[Jenne, Koji, Yuta]
We tried to scan of the Y arm but we couldn't scan for more than 1FSR.
In principle, we can do that because the error signal we are using, C1:ALS-BEATY_COARSE_I_IN1, has the range of ~ 40 MHz, which is about 10FSR (see elog http://nodus.ligo.caltech.edu:8080/40m/6815).
ALS stays for more than 10 min when we don't do the scan. If we put some offset gradually from C1ALS-OFFSETTER2, the lock breaks.
We monitored PZT output of the Y end laser, C1:GCY-SLOW_SERVO1_IN1, but it stayed in the range when scanning. So, there must be something wrong in the ALS loop.
Current in-loop arm length fluctuation is about 0.1 nm RMS (0.5 counts RMS).
Below is the spectrum of the error signal when the ALS is off(green) and on (pink,red). Below ~ 50 Hz, the measurement of the Y arm length is limited by ADC noise (~ 2uV/rtHz).