%Check for accidental resonances of HOM in the arms (maybe due to
%sidebands). At the moment there is only data for the y arm.
%Stuff one might change often
modeorder = 0:5; % Look for TEMmn modes where m,n run over modeorder
Kevin and I meaured the transfer function of the photodiode circuit using the Jenne laser and agilent in the 40m lab. The attached figures depict our measured transfer function over the modulation frequency ranges of 30kHz-30MHz and 1kHz-30MHz when the power of the laser was set to 69 and 95 μW. These plots indicate a clear roll off frequency around 300 kHz. In addition, the plots beginning at 1kHz display unstable behavior at frequencies below 30kHz. I am not sure why there is such a sharp change in the transfer function around 30kHz, but I suspect this to be due to an issue with the agilent or photodiode.
I've restarted the NDS2 process on Megatron so that we can use it for getting past data and eventually from outside the 40m.
1) from /home/controls/nds2 (which is not a good place for programs to run) I ran nds2-megatron/start-nds2
2) this is just a script that runs the binary from /usr/bin/ and then leaves a log file in ~/nds2/log/
3) I tested with DTT that I could access megatron:31200 and get data that way.
There is a script in usr/bin called nds2_nightly which seems to be the thing we should run by cron to get the channel list to get updated, but I' m not sure. Let's see if we can get an ELOG entry about how this works.
Then we want Jamie to allow some kind of tunneling so that the 40m data can be accessed from outside, etc.
I have done the following:
* installed the nds2-client in /ligo/apps/nds2-client
* moved the nds2 configuration directories to /ligo/apps/nds2/nds2-megatron
* set up a cron job to update the channel list every morning at 5 am. The cron line is:
15 5 * * * /usr/bin/nds2_nightly /ligo/apps/nds2/channel-tracker /ligo/apps/nds2/nds2-megatron
cron will send an email each time the channel list changes, at which point you will have to restart the server with:
* restarted nds2 with updated channel lists.
I have set the cron job up to restart the nds2 server automatically if the channel list changes. The only change is that the cron command was changes to /ligo/apps/nds2/nds2-megatron/test-restart.
The upgrade of megatron broke the nds2 service. I have fixed things by
1) installing the latest version of framecpp (1.19.32) from the lsc debian repository (this was necessary because I couldn't link to the existing version)
2) built nds2-server-0.5.11 and installed it in the system directories (/usr/bin)
3) there were a few scripts/links/etc that didn't seem to be set up correctly and I fixed them to correspond with my preious message.
nds2 is now running and the channel list should be updated regularly and the service restarted as appropriate.
Here's a Finesse modeling of what we're expecting to observe with this test. It uses Gautam's base model of the 40m IFO with appropriate modifications for the needed configuration.
The idea is to lock the IFO in the SRMI configuration, with the phase-locked AUX beam injected from the AS port. The AUX beam is imprinted with AM sidebands and slightly misaligned relative to the SRC so as to transfer power into HOM1. The RF network analyzer provides the drive signal for the AOM, and its frequency is swept to coherently measure the transfer function [reflected AUX beam / drive]. The reflected AUX beam is sensed by the AS110 PDA10CF.
It is also possible to drive PM sidebands as Koji suggests, but the squeezer group has encouraged using AM for practical advantages. The SNR with AM is a bit higher (less power lost into harmonics at large modulation index), there is a broadband AOM already available aligned to the SQZ beam at LLO, and there is also concern that driving strong PM could interfere with the SQZ control loops.
Attachment #1 shows the expected response to swept-AM in SRMI. Resolving just the FSR and the first-order mode splitting is sufficient to extract the SRC Gouy phase.
Since the 40m has not been opearted in SRMI since ~2016 (last done by Eric Q.), Gautam believes it may take some time to relock this configuration. However, the modeling indicates that we can likely obtain sufficient sensitivity in DRMI, which would allow us to proceed faster. Attachment #2 shows the expected response to swept-AM in DRMI. The PRC leakage signal turns out to be significantly smaller than the SRC reflection (a factor of ~30 in amplitude), so that the signal still retains its characteristic shape to a very good approximation. The tradeoff is a 10x reduction in SNR due to increased PSL shot noise reaching AS110.
Based on this, we should proceed with DRMI scans instead of PRMI next week.
The PRC FSR is, of course, very close to twice of our f1 moudlation frequency (11MHz x 2 = 22MHz) .
I still don't understand what response the measurement is looking for. I understood the idea of using the subcarrier as a stablized carrier to the PRC with a certain freq offset from the main carrier. I suppose what was swept was the AOM modulation frequency (i.e. modulation frequency of the AM applied to the subcarrier). If that is the case, the subcarrier seemed fixed at an arbitorary frequency (i.e. 50MHz) away from the carrier. If one of the AM sidebands hits the PRC resonance (i.e. 22, 44, 66MHz away from the main carrier), you still have the other sideband reflected back to the AS. Then the RF signal at the AS is still dominated by this reflected sideband. I feel that the phase modulation is rather suitable for this purpose.
If you are talking about ~MHz AM modulation by the AOM and scanning the PLL frequency from 1MHz to 60MHz, the story is different. And this should involve demodulation of the AS signal at the AM modulation frequency. But I still don't understand why we don't use phase modulation, which gives us the PDH type signal at the reflection (i.e. AS) port...
Not much progress today with the AUX cavity scans. I've determined there still are some alignment issues.
At the start of today a large AUX/PSL beat note was visible on the AS110 sensor, at a similar power as where we left off last night (-60 dBm). Proceeding from there, I attempted to reproduce Johannes' measurement of the cavity transmission resonances. I misaligned the X-arm, locked the Y-arm cavity, and scanned the AUX RF offset approximately 8 MHz in 2 kHz steps. This should have swept through two FSRs, but nothing was visible.
Further inspection revealed that none of the PSL light was making it back to through the AUX fiber to the PSL table. I take this to mean that the beam seen earlier on AS110 was the ITMY reflection, and that the AUX injection axis was no longer reaching ETMY. I also found that the AUX beam size just after the 90/10 beasmsplitter looks anomolously large. Maybe a lens was recently changed? In any case, the mode-matching looks like it is going to need to be readjusted.
[Jon, Keerthana, Sandrina]
Yesterday we carried out preliminary proof-of-concept measurements using the new AS-port-injected AUX laser to resolve cavity mode resonances.
At the time we started, I found the beat note levels consistent with what Johannes had reported the night before post-realignment. Hence we did not change the AUX alignment.
Test 1: YARM Mode Scan
Test 2: PRC Mode Scan
The SURFs have the data from last night's scans and will be separately posting plots of these measurements. We'll continue today with mode scans using AM sidebands rather than the AUX RF offset.
[Jon, Keerthana, Sandrine]
Thu.-Fri. we continued with PRC scans using the AUX laser, but now the "scanned" parameter is the frequency of AM sidebands, rather than the frequency of the AUX carrier itself. The switch to AM (or PM) allows us to coherently measure the cavity transfer as a function of modulation frequency.
In order to make a sentinel measurement, I installed a broadband PDA255 at an unused pickoff behind the first AUX steering mirror on the AS table. The sentinel PD measures the AM actually imprinted on the light going into the IFO, making our measurement independent of the AOM response. This technique removes not only the (non-flat) AOM transfer function, but also any non-linearities from, e.g., overdriving the AOM. The below photo shows the new PD (center) on the AS table.
With the sentinel PD installed, we proceeded as follows.
The below photo shows the measured transfer function [AUX Reflection / AUX Injection]. The measurement coherence is high to ~55 MHz (the AOM bandwidth is 60 MHz). We clearly resolve two FSRs, visible as Lorentzian dips at which more AUX power couples into the cavity. The SURFs have these data and will be separately posting figures for the measurements.
With the basic system working, we attempted to produce HOMs, first by partially occluding the injected AUX beam with a razor blade, then by placing a thin two-prong fork in the beam path. We also experimented with using a razor blade on the output to partially occlude the reflection beam just before the sensor. We were able to observe an apparent secondary dip indicative of an HOM a few times, as shown below, but could not repeat this deterministically. Besides not having fine control over the occlusion of the beams, there is also large few-Hz angular noise shaking the AS beam position. I suspect from moment to moment the HOM content is varying considerably due to the movement of the AS beam relative to the occluding object. I'm now thinking about more systematic ways to approach this.
Thanks to a discussion yesterday with Joe Betzweiser, I was able to identify and fix the remaining problem with the LLO GigE camera software. It is working now, currently only on rossa, but can be set up on all the machines. I've started a wiki page with documentation and usage instructions here:
This page is also linked from the main 40m wiki page under "Electronics."
This software has the ability to both stream live camera feeds and to record feeds as .avi files. It is described more on the wiki page.
In order to use the 0th-order deflection beam from the AOM for cavity mode scans, I've coaligned this beam to the existing mode-matching/launch optics set up for the 1st-order beam.
Instead of being dumped, the 0th-order beam is now steered by two 45-degree mirrors into the existing beam path. The second mirror is on a flip mount so that we can quickly switch between 0th-order/1st-order injections. None of the existing optics were touched, so the 1st-order beam alignment should still be undisturbed.
Currently the 0th-order beam is being injected into the IFO. After attenuating so as to not exceed 100 mW incident on the fiber, approximately 50 mW of power reaches the AS table. That coupling efficiency is similar to what we have with the 1st-order beam. With the Y-arm cavity locked and the AUX PLL locked at RF offset = 47.60 MHz (an Y-arm FSR), I observed a -50 dBm beat note at Y-end transmission.
I made the first successful AUX laser scan of a 40m cavity last night.
Attachment #1 shows the measured Y-end transmission signal w.r.t. the Agilent drive signal, which was used to sweep the AUX carrier frequency. This is a distinct approach from before, where the carrier was locked at a fixed offset from the PSL carrier and the frequency of AM sidebands was swept instead. This AUX carrier-only technique appears to be advantageous.
This 6-15 MHz scan resolves three FSR peaks (TEM00 resonances) and at least six other higher-order modes. The raw data are also enclosed (attachment #2). I'll leave it as an excercise for the SURFs to compute the Y-arm cavity Gouy phase.
# AG4395A Measurement - Timestamp: Jul 02 2018 - 18:55:04
#---------- Measurement Parameters ------------
# Start Frequency (Hz): 6000000.0, 6000000.0
# Stop Frequency (Hz): 15000000.0, 15000000.0
# Frequency Points: 801, 801
# Measurement Format: LOGM, PHAS
# Measuremed Input: AR, AR
#---------- Analyzer Settings ----------
# Number of Averages: 16
# Auto Bandwidth: Off, Off
More progress on the AUX-laser cavity scans.
Both data sets are attached.