constant to a modulo block.

 Quote: I did some work on the ETMY real and Sim. Set SUS coil gains to have the same quadropole arrangement as the magnets do (-1, 1, 1, -1) so that POS = POS instead of pringle. Set the Sim Magnet polarities to match this. These are the ETMY_CI filter banks. Found that the Xycom cable from the ETMY slow controls was unplugged at the Xycom side. This was preventing enabling the ETMY coil driver and so there was no real damping of the suspension going on. I plugged it in and checked that the mirror could now be moved. The C1SUS_ETMY master screen's BLUE output filter area is now mis-labeled. If you trust the screen you would set it up to drive the suspension incorrectly. This MUST be fixed along with all of the other misleading features of the screen. ETMY SUSSIDE filter bank had a 2048 Hz sample rate and was making the damping not work correctly. Fixed to 16384 Hz. 12 Hz, 4th order Cheby low pass added and turned on for the local damping filtering. This is not optimum, but its just there to give us some filtering without introducing some instability via phase lag around 3 Hz. ETMY OL beam re-aligned on ISCT-EX. TM Offset buttons not working on the main overview screen. It seems like there is still a problem with the input whitening filters. I believe the Xycom logic is set such that the analog whitening of the OSEM signals is turned ON only when the FM1 is turned OFF. Joe has got to fix this (and elog it) so that we can damp the suspension correctly. For now, the damping of the ETMY and the SETMY require different servo gains and signs, probably because of this.

4410   Fri Mar 18 11:29:36 2011 josephbUpdateCDSMinute trend issues

[Joe, Alex]

Steve pointed out to me today he couldn't get trends for his PEM slow channels like C1:PEM-count_full.

I experimented a bit and found for long time requests (over 20 days), it would produce minute trends up to the current time, but only if they started far enough back.  So the data was being written, but something was causing a problem for dataviewer/NDS to find it.

On further investigation it looks to be some incorrect time stamps at several points in the last few months are causing the problems.  Basically when Alex and I made mistakes in the GPS time stamp settings for the frame builder (daqd) code, the wrong time got written for hours to the raw minute trend data files.

So Alex is going to be running a script to go through the roughly 180 gigabytes of affected trend data to write new files with the correct time stamps.  Once it done, we'll move the files over.  We'll probably lose a few hours worth of recent trend data, depending on how quickly the scripts run, but after which minute trends should work as they are supposed to.

4415   Fri Mar 18 17:25:21 2011 josephbUpdateCDSLockins in c1sus update, suspension screens updated

I updated our lockin simulink pieces to use the newer, more streamlined lockin piece that is currently in CDS_PARTS (with new documentation block!).  It means we are no longer passing clock signals through three levels of boxes.

In order to use the piece, you need to right click on it after copying from CDS_PARTS and go to Link Options->Disable Link.  This forces the .mdl to save all the relevant information about the block rather than just a pointer to the library.  I talked with Rolf and Alex today and we discussed setting up another model file, non-library format for putting generically useful user blocks into, rather than using the CDS_PARTS library .mdl.

The BS, ITMX, ITMY, PRM, SRM, ETMX, ETMY now have working lockins, with the input matrix to them having the 2nd input coming from LSC_IN, the 3rd from the oplev pitch, and the 4th from oplev yaw.

This necessitated a few name changes in the medm screens.  I also changed the lockin clock on/off switch to a direct amplitude entry, which turns green when a non-zero value is entered.

Currently, the Mode cleaner optic suspension screens have white lockins on them.  I started modifying a new set of screens just for them, and will modify the generate_master_screens.  Unfortunately, this requires modifying two sets of suspension screens going forward - the main interferometer optics and the MC optics.

4431   Wed Mar 23 10:34:17 2011 josephbUpdateCDSTrend issue fixed

[Joe, Alex]

Yesterday during the day, Alex ran a script to fix the time stamps in the trends files we had messed up back during the daqd change overs around Feb 17th and 23rd.  See this elog for more information on the trend problem.

Due to how the script runs, basically taking all the data and making a new copy with the correct time stamps, the data collected while the script was running didn't get converted over.  So when he did the final copy of the corrected data, it created a several hour gap in the data from yesterday during the day time.

The original files still exist on the fb machine in /frames/trend/minute_raw_22mar2011 directory.

4438   Thu Mar 24 13:56:05 2011 josephbUpdateelogelog restarted at 1:55pm

Restarted elog.

4444   Fri Mar 25 11:16:19 2011 josephbFrogsGreen Lockingdigital frequency counting

I modified the c1gfd.mdl simulink model.  I made a backup as c1gfd_20110325.mdl.

The first change was to use a top_names block to put everything in.  The block is labeled ALS.  So all the channels will now be C1:ALS-GFD_SOMETHING.  This means medm channel names will need to be updated.  Also, the filter modules need to be updated in foton because of this.

I then proceeded to add the suggested changes made by Matt.  To avoid a divide by zero case, I added a saturation part which saturates at 1e-9 (note this is positive) and 1e9.

 Quote: Today we tried the Schmitt trigger DFD, and while it works it does not improve the noise performance.  At least part of our problem is coming from the discrete nature of our DFD algorithm, so I would propose that an industrious day job person codes up a new DFD which avoids switching.  We can probably do this by mixing the input signal (after high-passing) with a time-delayed copy of itself... as we do now, but without the comparator.  This has the disadvantage of giving an amplitude dependent output, but since we are working in the digital land we can DIVIDE.  If we mix the signal with itself (without delay) to get a rectified version, and low-pass it a little, we can use this for normalization.  The net result should be something like: output = LP2[ s(t) * s(t - dt) / LP1[ s(t) * s(t) ]], where s(t) is the high-passed input and LP is a low-pass filter.  Remember not to divide by zero.

Attachment 1: C1GFD.png
4445   Mon Mar 28 15:18:04 2011 josephbUpdateCDSCDS updates on Friday

Last Friday, we discovered a bug in the RCG where the delay part was not actually delaying.  We reported this to Alex who promptly put a fix in the same day.  This allowed Matt's newly proposed frequency discriminator to work properly.

It also required a checkout of the latest RCG code (revision 2328), and rebuild of the various codes.  We backed up all the kernel and executables first such as mbuf.ko and awgtpman.

We did the following:

2) Go to /opt/rtcds/caltech/c1/core/advLigoRTS/src/drv/mbuf and run make.  Copy the newly built mbuf.ko file to /diskless/root/modules/2.6.34.1/kernel/drivers/mbuf/mbuf.ko on the fb machine.

3) Use "sudo cp" to copy the newly built mbuf.ko file to /diskless/root/modules/2.6.34.1/kernel/drivers/mbuf/

4) Go to /cvs/cds/rtcds/caltech/c1/core/advLigoRTS/src/gds and run make.

5) Copy the newly built awgtpman executable to /opt/rtcds/caltech/c1/target/gds/bin/

6) Go to /opt/rtcds/caltech/c1/core/advLigoRTS/src/mx_stream/ and run make.

7) Copy the newly built mx_stream executable to /opt/rtcds/caltech/c1/target/fb/

4446   Mon Mar 28 15:49:18 2011 josephbUpdateCDSLessons from LST

[Koji,Joe]

## PART 1:

Koji was unable to build his c1lst model first thing this morning.  Turns out there was  a bug with RCG parser that was introduced on Friday when we did the RCG updates.  We talked Alex who did a quick comment fix.  The diff is as follows:

Index: Parser3.pm ==============================

===================================== --- Parser3.pm  (revision 2328) +++ Parser3.pm  (working copy) @@ -1124,8 +1124,8 @@   print "Flattening the model\n";   flatten_nested_subsystems($root); print "Finished flattening the model\n"; - CDS::Tree::do_on_nodes($root, \&remove_tags, 0, $root); - print "Removed Tags\n"; + #CDS::Tree::do_on_nodes($root, \&remove_tags, 0, $root); + #print "Removed Tags\n"; #print "TREE\n"; #CDS::Tree::print_tree($root);   CDS::Tree::do_on_nodes($root, \&remove_busses, 0,$root);

This was some code to remove TAGs from the .mdl file for some reason which I do not understand at this time.  I will ask tommorrow in person so I can understand the full story.

## PART 2:

Koji then rebuilt and started the c1lst process.  This is his new test version of the LSC code.  We descovered (again) that when you activate too many DAQ channels (simply uncommenting them, not even recording them with activate=1 in the .ini file) that the frame builder crashes.  In addition, the c1lsc machine, which the code was running on, also hard crashed.

When a channel gets added to the .ini file (or uncommented) it is sent to the framebuilder, irregardless of whether its recorded or not by the frame builder.  There is only about 2 megabytes per second bandwidth per computer.  In this case we were trying to do something like 200 channels * 16384 Hz * 4 bytes = 13 megabytes per second.

The maximium number of 16384 channels is roughly 30, with little to no room for anything else.  In addition, test points use the same allocated memory structure, so that if you use up all the capacity with channels, you won't be able to use testpoints to that computer (or thats what Alex has led me to believe).

The daqd process then core dumped and was causing all sorts of martian network slowdowns.  At the same time, the c1lsc computer crashed hard, and all of the front end processes except for the IOP on c1sus crashed.

We rebooted c1lsc, and restarted the c1sus processes using the startc1SYS scripts.  However, the c1susfe.ko apparently got stuck in a wierd state.  We were completely unable to damp the optics and were in general ringing them up severely.  We tried debugging, including several burt restores and single path checks.

Eventually we decided to reboot the c1sus machine after a bit of debugging.  After doing a burt restore after the reboot, everything started to damp and work happily.  My best guess is the kernel module crashed in a bad way and remained in memory when we simply did the restart scripts.

4499   Thu Apr 7 13:14:23 2011 josephbUpdateCDSProposed plan for ITMX/ITMY control switch

## Problem:

The controls (fast and slow both) think ITMX is ITMY and ITMY is ITMX.

## Solution:

After some poking around today, I have convinced myself it is sufficient to simply swap all instances of ITMX for ITMY in the C1_SUS-AUX1_ITMX.db  file, and then rename it to C1_SUS-AUX1_ITMY.db (after having moved the original C1_SUS-AUX1_ITMY.db to a temporary holding file).

A similar process is then applied to the original C1_SUS-AUX1_ITMY.db file.  These files live in /cvs/cds/caltech/target/c1susaux.  This will fix all the slow controls.

To fix the fast controls, we'll modify the c1sus.mdl file located in /opt/rtcds/caltech/c1/core/advLigoRTS/src/epics/simLink/ so that the ITMX suspension name is changed to ITMY and vice versa.  We'll also need to clean up some of the labeling

At Kiwamu and Bryan's request, this will either be done tomorrow morning or on Monday.

So the steps in order are:

1) cd /cvs/cds/caltech/target/c1susaux

2) mv C1_SUS-AUX1_ITMX.db C1_SUS-AUX1_ITMX.db.20110408

3) mv C1_SUS-AUX1_ITMY.db C1_SUS-AUX1_ITMY.db.20110408

4) sed 's/ITMX/ITMY/g' C1_SUS-AUX1_ITMX.db.20110408 > C1_SUS-AUX1_ITMY.db

5) sed 's/ITMY/ITMX/g' C1_SUS-AUX1_ITMY.db.20110408 > C1_SUS-AUX1_ITMX.db

6) models

7) matlab

8) Modify c1sus model to swap ITMX and ITMY names while preserving wiring from ADCs/DACs/BO to and from those blocks.

9) code; make c1sus; make install-c1sus

10) Disable all watchdogs

11) Restart the c1susaux computer and the c1sus computer

4515   Tue Apr 12 12:01:30 2011 josephbUpdateGeneralIFO controls, now with 10% less lying (ITMX/ITMY controls swapped)

# The ITMX/ITMY control swap is complete.

The steps from this elog were followed.

In addition, I did a burt restore of c1sus, c1mcs.

I then swapped all the gain settings from ITMX to ITMY, and reenabled the watchdogs.

I did some basic kick tests (1000 counts into UL coil) and confirmed channels like C1:SUS-ITMX_ULPD_VAR (watchdogs mV readback) corresponded to the correct optic.  I also checked that the POS, PIT, YAW, SIDE produced reasonable damping when engaged.

4516   Tue Apr 12 16:01:33 2011 josephbUpdateGeneralRFM errors

## Problem:

Currently the c1scy, c1mcs, and c1rfm models are reporting an error with receiving some data sent over the GE Fanuc Reflected memory cards.

To be more exact, the C1:SUS-ETMY_ALS signal from the c1gcv FE code on the c1ioo computer going too the Y end is not being received However, the C1:SUS-ETMY_LSC signal is.  So the physical RFM card seems to be working.

Similarly, the TRY signal is being sent correctly from the Y end computer.  The X end is working fine and receiving both LSC and ALS signals.

The c1mcs and c1rfm models also receive data from the c1ioo computer and reporting receiving errors.

## Theory:

Because the RFM cards are transmitting and receiving at least some channels, I'm guessing there was changes made to the C1.ipc file, which defines the memory locations of these various channels on the RFM network, and that when a model was rebuilt, a different one using the previous IPC file was not, and thus one of the computer is going to the wrong place to either read or write data.

Tomorrow, I'm planning on the  following:

1) Clean out the C1.ipc file (/opt/rtcds/caltech/c1/chans/ipc/)

2) Rebuild all models

3) Run activate_daq.py script

4) Restart models via script

If this doesn't clear up the problem, I'll continue  to bug hunt.

4518   Wed Apr 13 11:34:07 2011 josephbUpdateCDSFixed IFO_ALIGN.adl

## Problem:

I switched the ITMX and ITMY control channels yesterday, but forgot to update the IFO_ALIGN.adl file (/opt/rtcds/caltech/c1/medm/c1ifo/) which had the control labels swapped to make life easier.

## Solution:

I swapped ITMX and ITMY control locations on the screen.

## Question:

Are there any other screens involving ITMX and ITMY that had controls reversed to make life easier?

4524   Thu Apr 14 12:57:15 2011 josephbUpdateCDSRFM network happy again

[Joe, Alex]

## Problem Symptoms:

There were red lights on the status screen indicating RFM errors for the c1scy, c1mcs and c1rfm processes.

The c1iscey, c1sus machines were receiving data sent over the RFM network from the c1ioo computer with a bad time stamp, a few cycles too late.  The c1iscex computer was receiving data from c1ioo fine.

## Problem:

The c1iscex RFM card had gotten into a bad state and was somehow slowing things down/corrupting data.  It didn't affect itself, but due to the loop topology was messing everyone else up.  Basically the only one who wasn't throwing an error was the culprit.

## Solution:

Hard power cycling the c1iscex computer reset the RFM card and fixed the problem.

4543   Tue Apr 19 15:48:43 2011 josephbUpdateCDSMEDM screens and Front Ends updated to new Matrices

## Problem:

The original matrix naming conventions for the front ends was broken.  It used _11, _12,...,_1e, _1f, _110, _111 and so forth.  The code was changes to use _1_1, _1_2,...,_1_16,_1_17, and so on.

In addition the matrix of filter banks was modified to use the same naming convetion (instead of starting at zero, it now start with one).

## Work Done:

I rebuilt all the models, and restarted them all.

I wrote a simple script to modify the burt restore files to have the correct names for all the stored matrix values.

I also modified all the suspension screens, by modifying the default screens in /opt/rtcds/caltech/c1/medm/master/

The C1SUS, C1SCX, C1SPX, C1SCY, C1SPY, and C1MCS models had their foton filter files modified to put filters into the newly changed named filters

4545   Wed Apr 20 11:02:18 2011 josephbUpdateCDSMEDM screens and Front Ends updated to new Matrices
We simply didn't any matrices larger than 16x16. If we had, than that matrix would not have worked properly since the beginning.

 Quote: Just a curiosity: I just wonder how you have distingushed the difference between _111 and _111. They are equivalent alone themselves. Have you looked at the contexts of the lines? Or you just did not have the larger matrix than 16x16, did you?

## What was done:

1) Turn off MC1, MC2, MC3, BS, ITMX, ITMY, PRM, SRM watchdogs.

2) Turn c1sus computer off (sudo shutdown now)

3) Go connect monitor and keyboard to c1sus.  Turn c1sus on.

4) Hit "del" key at the right time to go to setup (BIOS).

6) Hit F10 to save and let the computer continue booting.

## What went wrong:

Once c1sus was up, I noticed several red lights and dead keep alives for the c1sus models.

Typing dmesg on c1sus revealed many messages like:

[  107.583420] c1x02: cycle 33737 time 20; adcWait 10; write1 0; write2 0; longest write2 0 [  107.583771] c1x02: cycle 33760 time 19; adcWait 11; write1 0; write2 0; longest write2 0

This indicates the Input/Output Processor (IOP) is not completing its duties within the 15 microseconds (1/64 kHz) that it has.  These lines indicate its take 20 or 19 microseconds.  (I saw messages ranging from 16 to 22 microseconds).

So this seems to agree with Rolf's observations that hyperthreading can cause a 5-10 microsecond increase in computation time.

So the next thing to do is modify which core the codes are running on, and try to get them paired up on the same physical core.

First, I disabled front end starts on boot up, and brought c1sus up.  I rebuilt the models for the c1sus computer so they had a new specific_cpu numbers, making the assumption that 0-1 were one real core, 2-3 were another, etc.

Then I ran the startc1SYS scripts one by one to bring up the models.  Upon just loading the c1x02 on "core 2" (the IOP), I saw it fluctuate from about 5 to 12.  After bringing up c1sus on "core 3", I saw the IOP settle down to about 7 consistently.  Prior to hyper-threading it was generally 5.

Unfortunately, the c1sus model was between 60 and 70 microseconds, and was producing error messages a few times a second

[ 1052.876368] c1sus: cycle 14432 time 65; adcWait 0; write1 0; write2 0; longest write2 0 [ 1052.936698] c1sus: cycle 15421 time 74; adcWait 0; write1 0; write2 0; longest write2 0

Bringing up the rest of the models (c1mcs on 4, c1rfm on 5, and c1pem on 6), saw c1mcs occasionally jumping above the 60 microsecond line, perhaps once a minute.   It was generally hovering around 45 microseconds.  Prior to hyper-threading it was around 25-28 microseconds.

c1rfm was rock solid at 38, which it was prior to hyper-threading.  This is most likely due to the fact it has almost no calculation and only RFM reads slowing it down.

c1pem continued to use negligible time, 3 microseconds out of its 480.

I tried moving c1sus to core 8 from core 3, which seemed to bring it to the 58 to 65 microsecond range, with long cycles every few seconds.

I built 5 dummy models (dua on 7, dub on 9, duc on 10, dud on 11, due on 1) to ensure that each virtual core had a model on it, to see if it helped with stabilizing things.  The models were basically copies of the c1pem model.

Interestingly, c1mcs seemed to get somewhat better and only taking to 30-32 microseconds, although still not as good as its pre-hyper-threading 25-28.  Over the course of several minutes it was no longer having a long cycle.

c1sus got worse again, and was running long cycles 4-5 times a second.

At this point, without surgery on which models are controlling which optics (i.e. splitting the c1sus model up) I am not able to have hyper-threading on and have things working.  I am proceeding to revert the control models and c1sus computer to the hyper-threading state.

4608   Tue May 3 10:41:35 2011 josephbUpdateCDSMorning maintenance

1) Filled in the C1SUS_BS_OLMATRIX properly so as to make the BS oplev work for Steve.

2) Turned on the ITMX damping.  Apparently it had tripped this morning, possibly due to work in the lab area.

3) The ETMX FE controller (c1scx) had ADC timed out and died sometime around 8:30 am.  The c1x01 (the IOP on the ETMX computer) was also indicating a FB status error (mismatch in DAQ channels).

The reported error in dmesg on c1iscex was:

[1628690.250002] c1spx: ADC TIMEOUT 0 3541 21 3605
[1628690.250002] c1scx: ADC TIMEOUT 0 3541 21 3605

Just to be safe, I rebuilt the c1x01 and c1scx models, ran ./activateDAQ.py, and used the scripts killc1spx, killc1scx, and killc1x01.

I finally restarted the process with startc1x01, startc1scx, and startc1spx.  Everything is currently alive and indicating all green.

4609   Tue May 3 10:59:31 2011 josephbUpdateCDS1Y2 binary output adapter board now powered

I temporarily turned off the power to the 1Y2 rack this morning while wiring in the binary output adapter board power (+/- 15V) into the cross connects.

The board is now powered and we can proceed to testing if can actually control the LSC whitening filters.

4665   Mon May 9 13:14:48 2011 josephbUpdateLSCC1:LSC-TRIG_MTRX : wrong matrix size

[Joe, Kiwamu]

There is a feature/bug of the RCG code that you can only have 1 receiving tag for every sending tag.  There were 5 tags which were being received by two tags each, for two different matrices.  Only the first tag was receiving, the second was apparently ignored.

This has been fixed temporarily by putting in direct lines in place of these 5 tags.

 Quote: I found that C1:LSC-TRIG_MTRX has a wrong matrix size. It needs to be fixed. It is designed to have a 11x8 matrix in the simlink model file, but it's been compiled as a 6x8 matrix.

4666   Mon May 9 15:21:36 2011 josephbUpdatePSLFixed channel names for PSL QPDs, fixed saturation, changed signs

[Valera, Joe]

## Software Changes:

First we changed all the C1:IOO-QPD_*_* channels to C1:PSL-QPD_*_* channels in the /cvs/cds/caltech/target/c1iool0/c1ioo.db file, as well as the /opt/rtcds/caltech/c1/chans/daq/C0EDCU.ini file.  We then rebooted the frame builder via "telnet fb 8087" and then "shutdown".

This change breaks continuity for these channels prior to today.

The C1:PSL-QPD_POS_HOR and C1:PSL-QPD_POS_VERT channels were found to be backwards as well.  So we modified the /cvs/cds/caltech/target/c1iool0/c1ioo.db file to switch them.

Lastly, we changed the ASLO and AOFF values for the C1:PSL-QPD_POS_SUM and the C1:PSL-QPD_ANG_SUM so as to provide positive numbers.  This was done by flipping the sign for each entry.

ASLO went from 0.004883 to -0.004883, and AOFF when from -10 to 10 for both channels.

## Hardware Changes:

The C1:PSL-QPD_ANG_SUM channel had been saturated at -10V.  Valera reduced the power on the QPD to drop it to about 4V by placing an ND attenuator in the ANG QPD path.

4677   Tue May 10 10:06:23 2011 josephbUpdatePSLFixed channel names for PSL QPDs, fixed saturation, changed signs

I added calculation entries to the /cvs/cds/caltech/target/c1iool0/c1ioo.db file which are named C1:IOO-QPD_*_*, as the channels were originally named.  These calculation channels have the identical data to the C1:PSL-QPD_*_* channels.  I then added the channels to the C0EDCU.ini file, so as to once again have continuity for the channels, in addition to having the newer, better named channels.

The c1iool0 machine ("telnet c1iool0", "reboot") and the framebuilder process ("telnet fb 8087", "shutdown") were both restarted after these changes.

These channels were brought up in dataviewer and compared.  The approriate channels were identical.

Quote:

[Valera, Joe]

## Software Changes:

First we changed all the C1:IOO-QPD_*_* channels to C1:PSL-QPD_*_* channels in the /cvs/cds/caltech/target/c1iool0/c1ioo.db file, as well as the /opt/rtcds/caltech/c1/chans/daq/C0EDCU.ini file.  We then rebooted the frame builder via "telnet fb 8087" and then "shutdown".

This change breaks continuity for these channels prior to today.

4679   Tue May 10 16:42:49 2011 josephbUpdateCDSNew upconversion model (c1uct)

[Ryan, Joe]

Ryan added the c1uct (upconversion tester) model to the c1ioo machine.   It is DCU_ID 32, CPU 6.  The relevant wiki page has been updated. It has been added to /diskless/root/etc/rtsystab on the fb machine and automatically comes up when the c1ioo computer is turned on.

Joe still needs to add its status to the status screen.

/opt/rtcds/caltech/c1/userapps/trunk/CDS/c1/models/c1uct.mdl

Ryan will expand upon its uses later.

4680   Tue May 10 16:45:19 2011 josephbUpdateCDSc1ass now receiving AS55I from c1lsc

[Valera, Joe]

We added a cdsPCIx_SHMEM connection between the c1lsc and c1ass models.  This connection is called C1:LSC-ASS_AS55I, and sends the normalized AS55I data to Lockin 11 of the c1ass model.

In addition, in order to get the c1ass model to compile, we had to place all the non-IO parts inside a subsystem block, which we called ASS, and gave the top_names tag to.

The c1lsc and c1ass models were rebuilt, the frame builder restarted, and the models restarted.

4748   Thu May 19 12:09:41 2011 josephbUpdateCDSAA filter box pulled from 1X5, optic suspensions currently off

[Steve, Joe]

Steve pulled the top AA filter box from 1X5 which handled some of the suspensions channels.  We turned off all the watchdogs before pulling it out, as well as recorded which cables were connected to which inputs.

The case  is undergoing a structural modification to have the ADC adapter card which previously was loosely connected via cables, securely attached to the case.

Steve still wants to do some cabling in the rack while the box is out, and will return it this afternoon once he has finished that.

4770   Tue May 31 11:26:29 2011 josephbUpdateCDSCDS Maintenance

1) Checked in the changes I had made to the c1mcp.mdl model just before leaving for Elba.

2) The c1x01 and c1scx kernel modules had stopped running due to an ADC timeout.

According to dmesg on c1iscex, they died at 3426838 seconds after starting (which corresponds to ~39 days).  "uptime" indicates c1iscex was up for 46 days, 23 hours. So my guess is about 8 days ago (last Monday or Tuesday),  they both died when the ADCs failed to respond quick enough for an unknown reason.

I used the kill scripts (in /opt/rtcds/caltech/c1/scripts/) to kill c1spx, c1scx, and c1x01.  I then used the start scripts to start c1x01, then c1scx, and then finally c1spx.  They all came up fine.

Status screen is now all green.  I renabled damping on ETMX and it seems to be happy. A small kick of the optic shows the approriately damped response.

4776   Wed Jun 1 11:31:50 2011 josephbUpdateCDSMC1 LR digital reading close to zero, readback ~0.7 volts

There appears to be a bad cable connection somewhere on the LR sensor path for the MC1 optic.

The channel C1:SUS-MC1_LRPDMon is reading back 0.664 volts, but the digital sensor channel, C1:SUS-MC1_LRSEN_INMON, is reading about -16.  This should be closer to +1000 or so.

We've temporarily turned off the LRSEN filter module output while this is being looked into.

I briefly went out and checked the cables around the whitening and AA boards for the suspension sensors, but even after wiggling and making sure everything was plugged in solidly.  There was one semi-loose connection, but it wasn't on the MC1 board, but I pushed it all the way in anyways.  The monitor point on the AA board looks correct for the LR channels, although ITMX LR struck me as being very low at about -0.05 Volts.

According to data viewer, the MC1 LR sensor channel went bad roughly two weeks ago, around 00:40 on 5/18 UTC, or 17:40 on 5/17 PDT.

UPDATE:

It appears the AA board (or possibly the SCSI cable connected to it) is the problem in the chain.

4800   Thu Jun 9 16:18:03 2011 josephbUpdateCDSSecond trends only go back 12 days

While answering a quick question by Kiwamu, I noticed we only had second trends going back to 99050000 GPS time, May 27th 2011.

Trends (I thought) were intended to be kept forever, and certainly longer than full data, which currently goes back several months.

Jamie will need to look into this.

4918   Thu Jun 30 06:54:07 2011 josephbUpdateCDSModified the automated scripts for producing model webviews

Dave Barker pointed out last week that the webview of our simulink model files, generated from the installed models (i.e. in /opt/rtcds/caltech/c1/target/<system name>/simLink/) was not handling libraries properly.  Essentially the web pages generated couldn't see inside library parts.

This was caused by 2 problems.  The first being the userapps not being in the matlab path when the slwebview call was done, so it couldn't even find the libraries.  The second problem is the slwebview code by default doesn't follow libraries and links, and needs a special command to be told to do so.

addpath('/opt/rtcds/caltech/c1/core/trunk/src/epics/simLink/lib') for sub = {'cds','isc','isi','sus','psl'}  for spath = {'common/models','c1/models/lib'}    addpath(['/opt/rtcds/caltech/c1/userapps/release/' sub{1} '/' spath{1}]);  end end

I also changed the following:

temp = slwebview(final_files{x},'viewFile',false);

became

temp = slwebview(final_files{x},'viewFile',false,'FollowLinks','on','FollowModelReference','on');

After confirming these changes worked, I have sent a corrected version to Dave and Keith.

The webview results can be found at: https://nodus.ligo.caltech.edu:30889/FE/

6051   Wed Nov 30 11:04:26 2011 josephbUpdateCDSFiltering Noise issue tracked down ???

 Quote: For now, I suppose we can just change this number to 1e-40 or so. I don't know how to calculate what the right number should be. Not sure why this underflow is not an issue for the BiQuad, however.

According to the RCG SVN logs, the reason it was removed was a more general change done to the compiled code, not specific to just the biquad.  Basically, the ability to have an underflow number (subnormal) has been turned off completely by having any number that underflows set to zero. I'm not positive, but from a quick search looks that the smallest number before hitting is an underflow as a double is 2.2250738585072014e-308.

Alex's entry from the SVN log for 2663:

Added new fz_daz() function to turn on two bits in the FPU SSE control register.
Bits FZ (flush underflows to zero) and DOZ (denorms are zeros) are set to
avoid runaway code on float/double denorms (really small numbers).

SVN log 2664:

Removed +- 1e-20 limiting code, this is taken care of by setting FZ/DOZ bits
in the CPU SEE control register (see mathInline.h)

SVN log 2665:

Kill the underflows and roll down float denorms to zero,
see fz_doz() in mathInline.h.

12849   Thu Feb 23 15:48:43 2017 johannesUpdateComputersc1psl un-bootable

Using the PDA520 detector on the AS port I tried to get some better estimates for the round-trip loss in both arms. While setting up the measurement I noticed some strange output on the scope I'm using to measure the amount of reflected light.

The interferometer was aligned using the dither scripts for both arms. Then, ITMY was majorly misaligned in pitch AND yaw such that the PD reading did not change anymore. Thus, only light reflected from the XARM was incident of the AS PD. The scope was showing strange oscillations (Channel 2 is the AS PD signal):

For the measurement we compare the DC level of the reflection with the ETM aligned (and the arm locked) vs a misaligned ETM (only ITM reflection). This ringing could be observed in both states, and was qualitatively reproducible with the other arm. It did not show up in the MC or ARM transmission. I found that changing the pitch of the 'active' ITM (=of the arm under investigation) either way by just a couple of ticks made it go away and settle roughly at the lower bound of the oscillation:

In this configuration the PD output follows the mode cleaner transmission (Channel 3 in the screen caps) quite well, but we can't take the differential measurement like this, because it is impossible to align and lock the arm but them misalign the ITM. Moving the respective other ITM for potential secondary beams did not seem to have an obvious effect, although I do suspect a ghost/secondary beam to be the culprit for this. I moved the PDA520 on the optical table but didn't see a change in the ringing amplitude. I do need to check the PD reflection though.

Obviously it will be hard to determine the arm loss this way, but for now I used the averaging function of the scope to get rid of the ringing. What this gave me was:
(16 +/- 9) ppm losses in the x-arm and (-18+/-8) ppm losses in the y-arm

The negative loss obviously makes little sense, and even the x-arm number seems a little too low to be true. I strongly suspect the ringing is responsible and wanted to investigate this further today, but a problem with c1psl came up that shut down all work on this until it is fixed:

I found the PMC unlocked this morning and c1psl (amongst other slow machines) was unresponsive, so I power-cycled them. All except c1psl came back to normal operation. The PMC transmission, as recorded by c1psl,  shows that it has been down for several days:

Repeated attempts to reset and/or power-cycle it by Gautam and myself could not bring it back. The fail indicator LED of a single daughter card (the DOUT XVME-212) turns off after reboot, all others stay lit. The sysfail LED on the crate is also on, but according to elog 10015 this is 'normal'. I'm following up that post's elog tree to monitor the startup of c1psl through its system console via a serial connection to find out what is wrong.

12851   Thu Feb 23 19:44:48 2017 johannesUpdateComputersc1psl un-bootable

Yes, that was one of the things that I wanted to look into. One thing Gautam and I did that I didn't mention was to reconnect the SRM satellite box and move the optic around a bit, which didn't change anything. Once the c1psl problem is fixed we'll resume with that.

 Quote: The fringes seen on the oscope are mostly likely due to the interference from multiple light beams. If there are laser beams hitting mirrors which are moving, the resultant interference signal could be modulated at several Hertz, if, for example, one of the mirrors had its local damping disabled.

Speaking of which:

Using one of the grey RJ45 to D-Sub cables with an RS232 to USB adapter I was able to capture the startup log of c1psl (using the usb camera windows laptop). I also logged the startup of the "healthy" c1aux, both are attached. c1psl stalls at a point were c1aux starts testing for present vme modules and doesn't continue, however is not strictly hung up, as it still registers to the logger when external login attempts via telnet occur. The telnet client simply reports that the "shell is locked" and exits. It is possible that one of the daughter cards causes this. This seems to happen after iocInit is called by the startup script at /cvs/cds/caltech/target/c1psl/startup.cmd, as it never gets to the next item "coreRelease()". Gautam and I were trying to find out what happends inside iocInit, but it's not clear to us at this point from where it is even called. iocInit.c and compiled binaries exist in several places on the shared drive. However, all belong to R3.14.x epics releases, while the logfile states that the R3.12.2 epics core is used when iocInit is called.

Next we'll interrupt the autoboot procedure and try to work with the machine directly.

Attachment 1: slow_startup_logs.tar.gz
12852   Fri Feb 24 20:38:01 2017 johannesUpdateComputersc1psl boot-stall culprit identified

[Gautam, Johannes]

c1psl finally booted up again, PMC and IMC are locked.

Trying to identify the hickup from the source code was fruitless. However, since the PMCTRANSPD channel acqusition failure occured long before the actual slow machine crashed, and since the hickup in the boot seemed to indicate a problem with daughter module identification, we started removing the DIO and DAQ modules:

1. Started with the ones whose fail LED stayed lit during the boot process: the DIN (XVME-212) and the three DACs (VMIVME4113). No change.
2. Also removed the DOUT (XVME-220) and the two ADCs (VMIVME 3113A and VMIVME3123). It boots just fine and can be telnetted into!
3. Pushed the DIN and the DACs back in. Still boots.
4. Pushed only VMIVME3123 back in. Boot stalls again.
5. Removed VMIVME3123, pushed VMIVME 3113A back in. Boots successfully.
6. Left VMIVME3123 loose in the crate without electrical contact for now.
7. Proceeded to lock PMC and IMC

The particle counter channel should be working again.

• VMIVME3123 is a 16-Bit High-Throughput Analog Input Board, 16 Channels with Simultaneous Sample-and-Hold Inputs
• VMIVME3113A is a Scanning 12-Bit Analog-to-Digital Converter Module with 64 channels

/cvs/cds/caltech/target/c1psl/psl.db lists the following channels for VMIVME3123:

Channels currently in use (and therefore not available in the medm screens):

• C1:PSL-FSS_SLOW_MON
• C1:PSL-PMC_PMCERR
• C1:PSL-FSS_SLOWM
• C1:PSL-FSS_MIXERM
• C1:PSL-FSS_RMTEMP
• C1:PSL-PMC_PMCTRANSPD

Channels not currently in use (?):

• C1:PSL-FSS_MINCOMEAS
• C1:PSL-FSS_RCTRANSPD
• C1:PSL-126MOPA_126MON
• C1:PSL-126MOPA_AMPMON
• C1:PSL-FSS_TIDALINPUT
• C1:PSL-FSS_TIDALSET
• C1:PSL-FSS_RCTEMP
• C1:PSL-PPKTP_TEMP

There are plenty of channels available on the asynchronous ADC, so we could wire the relevant ones there if we done care about the 16 bit synchronous sampling (required for proper functionality?)

Alternatively, we could prioritize the Acromag upgrade on c1psl (DAQ would still be asynchronous, though). The PCBs are coming in next Monday and the front panels on Tuesday.

The (nameless?) Windows 7 laptop that lives near MC2 and is used for the USB microscope was used for interfacing with c1psl. No special drivers were necessary to use the USB to RS232 adapter, and the RJ45 end of the grey homemade DB9 to RJ45 cable was plugged into the top port which is labeled "console 1". I downloaded the program "CoolTerm" from http://freeware.the-meiers.org/#CoolTerm, which is a serial protocol emulator, and it worked out of the box with the adapter. The standard settings fine worked for communicating with c1psl, only a small modification was necessary: in Options>Terminal make sure that "Enter Key Emulation" is set from "CR+LF" to "CR", otherwise each time 'Enter' is pressed it is actually sent twice.

12854   Tue Feb 28 01:28:52 2017 johannesUpdateComputersc1psl un-bootable

It turned out the 'ringing' was caused by the respective other ETM still being aligned. For these reflection measurements both test masses of the other arm need to be misaligned. For the ETM it's sufficient to use the Misalign button in the medm screens, while the ITM has to be manually misaligned to move the reflected beam off the PD.

I did another round of armloss measurements today. I encountered some problems along the way

• Some time today (around 6pm) most of the front end models had crashed and needed to be restarted GV: actually it was only the models on c1lsc that had crashed. I noticed this on Friday too.
• ETMX keeps getting kicked up seemingly randomly. However, it settles fast into it's original position.

General Stuff:

• Oscilloscope should sample both MC power (from MC2 transmitted beam) and AS signal
• Channel data can only be loaded from the scope one channel at a time, so 'stop' scope acquisition and then grab the relevant channels individually
• Averaging needs to be restarted everytime the mirrors are moved triggering stop and run remotely via the http interface scripts does this.

Procedure:

1.     Run LSC Offsets
2.     With the PSL shutter closed measure scope channel dark offsets, then open shutter
3.     Align all four test masses with dithering to make sure the IFO alignment is in a known state
4.     Pick an arm to measure
5.     Turn the other arm's dither alignment off
6.     'Misalign' that arm's ETM using medm screen button
7.     Misalign that arm's ITM manually after disabling its OpLev servos looking at the AS port camera and make sure it doesn't hit the PD anymore.
8.     Disable dithering for primary arm
9.     Record MC and AS time series from (paused) scope
10.     Misalign primary ETM
11.     Repeat scope data recording

Each pair of readings gives the reflected power at the AS port normalized to the IMC stored power:

$\widehat{P}=\frac{P_{AS}-\overline{P}_{AS}^\mathrm{dark}}{P_{MC}-\overline{P}_{MC}^\mathrm{dark}}$

which is then averaged. The loss is calculated from the ratio of reflected power in the locked (L) vs misaligned (M) state from

$\mathcal{L}=\frac{T_1}{4\gamma}\left[1-\frac{\overline{\widehat{P}_L}}{\overline{\widehat{P}_M}} +T_1\right ]-T_2$

Acquiring data this way yielded P_L/P_M=1.00507 +/- 0.00087 for the X arm and P_L/P_M=1.00753 +/- 0.00095 for the Y arm. With $\gamma_x=0.832$ and $\gamma_x=0.875$ (from m1=0.179, m2=0.226 and 91.2% and 86.7% mode matching in X and Y arm, respectively) this yields round trip losses of:

$\mathcal{L}_X=21\pm4\,\mathrm{ppm}$  and  $\mathcal{L}_Y=13\pm4\,\mathrm{ppm}$, which is assuming a generalized 1% error in test mass transmissivities and modulation indices. As we discussed, this seems a little too good to be true, but at least the numbers are not negative.

12869   Mon Mar 6 12:34:30 2017 johannesSummaryASSASS light injection scenarios

What we want from the light source for the AS port light injection:

• Frequency control for locking and maintaining known offset from arm cavity resonances -> see below
• Fast extinguishing light in the IFO -> AOM first order switching

We have four possible laser sources that we can use for the injection of 1064 nm from the back:

• There are ~65 mW of IR power coming from the PSL doubling oven, of which ~2mW are used for the fiber beat box. The remaining light is currently dumped on the PSL table and would be available. It is picked off after the PMC and does not have any of the sidebands.
• There is a ~200 mW Lightwave NPRO on the PSL table that is currently unused.
• Koji said he has a ~500mW NPRO in the OMC lab that has no PZT actuation. I contacted a couple companies about fiber-coupled variable AOM frequency shifters that we can pair with this laser.
• I don't think using the high power beam of the PSL itself is a good idea, especially if we want to map the loss on the optics, because' we'll need it for the dither locking

I think for maximum flexibility it's best to fiber-couple whichever source we choose on the PSL table and then just collimate it out of a fiber on the AS table. This way if we want to add fiber-coupled modulators of any kind it's a plug-and-play modification.

Different frequency control schemes are:

• Modulate sidebands on the light and stabilize directly to the arm, using POX/Y or back-reflection at AS
• Free-space resonant EOM
• Free-space broadband EOM with Rich's resonant amplifier attachment
• Fiber-coupled EOM
• Offset phaselock:
• PSL IR: Transfer mode-cleaner stability
• Can lock arms while measurement in progress, but will have PSL IR light on PDs
• Green from the end;
• Broadly tunable laser frequency and no interference from IR.

Either way we'll need a few things:

• required for PDH locking, optional if we phaselock instead
• AOM
• We have free-space available, looking into fiber-coupled ones with frequency tuning
• Fast switching electronics
• Various fiber stuff
• We have enough to set up the fiber coupling of one light source. I'm starting with the 200 mW NPRO but this is technically interchangable.

I'm working on how to best set this up at the AS port and interfere with normal operation as little as possible. Ideally we use a Faraday just like for squeezed light injection, but this requires some modification of the layout, although nothing that involves mode-matching.

ELOG V3.1.3-