We made ETMXT camera working.
We connected the camera to video mux, placed 10% pick off mirror in front of TRX PD, lead the beam go to ETMXT camera.
Transmission to the TRY PD was 23.8 uW, but now, it's 21.3 uW (2.3 uW goes to the camera).
So, we changed C1:LSC-TRX_GAIN from -0.00181818 to -0.00203158 (=-0.00181818*23.8/21.3).
There is a channel for power normalization, C1:LSC-TRX_POW_NORM, but is 1 and it looks like we are using this gain for the normalization. Situation of TRY is the same as TRX.