I tried to reduce BS 3.3 Hz motion with local damping. 3.3 Hz probably comes from the stack, but I want to reduce this because PRMI beam spot is moving in this frequency.
I tried it by putting some resonant gains to oplev servo and OSEM damping servo, but failed.
What I learned:
1. BS OSEM input matrix diagonalization looks impressively good. Below is the spectra of oplev pitch/yaw and OSEM pos/pit/yaw/side comparing with and without damping (REF is without). You can see mechanical resonances are well separated. Also, damping servos don't look like they are adding noise at 3.3 Hz.
2. 3.3 Hz motion is not stationary. Amplitude is sometimes high, but sometimes low. Amplitude changes in few seconds. You can even see 3.3 Hz in the dataviewer, too.
3. I set new oplev gains. I lowered them so that UGFs will be ~ 2.5 Hz. I turned ELP35 on.
C1:SUS-BS_OLPIT_GAIN = 0.2 (was 0.6)
C1:SUS-BS_OLPIT_GAIN = -0.2 (was -0.6)
4. All OSEM sensors feel about the same amount of 3.3 Hz motion.
5. OLPIT and OLYAW reduces if you put 3.3 Hz resonant gain to oplev servo, but it is maybe not true since they are in-loop error signals. You can't see the difference from OSEM sensors. Below is oplev pitch/yaw and OSEM pos/pit/yaw/side comparing with and without 3.3 Hz resonant gain (REF is without).