This week I took more data for the calibration of YARM. The summary of measurements taken is:
1. Peak-to-peak on Michelson
2. Michelson open loop
3. Excite ITMY and measure on AS55_Q_ERR
4. Excite ITMY and measure on POY11_I_ERR
5. Excite ETMY and measure on POY11_I_ERR
6. YARM open loop
Then I worked on comparing these measurements to the Simulink model of the interferometer control loop. The measured open loop transfer function of the YARM matched well with the model above about 20 Hz, after the gain was scaled properly to fit the data. Next is to fit the length response function of the model and the measurements, and then use DTT to calibrate the arm cavity's power spectrum.