|It took 11 days to fix earth quake triggered sus problems of ITMX, SRM and PRM|
Only ITMX north and BSC north vac doors were removed.
The PRM sus had to be removed form the vac envelope for "hip replacement"-new wire stand off was
epoxied in place.
Note: the PRM has no guide rod on the other side
ITMX, SRM and BS osems were optimized in place.
No crosscoupling optimization was performed.
Beam block was removed from ITMXC, it was too close to the main beam.
POX pick off mirror and mount will be replaced next vent.
Vac viewports were inspected from the inside.