I adjusted MC WFS offsets using /opt/rtcds/caltech/c1/scripts/MC/WFS/WFS_FilterBank_offsets.
Beam spot positions on MC mirrors came back to where it was past few weeks. See the trend below. Trend sometimes shows huge jump, but it's just a bad measurement caused by unlock of MC during the measurement.
I ran /opt/rtcds/caltech/c1/scripts/ASS/MC/mcassMCdecenter to measure beam spot whenever I feel like it (see elog #6727).
Beam spot doesn't move so much (~0.2 mm in standard deviation), which means incident beam from PSL table is quite stable.