CVI broadband AR coating was measured at the PSL-enclosure table around 9-10am today. The 2W Innolight first PBS S polarization beam was used with an other 1/2 wave plate and PBS.
W2-PW1-1004-C-633-1064-0 This 0.045" thick window has 0.7- 0.8 % reflected beam on each sides at 5 degrees of incidence, P polarization.
The specification is R avg <0.5 % per surface at 0 degree
Rana wants The device would be useless with such a high R, but R 0.1% is OK so I will get V coating.
CVI V-AR coating at 1064 nm, 0 degree, catalog item is R< 0.25% on each sides,
R <0.1 % is custom at much higher prices.
This custom order should go with other orders that has similar need.
From CVI: 5-6-2014
I checked the trace info on the W2-PW1-1004-C-633-1064-0, BBAR coated window that you received. It is side 1, 0.42%R & side 2, 0.53%R @ 1064nm. And with the shift, I’m not too surprised you ended up with 0.7%. A V coat would start with <0.25% (and more typically coming in at ~0.1%) per surface. As far as stock options, I have a 1”dia x 4mmT, fused silica window that is recorded as side 1, R=0.09 and Side 2, R=0.08% @ 1064. Is this too think or will it work for you?