- The data should be plotted in a log-log scale.
- The data points were only taken in the high current region.
- The plot may suggest that the amplifier saturate at the RF.
PDA255 has the nomial transimpedance gain of 10^4 Ohm.
The DC current of 10^-3 gives the output of 10V.
This plot may tell that the saturation starts even at the 1/10 of the full DC range.
The plot doesn't have many points below 0.1mA.
Consult with my plots for the similar measurements.
The measured points are logarithmically spaced. Use the same technique.
- It is also very unknown that how the noise level is calculated. No info is supplied in the plot or the elogs.