Most of my work has been on continuing to develop the Simulink model of the differential arm length control loop.
I have filled in transfer functions for the digital components after looking up the configuration of filters and
gains on the control screens. Filters that were active at the time included 1:50 and 1000:10 on C1LSC_YARM and
C1LSC_POY11 with a gain of 0.1. Jamie also introduced me to foton so that I could obtain the transfer functions
for the necessary filters. I have also continued to work on obtaining the open loop gain and length response
function from the model. The majority of the work now is to refine what I've accomplished so far. Adding details
to the arm cavity and the optics is one potential area for improvement.
I have also spent some time looking at real-time calibration methods from GEO and a proposal for a similar system
on LIGO in P040057-x0 from the DCC. While the work for this project may follow a different path for a real-time
calibration, having a sense for what's been accomplished so far should be helpful in working on a new system.