We replaced the MC Refl path BS with the Y1, as usual, so that the full ~100mW goes to the REFL PD, so we don't have WFS or MC refl camera.
The MC spots were all outside of 1mm, and some were beyond 2mm (for MC1,3, P,Y....MC2 is of course free since we have more DoFs than we need), so we touched (very, very slightly) the zigzag mirrors on the PSL table. We realigned the MC, and now the spots are centered to my satisfaction.
MC1,2,3 Pit, MC1,2,3 Yaw (in mm):
[0.46444020918749457, 8.2634316545130009, -0.41417975237831089, -0.89401481457980592, -0.9323196976382162, -1.543145765853893]
MC2 is way off in pitch, according to this measurement, and it's been consistently going down as we move the MC2 spot in the same direction (up on the monitor), but since we started at +15mm and are now at +8, and we've gone quite a ways, I'm not sure that we really want to go all the way to 0. Anyhow, MC1 and MC3 are the ones which define our input pointing, so we're quitting for tonight.
We will turn on the PZTs and begin with the official vent list for dummies tomorrow.