While doing initial measurements for the new global damping infrastructure I discovered that the ETMY loop between the OSEM actuation and the OSEM sensors has a gain that is 2.5 times greater than the ITMY. The result is that to get the same damping on both, the damping gain on the ETMY must be 2.5 times less than the ITMY. I do not know where this is coming from, but I could not find any obvious differences between the MEDM matrices and gains.
I uploaded a screenshot of measured transfer functions of the damped ITMY and ETMY sus's. Notice that the ETMY measurement is 2.5 times higher than the ITMY. The peak also has a lower Q, despite having the same damping filters running because of this mysterious gain difference. Lowering the damping gain of the ETMY loop by this 2.5 factor results in similar Q's.