I ran Zach's arbcav on our SRC with curved TTs and the situation looks much worse than the PRC.
I used the following parameters
SRM: RoC = 142 m, T = 10%
ITM: RoC = 83.1e3 m, T = 1.4%
SRC length: 5.37 m
In this case, with TT RoC of -600, the combined cavity g-factor = 0.9986, and astigmatism from SR3 makes the cavity patently not stable. You have to go up to an RoC of -710 before the cavity is just over the edge.