Tonight was kind of a wash.
We spent some time retaking single arm scans with Gautam's frequency counting code to confirm the linewidths he measured before his most recent round of code improvements. During this, ETMX was being its old fussy self, costing us gentle realignment time. For the time being, we started actuating on ITMX for single arm locks. Also, out of superstition, I changed the static position offset that had been at +1k for the last N months to -1k.
ETMX broke us out of a few DRFPMI lock trials as well, as did poor SRM alignment. I finally set up dither alignement settings for SRM in DRMI though, which helped (even in the arms-held-off-resonance situation). I still prefer doing the PRM/BS dither alignment in a carrier PRMI lock, because I think the SNR should be better than DRMI.
We know that the ETMX excursions can happen without length drive exciting them, but also that length drives certainly can excite them. For future locks, I'm going to try out avoiding ETMX drive altogether; the sites use a single ETM for their DARM actuation and let the CARM loop take care of the resultant cross coupling, so hopefully we can do the same without angering the mode cleaner.
Anyways, we didn't really ever make it far enough to do anything interested with the DRFPMI tonight
The oplev and the LSC are off.
Over the course of my investigations into the systematic errors in the frequency readout using digital frequency counting, I noticed that my counter variable that keeps track of the number of clock cycles between successive zero crossings was NOT oscillating between 2 values as I would have expected (because of there being a +/- 1 clock cycle difference between successive zero crossings due to the fixed sampling time of 1/16384 seconds), but that there were occassional excursions to values that were +/- 3 clock cycles away. I then checked the output of the SCHMITTTRIGGER CDS library part (which I was using to provide some noise immunity), and noticed that it wasn't triggering on every zero crossing at higher frequencies. I tested this by hooking up a digital oscillator to the SCHMITTTRIGGER part, and looked at its output for different frequencies. The parameters used were as follows:
SCHMITTTRIGGER lower threshold: -1.0
SCHMITTTRIGGER upper threshold: +1.0
I am attaching plots for two frequencies, 3000Hz and 4628Hz (Attachments #1 and #2) . I would have expected a flip in the state of the output of the schmitt trigger between every pair of horizontal red lines in this plot, but at 4628 Hz, it looks like the schmitt trigger isn't catching some of the zero crossings? Come to think of it, I am not even sure why the output of the schmitt trigger takes on any values other than 0 or 1 (could this be an artefact of some sort of interpolation in the visualization of these plots? But this would not affect the conclusion about the schmitt trigger missing some of the zero crossings?)
As an interim measure, I implemented a Schmitt trigger in my C code block - it was just a couple of extra lines anyways - I have designated the schmitt trigger output as a static variable that should hold its value in successive execution cycles, unless it is updated by comparing the input value to the thresholds (code attached for reference). Attachments #3 and #4 show the output of this implementation of a Schmitt trigger at the same two frequencies, and I am seeing the expected flip in the state between successive zero crossings as expected (though I'm still not sure why it takes on values other than 0 and 1?). Anyways this warrants further investigation. An elog regarding the implications of this on the systematic error in the frequency counter readout to follow.
The length of DRFPMI lock did not increase much tonight, but we got a ~80 second sensing matrix measurement, and got the CARM bandwidth up to 10k with two boosts on.
NB: I did not measure the CARM loop gain at its excitation frequency, so the plotted sensing element is supressed by the CARM loop. However, this is still useful for gauging the size of the PRCL signal vs. the residual CARM fluctuations. The excitations are fairly closely spaced between 309 and 316 Hz.
For comparison, I'm also re-plotting the DRMI sensing measurement from a few weeks back taken at CARM offset of -4. We can see some change in the PRCL sensing, likely due to the CARM-coupled path. MICH/PRCL sadly looks pretty degenerate, but REFL55 looks more reasonable.
I think the main limitation tonight was SRC stability. Even before bringing CARM to zero offset, we would see occasional sharp dives in AS110 power. One lockloss happened soon after such an occurance, but I checked the values, and it was not sufficient to trigger the Schmitt trigger down; instead it may have been a real optical loss of signal. The SRCL OLTF looks sensible.
I've installed new filters for the T240 -> PRM static online angular feedforward that were trained after some of the recent changes to the signal chain of the relevant signals (i.e. the counts->velocity calibration that Rana did for the seismometers, and fixing the improper dewhitening of the POP QPD channels used as the Wiener target.)
Quickly trying them out now shows about the same level of performance as the previous ones, but the real performance I care about is during after-hours locking-time, so I'll take more measurements tonight to be posted here.
MC WFS offsets were updated to have a better operating point.
CC1 and CC2 are working again. Why did they start working again ?
A handful of DRFPMI locks tonight, longest one was ~7 minutes.
EPICS/network latency has been a huge pain tonight. The locking script may hang between commands at an unstable place, or fail to execute commands altogether because it can't find the EPICS channel. This prevented or broke a number of locks.
I made some CARM OLG and crossover measurements, and found the AO gain for the right crossover freq (~100Hz) to be ~8dB different than what's in the PRFPMI script, which is weird. Right now, the CARM bandwidth / ability to turn on boosts is limited by the gain peaking in the IMC CLG due to the high-ish PC/PZT crossover frequency we're using.
Gautam turned on some sensing excitations during the last couple of locks, but they weren't on for very long before the lock loss. Hopefully I can pull out at least some angles from the data.
I'm also more convinced that the PRC angular FF needs retuning; there is more residual motion on the cameras than I'm used to seeing. I've taken more data that I'll use to recalculate a wiener filter tomorrow.
The PMC, ALSX beat and ITMX oplev all needed a reasonable pitch realignment tonight.
D'oh. Good point.
Reverted for now; I'm thinking about doing laser pointer->MC2 QPD...
I began my attempts to characterize the PDH loops at the X end today. My goal was to make the following measurements:
which I can then put into my simulink noise-budget scheme for the proposed IR beat setup.
I've made an Optickle model of a simple FP cavity and intend to match the measured PDH error signal from the X end to the simulated error signal to get the Hz/V calibration. I'll put the plots up for these shortly.
With regards to the other measurements, I was slowed down by remote data-acquisition from the SR785 - I've only managed to collect the analyzer noise floor data, and I plan to continue these measurements during the day tomorrow.
I found (an old) 10 dB coupler in the RF component shelves near MC2 - it has BNC connectors and not SMA connectors, but I thought it would be worth it to switch out the 20dB coupler currently on the X green beat PD on the PSL table with it. I used some BNC to SMA adaptors for this purpose. It appears that the coupler works, because I am now able to register an input signal on the X arm channel of the digital frequency counter (i.e. the coupled output from the green beat PD). I thought it may be useful to have this in place and do an IR transmissions arm scan using ALS for the X arm as well, in order to compare the results with those discussed here. However, the beat note amplitude on the analyzer in the control room looks noticeably lower - I am not sure if the coupler is responsible for this or if it has to do with the problems we have been having with the X end laser (the green transmission doesn't look glitchy on striptool though, and the transmission itself is ~0.45). In any case, we could always remove the coupler if this is hindering locking efforts tonight.
As per Eric's request, here is the code and TF measurement that was used to calculate the MC2 FF filter that is loaded in FM5. This filter module has the filter with the best subtraction performance that was achieved for MCL.
Sorry for the confusion - I did mean Green beat frequency, and I had neglected the factor of 2 in my earlier calculations. However, the fit parameter "c" in my fit was actually the half-width at half maximum and not the full width at half maximum. After correcting for both these errors (new fit is Attachment #1, where I have now accounted for the factor of 2, and the X axis is the IR beat frequency), I don't think the numbers change too much. It could be that the frequency counter wasn't reading out the frequency correctly, but looking at a time series plot of the frequency counter readout (Attachment #2), and my earlier trials, I don't think this is the case (38 MHz is a frequency at which I don't expect much systematic error - also, the offset was stepped from -3 to 3 over 45 seconds).
The revised numbers:
Fitted linewidth = 2*c = 10.884 kHz +/- 2 Hz (95% C.I.)
FSR of Y arm (from elog 9804) = 3.9468 MHz +/- 1.1 kHz
=> Y arm Finesse = FSR/fitted linewidth = 362.6 +/- 0.5
Total round trip loss = 2*pi/Finesse = 0.0173
The OSEMs cannot be used for coil balancing above ~10 Hz. The main coupling path from OSEM drive to sensor is not through the mirror motion, but instead direct electrical coupling of the drive wires to the sensing wires.
I put this in the elog every ~1-2 years since people keep trying it, but it keeps coming back like a zombie.
Better to use the MC angular sensors for L2A decoupling. Not perfect, but better than OSEMs. For the TMs we can use the OLs.
It is a nice scan. I'm still thinking about the equivalence of the moving average and the FIR low pass that I have mentioned in the meeting.
I'm confused by the plot. The bottom axis says "green beat frequency".
If you scan the IR laser frequency by df, you get 2*df shift of the green beatnote. You need to have this factor of two somewhere.
If you are looking at the IR beat freq, just the label is not correct. (I believe this is the case.)
Accepting the rather-too-low finesse of 363 (nominally 450), the total round trip loss is said to be 0.0173.
If we subtract the front transmission of 1.38% (and ignoring the transmission loss from the ETM),
the round trip loss is 3500ppm. Is this compatible with the following elog?
In fact, I'm afraid that the loss number in the above elog 11111 was not correct by a factor of 10.
Then, if so, can we believe this high loss number? (Nominally we expect ~100ppm loss per round trip...)
I've made quite a few changes in the software as well as the hardware of the digital frequency counting setup.
Eric helped me test the new setup by doing an arm scan through an IR resonance by ramping the ALS offset from -3 to +3 with a ramp time of 45 seconds. The data was acquired with the window size of the moving average set to 4096 clock cycles, and a 2 Hz low pass IIR filter before the frequency readout. Attachment 1 shows a plot of the data, and a fit with a function of the form trans = a/(1+((x-b)/c)^2), where a = normalization, b = center of lorentzian, and c = linewidth (FWHM) of the peak (the fitted parameter values, along with 95% confidence bounds are also quoted on the plot). In terms of the data acquisition, comparing this dataset to one from an earlier scan Eric did (elog11111) suggests that the frequency counting setup is working reasonably well - at any rate, I think the data is a lot cleaner than before implementing the moving average and having a 20Hz lowpass IIR filter. In any case, we plan to repeat this measurement sometime next week during a nighttime locking session. It remains to calculate the arm loss from these numbers analogous to what was done earlier for the X arm.
Calculation of loss:
Fitted linewidth = 10.884 kHz +/- 11Hz (95% C.I.)
Clean comparable scan at vacuum normal. There was no backstreaming.
Convectron gauge check:
Brand new 10 years old convectron gauge at atm was swapped into the place of existing gauges to see if they read close to 760 Torrs
They did reasonable well at the low end. I tried to imitate calibration and bring down the high end with little success.
( P1 and P2 were reading 7e-4 to ~660 Torr The correction of the upper end pushed up the lower end too. I will correct this later
P3 and P4 high ends are way off )
The point is that they work. Convectron gauges will be replaced and calibrated at the next vent.
Interlocks were not triggered during this test. I was expected to close the PSL shutter when P1 was reading 760 Torr
This hide some problem or not understanding.
It was good to see CC1 and CC4 working at the moment
Somehow I succeded opening VM1 and closed VM2 = vacuum normal
I just hope it stays open overnight to get comparable RGA scan.
[ericq, Gautam, Steve]
Following roughly the same procedure as ELOG 11354, c1vac1 and c1vac2 were rebooted. The symptoms were identical to the situation in that ELOG; c1vac1 could be pinged and telneted to, but c1vac2 was totally unresponsive.
The only change in the linked procedure was that we did not shut down the maglev. Since I unwittingly had it running for days without V4 open while Steve was away, we now know that it can handle shorter periods of time than that...
Upon reboot, many channels were readable again, unfortunately the channels for TP2 and TP3 are still blank. We were able to return to "Vacuum normal state," but because of unknowned communication problems with VM1's interlock, we can't open VM1 for the RGA. Instead we opened VM2 to expose the RGA to the main IFO volumn, but this isn't part of the "Normal" state definite, so things currently read "Undefined state".
1, Pressure gauges had no communication ( NO COMM ) with c1vac2
2, Lost N2 supply on Oct 9 This triggered a normal all valve closed condition. At this point you replace N2 cylinders and manually swich valves to recreate VAC NORMAL configuration in the correct sequential order.
The very last thing you do is open V1 gate valve.
a, check TP2 that is the forepump of the Maglev. Foreline pressure to drypump ~ 10- 100 mTorr, rotation speed 50 Krpm
b, open V4 if P2 <1Torr
c, check Maglev rotating at 560 Hz
d, open V1 if P1 <500 mTorr
e, check TP3 foreline, rotation speed and open V5 if P3 <1 Torr with VA6 closed
f, open VA6 if PAN <1 Torr
g, open annulos valves one by one , like VASE if PASE <1 Torr and so on...........Now the Current State: should read Vac Normal
3, Maglev run for 7 days with V4 closed. This encreased its foreline pressure to estimated few Torrs and its body temp rose ~30C on the outside.
So it was sweating and it may be back streamed.
The present RGA data is indicating that it had to be very mild.
The RGA will have better sensitivity with VM1 open and VM2 closed.
The PSL output shutter stayed open during these period is pointing to IFO pressure stayed P1 <3 mTorr
PROBLEM: P1 and P2 plot should show nothing where there is no communication.http://nodus.ligo.caltech.edu:8080/40m/151016_182003/oct16Fpm2015.png
How do we check if pressure based software interlocks are working in this no communication condition?
I'm working on setting up a moving-average in the custom C code block that counts the zero crossings to see if this approach is able to mitigate the glitchy frequency readout due to mis-counting by one clock cycle between successive zero crossings. I'm storing an array the size of the moving average window of frequency readouts at each clock cycle, and then taking the arithmetic mean over the window. By keeping a summing variable that updates itself each clock cycle, the actual moving average process isnt very intensive in terms of computational time. The array does take up some memory, but even if I perform the moving average over 1 second with 16384 double precision numbers stored in the array, its still only 130 kB so I don't think it is a concern. Some tests I've been doing while tuning the code suggest that with a moving average over 16384 samples (i.e. 1 second), I can eliminate glitches at the 1Hz level in the frequency readout for frequencies up to 5 kHz (generated digitally using an oscillator block). Some tuning still needs to be done, and the window could possibly be shortened. I also need to take a look at the systematic errors in this revised counting scheme, preferably with an analog source, but this is overall, I think, an improvement.
On a side note, I noticed some strange behaviour while running the cds average command - even though my signal had zero fluctuations, using z avg 10 -s C1:TST-FC_FREQUENCY_OUT gave me a standard deviation of ~1 kHz. I'm not sure what the problem is here, but all the calibration data I took in earlier trials were obtained using this so it would be useful to perform the calibration again.
Using a modified version of Hang's deMod_deCoup scripts, I tuned the MC2 coil output matrix to minimize the appearance of POS drive in the SUSPIT signal at 28Hz. Up until now, there was no F2P compensation. This reduced the force to pitch coupling at 28Hz by 8dB.
Old: POS -> 1 x UL, 1 x UR, 1 x LL, 1x LR
New: POS -> 1.1054 x UL, 1.1054 x UR, 0.8946 x LL, 0.8946 LR
I checked the MCL spectrum before and after this change with OAF on, this did not spoil the feedforward length subtraction in any noticible way.
The script lives in userapps/release/isc/c1/scripts/decoup, but I've symlinked it to /opt/rtcds/caltech/c1/scripts/decoup.
The script modification I made had to do with how the I and Q data is collected. Before, it was sporadically probing the I and Q FM output monitor EPICS values; I changed it to use the avg function of cdsutils, which calculates the mean and std from the 16kHz data and have seen it improve results by 1dB or so. I've been in touch with Jenne to propagate this to the sites.
For CARM and DARM, the A channels are used for the ALS signals, whereas the B channels are used for blending the RF signals.
For the DRMI, the A channels are used for the 1F signals, whereas the B channels are used for the 3F signals. The settings for transitioning to 1F after locking the DRFPMI have not yet been determined.
These settings are currently saved in the DRMI configurator, but the demod angles are set for DRFPMI lock, so the settings don't reliably work for misaligned arms.
The REFL33 element in SRCL_B is to reduce the PRCL coupling, was found empirically by tuning the relative gains with the arms misaligned and looking at excitation line heights. The offsets were found by locking the DRMI on 1F signals with arms misaligned, and taking the average value of these 3F error signals.
The CARM and DARM ALS settings are largely scripted by scripts/ALS/Transition_IR_ALS.py, which takes you from arms POX/POY locked to CARM and DARM ALS locked. The DRMI settings are usually restored from the IFO_CONFIGURE screen.
When arms are POX/POY locked, and the green beatnotes are appropriately configured, calling scripts/DRFPMI/carm_cm_up.sh initiates the following sequence of events:
When CARM and DARM are buzzing around true zero, powers maximized:
This is as far as we've taken the DRFPMI so far, but the CARM bandwidth is still only at a few kHz. Based on PRFPMI locking, the next steps will be:
Last Friday, I installed some RF couplers on the green BBPDs' outputs, and sent them over to Gautam's frequency divider module. At first I tried 20dB couplers, but it seemed like not enough power was reaching the dividers to produce a good output. I could only find one 10dB coupler, and I stuck that on the X BBPD. With that, I could see some real signals come into the digital system.
I don't think it should be a problem to leave the couplers there during other activities.
Gautam was working on his digital frequency counter stuff when the c1als model crashed. I had trouble bringing it back until I realized that, for reasons unknown to me, the safe.snap file that the model looks for at boot had been deleted. (This file lives in /target/c1als/c1alsepics/burt). I copied over this morning's version from Chiara's local backup.
At the sites, these files are under version control in the userapps svn repository, presumably symlinked into the target directory. We should definitely do something along these lines.
Here is a longer lock, about 100 seconds RF only, from later that same night. The in-loop CARM and DARM error signals have the order of magnitude of 1nm per count.
From ~-150 to -103, we were fine tuning the ALS offsets to try and get close to the real CARM/DARM zero points then blending the RF CARM signal.
At -100, the CARM bandwidth increases to a few kHz and stabilizes the arm powers. By -81, the error signals are all RF. At -70, I turned on the transmon QPD servos, which brought the power up a bit.
If I recall correctly, lock was lost because I put waaaay too big of an excitation on DARM with the goal of running its UGF servo for a bit. The number I entered was appropriate for ALS, but most certainly too huge for AS55...
Steve pointed out that in the aftermath of the Nitrogen running out a couple of times last week, the RGA had shut itself off thinking that there was a leak and so it was not performing the scheduled scans once a day. So the data files from the scheduled scans were empty in the /opt/rtcds/caltech/c1/scripts/RGA/logs directory. The wiki page for getting it up and running again is up-to-date, but the script RGAset.py did not exist on the c0rga machine, which the RGA is communicating with via serial port. I copied over the script RGAset.py from rossa to c0rga and ran the script on that machine - but the error flags it returned were not all 0 (indicating some error according to the manual) - so I edited the script to send just the initialize command ('IN0') and commented out the other commands, after which I got error flags which were all 0. After this, I ran a manual scan using 'RGAlogger.py', and it appears that the RGA is now able to take scans again - I'm attaching a plot of the scan results. We've saved this scan as a reference to compare against after a few days.
in addition to Koji's words I feel like we should also thank those who made small but positive contributions. Its hard not to notice that this locking only happened after the new StripTool PEM colors were implemented...
From the times series plot I guess that the fuzz of the in-loop DARM is 1 pm RMS (based on memory). This means that the ALS was holding the DARM at 10 pm from the RF resonances.
There is no significant shift in the DRMI error signals, so new weird CARM effect. Would be interesting to see what the 1f signals do in the last 60 seconds before RF lock.
For documentation, perhaps Gautam can post the loop gain measurements of the 5 loops on top of the Bode plots of the loop models.
Vacuum normal configuration with VM1-closed, VM2-open valve positions. Power load normal 24V 0.2A
Maglev rotation 560 Hz at room temp body temp.
"NO COMM" error message on medm screen. Gauge controller pressures are read able.
All vac comp LEDs are green. We have to reboot on Monday to enable communication.
None of the links here seem to work. I forgot what the story is with our special apache redirect
The story is: we currently don't expose the whole /users/public_html folder. Instead, we are symlinking the folders from public_html to /export/home/ on nodus, which is where apache looks for things
So, I fixed the links on the Core Optics page by running:
controls@nodus|~ > ln -sfn /users/public_html/40m_phasemap /export/home/
Fast ALS was still a problem tonight. I don't think high frequency ALS noise saturating the PC drive is the issue; I put two 10k poles before the CM board (shooting for just 2-3kHz bandwidth), and the PC drive levels would be stable and low up until the lockloss, which was always conincident with a step in the AO gain.
After working with that for a few hours, we turned back to our more standard locking attempts. First, we dither aligned the PRMI, and then centered the REFL beam on REFL11. It's hard to say for certain, but we may have been a little close to the edge of the PD. The only other thing that differed from Monday's attempts was using 6dB less AO gain when trying the up the overall gain.
The script now reliably breaks through to stable high powers, we had a handful of pure-RF locks tonight. The digital DARM gain needs tuning, and the CARM bandwidth still isn't at its final state, but these are very tractable. Off the top of my head, the way forward now includes:
Unrelated: I feel that the PRC angular FF may have deteriorated a bit. I'm leaving the PRC locked on carrier to collect data for wiener filter recalculation.
For real this time.
I carried out some further diagnostics and found some ways in which I could optimize the zero-crossing-counting algorithm, such that the error in the measured frequency is now entirely within the expected range (due to a +-1 clock cycle error in the counting). We can now determine frequencies up to ~60 MHz with less than 1 MHz systematic error and <10 kHz statistical error (fluctuations after the 20 Hz lowpass). This should be sufficient for slow control of the end-laser temperatures.
The conclusion from my earleir tests was that there was possibly an improvement that could be made to setting the thresholds for the Schmitt trigger stage in the model. In order to investigate this, I wanted to have a look at the 64K sampled raw input to the ADCs. Yesterday Eric helped me edit the appropriate .par file for viewing these channels for c1x03, and for an input frequency of 70MHz (after division, ~4.3 kHz square wave), the signal looked as expected (top left plot, attachment #1). This prompted me to check the counting algorithm again with the help of various test points I had setup within the model. I found that there was a tendency to under-count the number of clock-cycles between zero-crossings by more than 1 clock cycle, due to the way my code was organized. I fixed this and found that the performance improved dramatically, compared to my previous trials. With the revised counting algortihm, there was at most a +-1 clock cycle error in the counting, and the systematic error between the measured and requested RF frequencies is now completely accounted for taking this consideration into account. The origin of this residual error can be understood by looking at the top right plot in Attachment #1 - presumably because of the effects of some downsampling filter, the input signal to the Schmitt trigger isnt a clean square wave (even at 4kHz) - specifically, the time spent in the LOW and HIGH states of the Schmitt trigger can vary between successive zero crossings because of the shape of the input waveform. As a result, there can be a +-1 clock cycle error in the counting process. Attachment #2 shows this - the red and blue lines envelope the measured frequency for the whole range investigated: 10-70MHz. Attachment #3 shows the systematic error as a function of the requested frequency.
If there was some way to bypass the downsampling filter, perhaps the high-frequency performance could be improved a little.
Eric needed a buffer to drive low input impedance (~130Ohm) of his pomona box, I quickly made a non-iverting buffer with G=+10. The DC power is obtained from the back of SR560. It uses 1.02K and 9.09K
to have the gain of ~10. The chip is OP27. In fact this limits the output swing to be +/-5V for the load resistance of 130Ohm. Eric thinks this is enough. If we need more, we need to swap the chip.
As SR560s tend to saturate too quickly, it would be very useful to have this kind of kit in all the labs
once it is packed in a box.
After some discussion at last week's 40m meeting, I increased the frequency of daqd trying to write out minute trends from hourly to every two hours.
This has eliminated the hourly crashes. daqd still crashes sometimes, but only a few times per day.
However, looking at the oplev summary pages that actually use the minute trends, it looks like they're only sporadically getting succesfully written out.
Also, I was having a lot of problems with the frontends' EPICS processes dying when I would try to update the SDF table. I rebuilt all of the frontends with RCG 2.9.6, which differs from the 2.9.4 that we had been running by SDF bugfixes and an RMS calculation bugfix. The SDF procedures are much more stable now.
I have not yet discovered anything broken by this chage, and the tests I made for the last upgrade were all fine; last weeks tiny DRFPMI lock was achieved after this change.
I've made a cascaded passive 2-pole pomona box for fast ALS use, using LISO to check that it'll give the right shape when hooked up to the CM board's input stage.
First stage is a 133Ohm + 10uF cap for ~120Hz LP, second is 1.15kOhm + 47nF cap for ~3.8kHz LP. The DC gain is ~0.75, which is much better than what I was doing before. The second stage would normally make a 2.9kHz LPF on its own, but the loading of the input stage moves the corner up.
It seems the 133 Ohm resistor is a reasonable load on the output AD829 of the ALS demod board (short-circuit output current of 32mA and a series output resistor of 499Ohm). To be able to use the digitized ALSX I and the lowpassed analog version simultaneously, I had to buffer the signal with a SR560 before the pomona box, otherwise the signals looked distorted. This isn't a good long-term solution. Maybe I can used the further-buffered differential output to drive the LPF+CM board.
The LISO files used to model the filter and CM board input stage, and fit the pole frequencies are attached.
I made some attempts to get the AO path going today, but I suspect this daytime noise is just too much; the PC drive seems too irritable
Despite our best efforts, the grappa remains out of reach: the DRFPMI was not locked tonight.
We spent a fair amount of time with the AUX X laser, as it was glitching madly again.
DRMI was finicky until I found some more reliable triggering settings; namely aquiring with AS110Q, but after that transitioning the trigger to the same POP22+POPDC combo as PRCL and MICH. With this in place, the DRMI lock seems really indefinite no matter what CARM seems to do; or at least, I always lost lock due to CARM shenanigans after this.
The most frustrating part was the fact that I just couldn't cross over the AO path stably. It never "clicked" into high circulating power as it normally does (either in PRFPMI, or how it was last week). Various crossover filters and tweaks were attempted to no avail. Morning traffic starts soon, so we're calling it a night.
I carried out some more tests on the digital frequency counting system today, mainly to see if the actual performance mirrors the expected systematic errors I had calculated here.
Setup and measurement details:
I used the Fluke 6061A RF signal generator to output an RF signal at various frequencies, one at a time, between 10 and 70 MHz. I split the signal (at -15 dBm) into two parts, one for the X-channel and one for the Y-channel using a mini-circuits splitter. I then looked at the input signal using testpoints I had set up within the model, to decide what thresholds to set for the Scmitt trigger. Finally, I averaged the outputs of the X and Y channels using z avg -s 10 C1:ALS-FC_X_FREQUENCY_OUT and also looked at the standard deviation as a measure of the fluctuations in the output (these averages were taken after a low-pass filter stage with two poles at 20Hz, chosen arbitrarily).
The new ZKL-1R5 RF amplifier that Steve ordered arrived yesterday. I installed this in the frequency divider box and did a quick check using the Fluke RF signal generator and an oscilloscope to verify that both the X and Y paths were working.
I've now installed the box in the 1X2 rack where the olf "RF amplifiers for ALS and FOL" box used to sit (I swapped that out as I needed the L brackets on that chassis to mount mine, see Attachment #1 for the new layout). The power cable that used to power the old chassis was available, but the connector was of the wrong gender, so I had to switch this out. After verifying that I was getting the correct voltage (+15V), I connected it to the chassis.
I then did a quick check with the Fluke generator to make sure that all was working as expected - Eric had set up some ADC channels for me earlier today in the C1ALS model, and I copied over my frequency counting module from C1TST into C1ALS, and recompiled the model. The RF generator was set to generate a 25MHz signal at -20dBm, which I then split using an RF power splitter between the X and Y arms. I then checked the output using dataviewer - I recovered an output frequency of ~27.64 MHz with a jitter of ~0.02 MHz with a 20Hz low-pass filter in place (see Attachment #2), which looks consistent with the systematic error inherent in the zero-crossing counting algorithm and random fluctuations I had observed in my earlier trials, discussed here. But a more systematic investigation needs to be carried out in this regard. The interfacing between the hardware and software seems to be working alright though. I've left the RF generator near the 1x2 rack for now, though its powered off.
The mode cleaner unlocked quite a few times while I was working but looks stable now.
A few minutes ago, Gautam and I were poking around the IOO rack, looking at where he should power his frequency divider box, and what ADC innputs to use.
Looking at the mode cleaner signals, it looks like we may have jostled something in a good way. Weird.
Ah, I understand it now! Since the additive offset path keeps the post-cavity frequency TF flat, the pre-cavity frequency must grow above the cavity pole, which is why ALS sees a zero.
Ok, so this means we want to apply two lowpasses to the ALS signal for use as fast CARM control, if we want it to be capable of scalar blending with REFL11: one at ~120Hz to imitate the CARM coupled cavity pole present in REFL11, and one at ~3.8kHz to undo the "IMC cavity zero" present in ALS.
At this point, I'm starting to prefer an active circuit to do this lowpassing; using LISO to check designs for two cascaded passive LPFs it looks like the ALS signal would have to be attenuated by a factor of ~20 at DC if we don't use resistors smaller than 1k, given the low input impedence of the CM board.
ALS is the comparison of the PSL laser freq vs the end laser freq that is locked to the arm cavity resonant freq
On the other hand, the AS55 PDH is the comparison of the PSL laser freq after the IMC vs the arm cavity resonant freq. Here the PDH signal involves the arm cavity pole.
In total you observe the difference by the IMC cav pole + the arm cav pole.
To get a better look at how to do fast ALS, I took some "Plant TF" measurements of the X arm.
Specifically, in single arm POX lock and the both Y TMs misaligned, I used the SR785 to inject into EXC B of the common mode board with the CM fast output gain and IMC IN2 gain both at 0dB, and looked at the transfer function of that excitation into the analog ALSX I and AS55 Q out-of-loop signals. (ALSX I tuned to a zero crossing via the delay line box as usual.)
My expectation was to see them only differ by the IR single arm cavity pole, which should be around 8-9kHz ( FSR/450 = 3.9MHz/450 ~ 8.6kHz). The green cavity pole at ~18k shouldn't show up since we're not touching the green light, and the IMC pole at ~3.8kHz shouldn't show up since this is well within the IMC loop bandwidth and we're actuating on its error point.
Instead, I see them differ by a double pole at 4.3kHz. (or zero, if you look at it the reciprocal way). Vectfit actually fits them as a slightly complex pair, with a Q of 0.53/ I imagine that the wiggles are due to the digital control loop.
My question is: why is there a double zero here? Where has my reasoning led me astray?
At 10:02AM, the N2 Pressure fell below 60 PSI. The watch script saw this happen, but I did not recieve the email it is supposed to send
C1:Vac-P1_pressure reads 7e-4, which is the same as it has for the past ~2 days, so the V1 interlock worked fine.
I've put some fresh N2 into the system, and Bob will pop in over the weekend to check it. I'll stay on top of it until Steve gets back.
After consulting ELOGs and the 40m wiki, I reasoned it was ok to open the V1 to reconnect the turbo pump to the main IFO volume and VM1 to reconnect the RGA, and have now done so.
I hope the grappa was already cold, and ready to drink!
Look upon this three second lock, ye Mighty, and rejoice!