We have stabilized the ALS for Y arm and concluded that although the PDH servo could be stabilized, it drifts and loses stability over a span of few hours. (See masayuki's elog today)
We wanted to follow the same systematic procedure like in the previous elog to look at the condition of the X arm as well.
In order to stabilize the green PDH servo, we held the arm using the IR PDH and aligned the end-green to the X arm.
We see 2 TEM00-like modes and one oblong TEM00+TEM01 mode that can lock to the cavity. It is not clear to me as yet as to how to differentiate between these 2 TEM-00 like modes and how we should decide between them.
One of the TEM00-like mode is strongly matched to the arm cavity. Normalized GTRX measures 0.6 counts. The other TEM00-like mode is weakly matched to the cavity. Normalized GTRX measures 0.12 counts. This might be the reason why Jenne and Masayuki were seeing a lower beat amplitude. Camera images are shown below.
On another note, we found that an oblong mode (looks like a TEM00+TEM10 mode) also locks to the cavity. The mode looks weird in that, only one half of the mode is seen moving due to seismic noise and the other part does not. I am not sure how I can describe this...so here is a 10 second video of how this mode looks like.