To check the basolute frequency stability of the old monochrome HP 8591E RF Spectrum analyzer that we're using for the ALS beat readout, I hooked its 10 MHz reference output (from its rear panel) into the A channel of the SRS SR620 frequency counter. The SR620 is locked to the FS 720 Rubidium clock via the 10 MHz connections in their rear panels.
So, we can assume that this is a good absolute readout. It reads 9.999860.7 +/- 0.3 Hz. So its 139.1-139.4 Hz lower than 10 MHz. The +/- 0.3 is just a slow drift that I see over the course of 10 minutes.
So, let's say that the analyzer is low by 10 ppm, so the arm length estimates are short by ~0.4 mm. A negligible correction, so there's no need to use atomic clocks to measure our arm lengths.