TRY path fixed and ready for normalization.
I used 2" BS at R=50 and R=98 to reflect the Y arm transmission at QPD-Y and TRY PD respectively. The residual beam transmitted by the BS is now steered by a Y1mirror to the camera. With Y arm locked, transmission currently measures 40mW against the expected 70mW. TRY shows 0.45 counts in dataviewer.
I think it is too much. Incident power to IFO is 1.3 W. Even if we assume no losses and pick-offs on the path to the arms, we should get ~100 uW out of the cavity. I measured X and Y arms transmission to be 60 uW. Did you disable triggering during your measurement?