I'm working on locking the Michelson now in order to put an excitation on one of the input test masses and measure the resulting error signal at the anti-symmetric port. I aligned the beams from ITMX and ITMY by looking at the AS camera with the video screens, but the fringes were not destructively interfering. Jenne advised that I look at the gain on the MICH servo filter modules in the LSC screen. We flipped the sign on the gain (it was 0.120 and it is now -0.120) and the fringes destructively interfered as desired after this change.
For purposes of documentation, I locked the YARM earlier in the morning before moving on to the Michelson. The purpose of this was to put another excitation on C1:SUS-ETMY_LSC_EXC and then measure the error signal on C1:LSC-POY11_I_ERR.