I guess this is normal. DARM has (almost) the same effect of MICH on the corner interferometer signals, just increased in amplitude by the arm cavity amplification. When the arm is out of resonance, DARM is almost completely depressed and it is not affecting MICH at all. On the other hand, when the cavities are exactly at resonance, DARM signal is amplified w.r.t. MICH by the cavity gain (2F/pi).
Since DARM is still controlled with ALS, it is probably noisy. The closer to resonance you move the cavities, the more ALS noise in DARM will affect MICH.
When looking at the data, I see that the MICH error signal gets fuzzier when the arms get close to resonance. (Note here that because I forgot to zero the carm offset before finding the resonances, -3 is my zero point for this plot and the next.)